Effect of surgeon on transprosthetic gradients after aortic valve replacement with Freestyle® stentless bioprosthesis and its consequences: A follow-up study in 587 patients
© Albert et al; licensee BioMed Central Ltd. 2007
Received: 06 January 2007
Accepted: 05 October 2007
Published: 05 October 2007
The implantation of stentless valves is technically demanding and the outcome may depend on the performance of surgeons. We studied systematically the role of surgeons and other possible determinants for mid-term survival, postoperative gradients and Quality of Life (QoL) after aortic valve replacement (AVR) with Freestyle® stentless bioprostheses.
Between 1996 and 2003, 587 patients (mean 75 years) underwent AVR with stentless Medtronic Freestyle® bioprostheses. Follow-up was 99% complete. Determinants of morbidity, mortality, survival time and QoL were evaluated by multiple, time-related, regression analysis. Risk models were built for all sections of the Nottingham Health Profile (NHP): energy, pain, emotional reaction, sleep, social isolation and physical mobility
Actuarial freedom from aortic valve re-operation, structural valve deterioration, non-structural valve dysfunction, prosthetic valve endocarditis and thromboembolic events at 6 years were 95.9 ± 2.1%, 100%, 98.7 ± 0.5%, 97.0 ± 1.5%, 79.6 ± 4.3%, respectively. The actuarial freedom from bleeding events at 6 years was 93.1 ± 1.9%. Estimated survival at 6 years was similar to the age-matched German population (61.4 ± 3.8 %). Predictors of survival time were: diabetes mellitus, atrial fibrillation, peripheral vascular disease, renal dysfunction, female gender > 80 years and patients < 165 cm with BMI < 24. Predictive models showed characteristic profiles and good discriminative powers (c-indexes > 0.7) for each of the 6 QoL sections. Early transvalvular gradients were identified as independent risk factors for impaired physical mobility (c-index 0.77, p < 0.002). A saturated propensity score identified besides patient related factors (e.g. preoperative gradients, ejection fraction, haematological factors) indexed geometric orifice area, subcoronary implantation technique and individual surgeons as predictors of high gradients.
In addition to the valve size (in relation to body size), subcoronary technique (versus total root) and various patient-related factors the risk of elevated gradients after stentless valve implantation depends, considerably on the individual surgeon.
Although there was no effect on survival time and most aspects of QoL, higher postoperative transvalvular gradients affect physical mobility after AVR.
Stented bioprostheses are considered at risk of structural failure and a non physiological flow pattern. Since the rigid stent is considered to be incremented in these disadvantages, stentless bioprostheses were developed. During the last years stentless bioprostheses have been used frequently and clinical outcome has been demonstrated [1–3].
The implantation of stentless valves is considered as technically more demanding. Nevertheless no increase in peri-operative risk in comparison to the implantation of stented bioprostheses or mechanical valves was observed in our own  and the experience of others [5–8] and favourable hemodynamic performance of the stentless valves has been demonstrated [1, 2, 9–11]. However, different groups have observed a heterogeneity in transvalvular Doppler gradients early after implantation and some patients have been found to have higher transvalvular gradients than anticipated [1, 12]. It is still unclear to what extent patient related factors (e.g. preoperative gradient), characteristics of the early postoperative period like elevated stroke volume, local oedema and hematoma or other surgical factors are associated with this phenomena. ; however it was assumed that elevated postoperative gradients observed in stentless valves depend to a larger extent on the surgeon's skill and experience [1, 12]. It is also a matter of debate if higher gradients early after surgery are just a transient phenomena or are persistent over time, affecting clinical outcome [1, 12–14].
The purpose of the present study is to estimate the clinical importance of the individual surgeons for quality of life (QoL) and survival after aortic valve replacement (AVR) with Freestyle® stentless bioprostheses. Thereby early postoperative gradients, the relationship between prosthesis-size and patient size, the implantation technique (subcoronary versus total root) and a wide spectrum of patient's characteristics including online accessible laboratory values were used as risk-adjustment variables in multivariate analysis.
Between April 1996 and December 2003, 587 patients older than 60 years underwent AVR with the stentless biological Medtronic Freestyle® Prosthesis (for age distribution see additional file 1). This group of patients represents 31 % of all patients receiving aortic valve prosthesis in this age group in our centre. The choice of the prosthesis type was according to surgeon's preference and patient's choice after informed consent. Additional files 2&3 show the operative and preoperative characteristics of the patient population. All patients have signed an informed consent for the operation, for quality control measures and the follow-up studies.
The indexed geometric orifice area (IGOA) were calculated by the internal diameters for corresponding valve sizes reported by the company [Data from Medtronic: Valve size 19, 21, 23, 25, 27 – Internal Diameters 16, 18, 20, 21.5, 23.5] divided by BSA.
Technique of implantation and the individual surgeons
All operations were performed using standard cardio-pulmonary bypass techniques with systemic normothermia and both antegrade and retrograde hyperkalemic cold blood cardioplegia. The subcoronary implantation was performed using a 4/0 Prolene continuous running suture for both, the first and second suture line. The prosthesis usually was 120° rotated in order to place the Dacron covered muscular part of the implant towards the human non-coronary sinus. Total root implantation was performed using a 4/0 Prolene continuous running suture for the annular implantation and distal anastomoses to the aorta and a 5/0 Prolene suture for coronary ostia re-implantation. The individual surgeons were included into the multivariate models as nominal data. The majority of the stentless valves were implanted by 5 surgeons (surgeons A-E), one surgeon has implanted only 28 valves (surgeon F); a group of younger surgeons, who operated a total of 15 valves, were subsumed as a mixed group M.
Intraoperative transesophageal echocardiography and transthoracic echocardiographic control before hospital discharge (4 to 7 days after AVR) under rest were performed by experienced cardiologists. Gradients from the pre-discharge echocardiography were used as variable for the present study. Mean transvalvular gradient was calculated by continuous wave Doppler using the modified Bernoulli equation. In cases with unusual higher gradients or other irregular findings a second opinion by a colleague was asked for.
Follow-up information was obtained 6 month after surgery, in 2000, 2001 and 2003 by mailed questionnaires and completed by telephone interviews. Follow-up was 99% complete. Mean follow-up time was 32 ± 23 months. The follow-up questionnaire consisted of a quality of life (QoL) assessing questionnaire, the Nottingham Health Profile (NHP) , and general questions concerning postoperative complications, further hospitalisation and NYHA status.
Statistical analysis was performed using the software package SPSS (SPSS Inc, Chicago, IL).
For each patient included in the present study, 49 pre-operative characteristics were retrieved from the clinical information system and the consolidated database of our Data Mart system . The data is based on continuous, online input from the anaesthesiological, cardio-surgical quality assurance and the laboratory data of the clinical chemistry [Tables a1, a2, additional files].
Valve-related morbidity was estimated by life-table analysis
By logistic regression the following categorical outcome variables were analysed: 30-days- and 6-months-mortality and QoL. NHP-scores above the values of the general German population of men and women of the same age  were defined as an impaired QoL and were coded by 1. We did not analyze the QoL as continuous data, because the NHP-scores were not normally distributed. In order to assess the effect of missing data in the models for Quality of Life, we have recalculated all models. Firstly, we inputted all missing data as "impaired" (worst case); secondly, we inputted all missing data as "non impaired" (best case).
Predictors for survival time were identified by Cox-regression analysis
For variable selection of all multivariate regression models, the Akaike Information Criterion (AIC = Deviance of the model + 2 * number of included parameters) was calculated for variables showing a difference for the outcome variable with a p value smaller than or equal to 0.25. The variables were included into multiple regression models in a stepwise way. The AIC was calculated each time a variable was included. The final model was reached when no more reduction in AIC was observed. To establish linear dependency of outcome from continuous variables, patient population was divided into subgroups of the same size. The logit of the outcome variable was calculated within each group. In the case of a nonlinear increase or decrease of an event, the AIC of the models was determined for several cut points or quadratic relations.
For Cox-analysis the different Kaplan-Meier curves were inspected after dividing the patient population in equally sized subgroups. The relevant cut points were then determined by minimization of the AIC of the Cox-model.
76% of all survivors (356 patients) responded to the mailed questionnaire, assessing NYHA-status and QoL and those who did not respond were contacted by telephone. Of all non-responders (N = 113) the reasons for non-responding were non-cardiac-illness (8%), cardiac illness (19%), disagreement (34%), staying abroad (2%) and without reason (willingly answered on the phone, however, did not return the questionnaire) (38%).
Actuarial freedom from aortic valve re-operation, structural valve deterioration, non-structural valve dysfunction, prosthetic valve endocarditis and thrombembolic events at 6 years were 95.9 ± 2.1%, 100%, 98.7 ± 0.5%, 97.0 ± 1.5%, 79.6 ± 4.3%, respectively. In 1389 patient-years, twelve patients had to be re-operated (0.9%/pt-yr). The indications to re-operation were mainly due to outflow obstruction. No significant aortic insufficiency was observed (additional file 4). The quality of life after re-replacement of aortic valve was impaired for the sections energy, pain, sleep and physical mobility.
80 (16.8%) patients required anticoagulation therapy due to co-morbidities and 15 major bleeding events were observed (1.1%/pt-yr). The actuarial freedom from bleeding events at 6 years was 93.1 ± 1.9%.
Predictors of valve size
Male patients usually received larger stentless prostheses than female patients: valve size 23 and 25 was implanted in 73% of the male patients and valve size 21 and 23 in 86% of the female patients.
The linear regression model revealed subcoronary implantation technique (p < 0.001), female gender (p < 0.001, body height (p < 0.001), age (p = 0.02), BMI (p = 0.046), re-replacement of the aortic valve (p = 0.01) as predictors for valve size.
Determinants of mortality and survival time
Predictors of 30-day- and 6-months-mortality and survival time. Apparently valve size is a risk factor in the first step of the analysis. However, after adjustment for gender, age, BSA, BMI and body size, valve size disappeared (p < 0.05)
Preoperative risk factors
Peripheral vascular disease
Antithrombine III (%)
Urea concentration (mg/dl)
p = 0.009
p = 0.05
p = 0.02
After Variable transformation:
Female and age > 74 years
BSA < 1.78 m2
Female and age > 80 years
BMI < 24 and body size < 165 cm
p = 0.13
p = 0.47
p = 0.22
After adjusting for these factors, valve size was no longer a risk factor in all 3 models (table 1), nor was indexed geometric orifice area.
Determinants of QoL
The percentage of patients after AVR with NHP-scores higher than the age and gender matched general German population was 43% in the section energy, 25% in the section pain, 29% in the section emotional reaction, 27% for sleep, 18% for social isolation and 41% for physical mobility.
Significant predictors of impaired QoL are listed
Emotional reaction [OR;95%CI]
Social isolation [OR;95%CI]
Physical mobility [OR;95%CI]
(> 76), 2.8; 1.7–4.9
(> 79), 2.6; 1.4–4.8
(> 70), 5.6; 1.9–16.7),
Lower potassium 1
Higher creatinine 2
History of syncope
(< 13) 2.0; 1.2–3.5
History of MI 4
De Ritis-ratio (AST/ALT) 6
Pacemaker before AVR
Mean gradients (1 mmHg)
Including the size of implanted valves and the indexed geometric orifice area did not improve the models significantly (p < 0.1).
In order to assess the effect of missing data in the models for Quality of Life, we have recalculated all models for the worst and best case scenarios, where all missing data were imputed as "impaired QoL" respectively "non impaired QoL". In both, the worst- and best case scenarios, the identified risk factors for Quality of Life did not change and the c-indexes for all models decreased, suggesting that firstly, the models have enough stability and secondly, that the non responding patients, have a mixed risk profile, not exclusively belonging to the high risk or low risk group.
16% of the patients (N = 58) took coumarin during follow-up, mainly due to atrial fibrillation.
Impact of early postoperative gradient on mortality and QoL
Mean and maximum transvalvular pressure gradients (mmHg) at discharge (5 to 7 days after AVR)
Total root replacement
29.8 ± 14.4
54.2 ± 23.2
22.0 ± 8.9
40.4 ± 14.5
13.7 ± 4.2
25.7 ± 7.5
19.6 ± 7.6
35.5 ± 12.8
13.5 ± 6.0
25.3 ± 10.1
17.0 ± 6.9
31.0 ± 14.9
10.8 ± 5.0
19.8 ± 10.9
14.8 ± 5.7
26.7 ± 9.4
7.2 ± 2.0
13.2 ± 4.3
The postoperative mean and maximum transvalvular pressure gradients had no impact on the QoL sections energy (p = 0.54, p = 0.33), pain (p = 0.34, p = 0.19), emotional reaction (p = 0.33, p = 0.08), sleep (p = 0.9, p = 0.89) and social isolation (p = 0.71, p = 0.81, respectively for mean and maximum pressure gradients). However, increasing postoperative mean transvalvular pressure gradients were identified as independent risk factors for impaired physical mobility (p = 0.002, Table 2).
In a logistic regression model for non-responding the postoperative mean and maximum transvalvular pressure gradients were no risk factors (p-values for model improvement: 0.69 and 0.42, respectively).
Risk factors for higher transprosthetic gradients and influence of the surgeons
Saturated propensity score predicting a mean transvalvular gradient after AVR > 20 mmHg
Body height (cm)
Subcoronary implantation technique
Indexed geometric orifice area
Ejection fraction (%?)
Preoperative maximum transvalvular gradient (mmHg)
White blood cell count (cells*1000/μl)
Total serum protein (g/dl)
Glutamic-oxalacetic transaminase (U/l)
Mean corpuscular volume (fl)
Chronic pulmonary disease
Peripheral occlusive arterial disease
Experience of the surgeons at end of 2004
Years in Cardiac Surgery
Major cardiac surgery performed [N]
Valve implantation performed [N]
Freestyle valve implantation performed [N]
Mean transprosthetic gradients in subcoronary technique (ANOVA, p < 0.001)
Mean transvalvular gradients for the main valve sizes by surgeon
Mean ± StDev [mmHg]
Mean ± StDev [mmHg]
Mean ± StDev [mmHg]
19 ± 2.4
16 ± 1.8
17 ± 2.3
24 ± 2.3
22 ± 1.7
16 ± 2.3
25 ± 1.9
21 ± 1.3
19 ± 1.8
21 ± 1.1
18 ± 0.8
16 ± 0.9
23 ± 2.7
26 ± 1.7
19 ± 2.8
20 ± 1.9
17 ± 1.4
13 ± 1.7
10 ± 8.2
12 ± 2.0
12 ± 2.6
P = 0.1
p < 0.001
P = 0.1
Encouraged by previous reports about the excellent hemodynamic performance and the good initial experience, stentless valves are frequently implanted by all surgeons in our institution.
The 30 day mortality in our study equals to the reports of a multicenter evaluation of the Freestyle valve (3.1% versus 3.0%)  and the survival rate, after 5 years was similar to a series with a comparable mean age of 75 years  (73%, versus 72% in our study). In accordance with other studies we observed virtually no structural valve deterioration or important aortic insufficiency with the Freestyle stentless valves .
Nevertheless, we observed in a considerable percentage of patients higher transvalvular gradients. They were in average higher than reported previously [1, 1, 3, 11, 20, 21], what may be partly explained by the more frequent use of subcoronary technique in patients with small aortic roots., but also previous studies reported a heterogeneity of the transvalvular gradients and suboptimal hemodynamics with the freestyle stentless valve early postoperative [1, 12, 13, 22]. Up to now, both the risk factors for higher gradients and their clinical relevance are still poorly defined. It was argued that echocardiography tends to overestimate the transvalvular gradient in stentless valves, because the velocity profile of stentless valves is more parabolic in the early postoperative period,  and furthermore that elevated gradients will decline by resolution of paravalvular haematoma and the normalization of postoperative hemodynamics. [1, 3, 12]. On the other hand, previous studies showed that transvalvular gradients remained high at follow-up and left ventricular mass regression was subsequently less [12, 22]. Up till now, a possible association of postoperative gradients with clinical symptoms was not studied in stentless valves. Even, in other valve types, the clinical significance of residual left ventricular outflow tract obstruction after AVR is still under debate .
We could not detect an impact of neither the early postoperative transvalvular gradient nor the IGOA on morbidity or mid-term mortality, but physical mobility at follow-up was impaired with increasing mean transvalvular gradient especially with mean gradients > 20 mmHg. Also our finding that apart from physical mobility, none of the other NHP sections: energy, pain, emotional reaction, sleep, or social isolation were affected, indicates an exercise dependent mechanism similar to what occurs in patients with hemodynamically important stenosis. Previous studies have suggested that the residual transprosthetic pressure gradient results in an increased LV workload, thus hampering the regression of LV mass after AVR and subsequently functional recovery after AVR [24, 26]. In addition it was demonstrated that, in patients with AS and angiographically normal coronary arteries, the improvement of coronary flow reserve after AVR is directly dependent on the improvement of valve EOA that is achieved with AVR . Hence, the increased LV systolic pressure associated with residual transvalvular gradients may compromise the normalization of coronary flow reserve after AVR. Both mechanisms predispose to decreased exercise tolerance and explain our finding of an impaired physical mobility in patients with higher gradients after AVR.
In a recent study no risk-adjusted impact of mean transvalvular gradients or IGOA on functional health-related QoL was detected after AVR with mainly stented valves . The Duke Activity Status Index (DASI), used in this study to measure activities of daily living such as household tasks, ambulation or personal care, shows similarities to the section physical mobility of the NHP. Nevertheless, this study was different from ours regarding the study design. It may be possible that, due to the fact that the mean follow-up time in our study was more than 3 times longer than in the above mentioned study, we were able to prove an impact of the postoperative gradient on QoL.
According to our data in stentless valves higher gradients depend beyond the physical valve sizes and its relation to body size, mainly on the interaction of subcoronary technique, the preoperative gradients with the individual surgeons.
By solving the multivariate equations in figure 2 we had illuminated how with total root technique virtually no increase of the risk for higher gradients with decreasing IGOA was observed, whereas with subcoronary technique the risk increase exponentially. Thus previous reports could be redefined, where a threshold of a valve size equal or smaller than 23 was mentioned as risk factor for higher gradients [1, 12]. In contrary to a study about the outcome after human tissue valves for aortic valve replacement, a limited performance of the individual surgeon in subcoronary technique did not result in more initial aortic regurgitation and early reoperation our series. 
It was already assumed that the performance of stentless valves depends to a larger extent on the surgical experience [1, 12]. Up to now only data were available concerning 30 day mortality, where a significant decrease with operator experience for both subcoronary and total root technique was described . In our study no significant effect of individual surgeons on mortality was detectable, but we were able to show to which degree the hemodynamic performance of stentless valves depends on the individual surgeon and proved its clinical consequences. In contrary to the study about learning curve of tissue valve implantation, where an experience of more than 38 cases was associated with better performance, we could not define the number of cases needed, until the constantly better gradients where achieved, due to large variations between the surgeons; their skills and differences in their individual histories of training seems to be more important than experience over time and the number of cases.
According to our and others experiences [1, 12] higher transvalvular gradients through stentless valves may develop by already slight distortions of the valve, horizontal or vertical folding of valve tissue into the outflow tract, oversizing, impaired movements of the non-coronary cusp or due to paravalvular hematoma. Especially in patients with a discrepancy between annulus and sinotubular junction dimensions, calcification in the aortic sinuses, unusual angles between the coronary ostia, bicuspid valves or small aortic roots, the difficulty of stentless valves implantation increases whereas the reproducibility decreases.
In our study we considered both transvalvular gradients and IGOA as possible determinant of outcome, but the Effective Orifice Area (EOAs) was not assessed. Nevertheless a correlation of the mean transvalvular gradients with the EOA, the impedance to the left ventricular outflow and left ventricular mass regression is well established for native valves, conventional prosthesis  and stentless xenografts, in rest and in exercise [1, 3, 8, 12, 13, 31]. Despite its known limitations, an accurate measurement of the pressure gradient is still sufficient to make clinical decision . In order to adjust for remaining discrepancies between transvalvular gradient and EOAs, we adjusted for ejection fraction and other variables known to influence the gradients like blood composition, ejection fraction, atrial fibrillation and preoperative gradients.
The implantation of Freestyle stentless valves in subcoronary technique is technically demanding and therefore affords an extraordinary expertise, especially in managing small valve sizes (in relation to BSA), cases with a difficult aortic root geometry, sinus calcifications and severe left ventricular hypertrophy. Transvalvular gradients rise easily when the valve does not smoothly fit within the aortic root (whereas aortic insufficiency is extremely uncommon). Despite the well-known limitations of the transvalvular gradient for measuring performance of aortic valves, in the majority of patients it reflected sufficiently the technical quality of implantation and correlated significantly with physical mobility years after the operation. In order to avoid unfavourable performance curves ("learning curves") in technically demanding procedures like stentless valve implantation it is recommendable to apply more profoundly the recognized methods of surgical teaching (e.g. prolonged expert guidance, cognitive task analysis, model training ) and thereby optimizing the diffusion of such a new technology .
- Jin XY, Westaby S: In vivo hemodynamic characteristics of porcine stentless aortic valves. Semin Thorac Cardiovasc Surg. 2001, 13: 67-74.PubMedGoogle Scholar
- Kunadian B, Vijayalakshmi K, Thornley AR, de Belder MA, Hunter S, Kendall S, Graham R, Stewart M, Thambyrajah J, Dunning J: Meta-analysis of valve hemodynamics and left ventricular mass regression for stentless versus stented aortic valves. Ann Thorac Surg. 2007, 84: 73-9. 10.1016/j.athoracsur.2007.02.057.View ArticlePubMedGoogle Scholar
- Bach S, Kon ND, Dumesnil JG, Sintek CF, Doty DB: Eight-year results after aortic valve replacement with the Freestyle stentless bioprosthesis. J Thorac Cardiovasc Surg. 2004, 127: 1657-63. 10.1016/j.jtcvs.2004.01.023.View ArticlePubMedGoogle Scholar
- Florath I, Albert A, Rosendahl U, Alexander T, Ennker IC, Ennker J: Mid term outcome and quality of life after aortic valve replacement in elderly people: mechanical versus stentless biological valves. Heart. 2005, 91: 1023-9. 10.1136/hrt.2004.036178.View ArticlePubMedPubMed CentralGoogle Scholar
- Van Nooten G, Caes F, Francois K, Belleghem YV, Taeymans Y: Stentless or stented valve implants in elderly patients?. Eur J Cardio-Thorac. 1999, 15: 31-6. 10.1016/S1010-7940(98)00288-7.View ArticleGoogle Scholar
- David TE, Puschmann R, Ivanov J, Bos J, Armstrong S, Feindel CM, Scully HE: Aortic valve replacement with stentless and stented porcine valves: a case-match study. J Thorac Cardiovasc Surg. 1998, 116: 236-41. 10.1016/S0022-5223(98)70122-9.View ArticlePubMedGoogle Scholar
- Del Rizzo DF, Freed D, Abdoh A, Doty D, Goldman BS, Jamieson WR, Westaby S: Midterm survival of stented versus stentless valves: does concomitant coronary artery bypass grafting impact survival?. Semin Thorac Cardiovasc Surg. 2001, 13: 148-55.PubMedGoogle Scholar
- Cohen G, Christakis GT, Joyner CD, Morgan CD, Tamariz M, Hanayama N, Mallidi H, Szalai JP, Katic M, Rao V, Fremes SE, Goldman BS: Are stentless valves hemodynamically superior to stented valves? A prospective randomized trial. Ann Thorac Surg. 2002, 73: 767-75. 10.1016/S0003-4975(01)03338-0.View ArticlePubMedGoogle Scholar
- Silberman S, Shaheen J, Merin O, Fink D, Shapira N, Liviatan-Strauss N, Bitran D: Exercise hemodynamics of aortic prostheses: comparison between stentless bioprostheses and mechanical valves. Ann Thorac Surg. 2001, 72: 1217-21. 10.1016/S0003-4975(01)03064-8.View ArticlePubMedGoogle Scholar
- Dumesnil JG, LeBlanc MH, Cartier PC, Metras J, Desaulniers D, Doyle DP, Lemieux MD, Raymond G: Hemodynamic features of the freestyle aortic bioprosthesis compared with stented bioprosthesis. Ann Thorac Surg. 1998, 66: S130-3. 10.1016/S0003-4975(98)01119-9.View ArticlePubMedGoogle Scholar
- Yun KL, Jamieson WR, Khonsari S, Burr LH, Munro AI, Sintek CF: Prosthesis-patient mismatch: hemodynamic comparison of stented and stentless aortic valves. Semin Thorac Cardiovasc Surg. 1999, 11: 98-102.PubMedGoogle Scholar
- Bach DS, Cartier PA, Kon N, Johnson KG, Dumesnil JG, Doty DB: Impact of high transvalvular to subvalvular velocity ratio early after aortic valve replacement with Freestyle stentless aortic bioprosthesis. Sem Thorac Cardiovasc Surg. 2001, 13: 75-81.Google Scholar
- Bach DS, Lemire MS, Eberhart D, Armstrong WF, Deeb GM: Impact of high transvalvular velocities early after implantation of Freestyle stentless aortic bioprosthesis. J Heart Valve Dis. 2000, 9: 536-43.PubMedGoogle Scholar
- Kon ND, Westaby S, Amarasena N, Pillai R, Cordell AR: Comparison of implantation techniques using freestyle stentless porcine aortic valve. Ann Thorac Surg. 1995, 59: 857-62. 10.1016/0003-4975(95)00066-T.View ArticlePubMedGoogle Scholar
- Kohlmann T, Bullinger M, Kirchberger-Blumstein I: Die deutsche Version des Nottingham Health Profile (NHP): Übersetzungsmethodik und psychometrische Validierung. Soz.-Präventivmed. 1997, 42: 175-85. 10.1007/BF01300568.View ArticlePubMedGoogle Scholar
- Arnrich B, Walter J, Albert A, Ennker J, Ritter H: Data Mart based research in heart surgery: challenges and benefit. Medinfo. 2004, 8-12.Google Scholar
- Hinz A, Klaiberg A, Schumacher J, Brähler E: The psychometric quality of the Nottingham Health Profile (NHP) in the general population. Psychother Psych Med. 2003, 53: 353-8. 10.1055/s-2003-40948.View ArticleGoogle Scholar
- Bach DS, Cartier PC, Kon ND, Johnson KG, Deeb GM, Doty DB, Freestyle Valve Study Group: Impact of implant technique following freestyle stentless aortic valve replacement. Ann Thorac Surg. 2002, 74: 1107-13. 10.1016/S0003-4975(02)03832-8.View ArticlePubMedGoogle Scholar
- Doty DB, Cafferty A, Cartier P, Huysmans A, Kon ND, Krause AH, Millar RC, Sintek CF, Westaby S: Aortic valve replacement with Medtronic Freestyle bioprosthesis: 5-year results. Sem Thorac Cardiovasc Surg. 1999, 11: 35-41.Google Scholar
- Rao V, Jamieson WR, Ivanov J, Armstrong S, David TE: Prosthesis-patient mismatch affects survival after aortic valve replacement. Circulation. 2000, 102: III5-III9.View ArticlePubMedGoogle Scholar
- Yun KL, Sintek CF, Fletcher AD, Pfeffer TA, Kochamba GS, Hyde MR, Torpoco JO, Khonsari S: Aortic valve replacement with the Freestyle stentless bioprosthesis: Five-Year experience. Circulation. 1999, 100: II17-II23.PubMedGoogle Scholar
- Matsue H, Sawa Y, Matsumiya G, Matsuda H, Hamada S: Mid-term results of freestyle aortic stentless bioprosthetic valve: clinical impact of quantitative analysis of in-vivo three-dimensional flow velocity profile by magnetic resonance imaging. J Heart Valve Dis. 2005, 14: 630-6.PubMedGoogle Scholar
- David TE: Is prosthesis-patient mismatch a clinically relevant entity?. Circulation. 2005, 111: 3186-7. 10.1161/CIRCULATIONAHA.105.545657.View ArticlePubMedGoogle Scholar
- Mohty-Echahidi D, Malouf JF, Girard SE, Schaff HV, Grill DE, Enriquez-Sarano ME, Miller FA: Impact of prosthesis-patient mismatch on long-term survival in patients with small St Jude Medical Mechanical prostheses in aortic position. Circulation. 2006, 113: 420-426. 10.1161/CIRCULATIONAHA.105.546754.View ArticlePubMedGoogle Scholar
- Pibarot P, Dumesnil JG, Lemieux M, Cartier P, Metras J, Durand LG: Impact of prosthesis-patient mismatch on hemodynamic and symptomatic status, morbidity and mortality after aortic valve replacement with a bioprosthetic heart valve. J Heart Valve Dis. 1998, 7 (2): 211-8.PubMedGoogle Scholar
- Ali NF, Mahadevan VS, Muir A, Maguire C, Young D, Campalani G, Campbell NP, Danton MH: The influence of prosthesis size and design on exercise dynamics after aortic valve replacement. J Heart Valve Dis. 2006, 15 (6): 755-62.PubMedGoogle Scholar
- Rajappan K, Rimoldi OE, Camici PG, Bellenger NG, Pennell DJ, Sheridan DJ: Functional changes in coronary microcirculation after valve replacement in patients with aortic stenosis. Circulation. 2000, 107 (25): 3170-5. 10.1161/01.CIR.0000074211.28917.31.View ArticleGoogle Scholar
- Koch CG, Khandwala F, Estafanous FG, Loop FD, Blackstone EH: Impact of prosthesis-patient size on functional recovery after aortic valve replacement. Circulation. 2005, 111: 3221-9. 10.1161/CIRCULATIONAHA.104.505248.View ArticlePubMedGoogle Scholar
- Willems TP, Takkenberg JJ, Steyerberg EW, Kleyburg-Linkers VE, Roelandt JR, Bos E, van Herwerden LA: Human tissue valves in aortic position: determinants of reoperation and valve regurgitation. Circulation. 2001, 103: 1515-21.View ArticlePubMedGoogle Scholar
- Feigenbaum H: Hemodynamik information derived from echocardiography. Echocardiography. Edited by: Feigenbaum H. 1994, LEA& Febiger, Philadelphia, 181-215. 5Google Scholar
- Chambers J, Rimington H, Rajani R, Hodson F, Blauth C: Hemodynamic performance on exercise: comparison of a stentless and stented biological aortic valve replacement. J Heart Valve Dis. 2004, 3: 729-33.Google Scholar
- Wanzel KR, Ward M, Reznick RK: Teaching the surgical craft: From selection to certification. Curr Probl Surg. 2002, 39: 573-659. 10.1067/mog.2002.123481.View ArticlePubMedGoogle Scholar
- Edmondson Amy C, Winslow Ann B, Bohmer Richard MJ, Pisano Gary P: Learning how and learning what: effect of tacit and codified knowledge on performance improvement following technology adoption. Decision Sciences. 2003, 34: 197-222. 10.1111/1540-5915.02316.View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.