The effectiveness of rigid pericardial endoscopy for minimally invasive minor surgeries: cell transplantation, epicardial pacemaker lead implantation, and epicardial ablation
© Kimura et al.; licensee BioMed Central Ltd. 2012
Received: 30 March 2012
Accepted: 3 November 2012
Published: 9 November 2012
The efficacy and safety of rigid pericardial endoscopy as the promising minimally invasive approach to the pericardial space was evaluated. Techniques for cell transplantation, epicardial pacemaker lead implantation, and epicardial ablation were developed.
Two swine and 5 canines were studied to evaluate the safety and efficacy of rigid pericardial endoscopy. After a double pericardiocentesis, a transurethral rigid endoscope was inserted into the pericardial space. The technique to obtain a clear visual field was examined, and acute complications such as hemodynamic changes and the effects on intra-pericardial pressure were evaluated. Using custom-made needles, pacemaker leads, and forceps, the applications for cell transplantation, epicardial pacemaker lead implantation, and epicardial ablation were also evaluated.
The use of air, the detention of a stiff guide wire in the pericardial space, and the stretching of the pericardium with the rigid endoscope were all useful to obtain a clear visual field. A side-lying position also aided observation of the posterior side of the heart. As a cell transplantation methodology, we developed an ultrasonography-guided needle, which allows for the safe visualization of transplantation without major complications. Pacemaker leads were safely and properly implanted, which provides a better outcome for cardiac resynchronizing therapy. Furthermore, the success of clear visualization of the pulmonary veins enabled us to perform epicardial ablation.
Rigid pericardial endoscopy holds promise as a safe method for minimally invasive cell transplantation, epicardial pacemaker lead implantation, and epicardial ablation by allowing clear visualization of the pericardial space.
KeywordsRigid pericardial endoscopy Minimally invasive surgery Cell transplantation Epicardial pacemaker lead implantation Epicardial ablation
Although tissue engineering research has developed various techniques to establish stem cell differentiation, there have been no established techniques to transplant these resources. The intracoronary infusion of cell resources has been clinical used for acute myocardial infarction , ischemic heart disease , and nonischemic cardiomyopathy . However, this approach faced major complications due to microvascular embolization . Intramyocardial delivery, which is a direct injection into the myocardium, is more beneficial in cell retention and therapeutic effects compared to systemic, intracoronary, or intravenous infusion methods [4, 5]. It can be achieved by a transepicardial or transendocardial approach . Transepicardial delivery is usually performed by way of a sternotomy during coronary artery bypass , subdiaphragmatic access , or coronary sinus . Transendocardial delivery requires a retrograde left ventricular approach via the femoral artery . For this purpose, the NOGA XP Cardiac Navigation system™ (Biologics Delivery Systems Group, Cordis Corporation, Diamond Bar, CA, USA)  and the MyoStar™ catheter (BiosenseWebster, Diamond Bar, CA, USA)  are used as cell delivery navigation systems. These systems allow recognition of the specific areas upon which to deliver laser energy , genes , and cells  under magnetic and electromechanical guidance; moreover, these methods have been reported to be effective for ischemic  and nonischemic disease . However, there is still a risk of microvascular embolization , and the limitations due to positional errors during mapping remain to be solved.
Therefore, we have evaluated the feasibility of flexible pericardial endoscopy as a minimally invasive intramyocardial delivery method with direct visualization of the pericardial space . The flexible endoscope was inserted from the subxyphoid area by using the Seldinger technique under fluoroscopic guidance. The small pericardial space was inflated by air to obtain a clear visual field to maintain a certain distance between the endoscope and the heart. The flexible pericardial endoscope led to no acute or chronic complications. Too much air in the pericardial space induced transient blood suppression, which was called “air tamponade”. This was immediately resolved by suctioning the excess air. The endoscopic image was so vivid no other guidance systems were necessary for minor surgery applications.
Direct endoscopic imaging is free from the limitations of mapping error and can be a promising technique for minor surgery in the pericardial space. To explore these advantages and to establish pericardial endoscopic surgery, we evaluated the use of rigid pericardial endoscopy for a better visual field and easier access to the pericardial space. In this study, we invented a safer ultrasonography-guided cell transplantation system. We focused on further optimization of epicardial approach devices to treat arrhythmias, which included epicardial pacemaker lead implantation and epicardial ablation.
All of the experimental protocols were approved by the institutional ethical committee.
A double sheath puncture procedure
Five canines weighing 18–25 kg and 2 swine weighing 30–35 kg were studied. Following the infusion of 0.5 mg/kg pentobarbital, the animals were intubated and ventilated with room air by a constant-volume cycled respirator (Model SN-480-3, Shinano Inc., Tokyo, Japan). The animals were anesthetized with 1.5% halothane, and a 50 ml/h normal saline infusion was used to compensate for body fluid loss and drug delivery. The right femoral arterial blood pressure, electrocardiogram, and pulse oximetry were continuously monitored during the procedures.
After local anesthesia, sheaths were placed in the pericardial space by using the modified Seldinger technique . An 18 Gauge needle for epidural anesthesia was used for percutaneous pericardiocentesis from the epigastric fossa under ultrasonography and fluoroscopy guidance. After safely puncturing the pericardium with a puff of contrast medium, a guide wire (outer diameter (OD) = 0.35 mm) was inserted into the pericardial space. The cutaneous hole was gradually upsized by several polypropylene dilators (8 French, 12 French). Afterwards, 2 types of sheaths, which are described below, were inserted. For a double sheath maneuver, the second sheath was inserted by repeating the same procedure. The second puncture site was directed as far as possible from the first site to avoid interference of both sheaths.
Two kinds of rigid endoscopes (Model WA50373B and A22003A, Olympus Medical Systems Corporation, Tokyo, Japan) were used (Figure 1, R-30, R-70). These endoscopes were originally designed for transurethral resection of cancers. These endoscopes provided a 30° (R-30) and a 70° (R-70) cross-eyed visual field. The R-30 endoscope provided a 70° viewing angle and the R-70 endoscope provided a 90° viewing angle. The viewing angles were sufficiently wide to observe the surface of the heart at close range. Other information, such as model numbers, sizes, and specifications are shown in Figure 1.
We used 2 types of handmade sheaths: a smaller sheath (Figure 1, Sh-S) for the rigid endoscopes and a larger sheath (Figure 1, Sh-L) for forceps and needles. The sheaths were fluoroscopically visible and were equipped with a check valve to seal the air in the pericardium. A clasp for the guide wire was also attached to prevent the unintentional extraction of the guide wires.
We arranged 2 types of guide wires; a soft, plastic, readymade guide wire was used for the Seldinger technique, which was later replaced by a stiff guide wire (Radifocus™ Guide Wire M, Terumo Corporation, Tokyo, Japan) that was detained inside of the pericardial space and fixed by the wire clasp during sessions.
Two types of needles for cell transplantation were designed by modifying abdominal forceps. A needle with a grip (Figure 1, Nd-G) was inserted through Sh-L; this consisted of a 25 Gauge needle attached to the inside of the metal forceps and connected to a syringe. By grasping the grip, the needle extended 3–5 mm from the tip of the forceps. Moreover, we developed an ultrasonography-guided needle (Figure 1, Nd-U). The tip of the readymade ultrasonographic needle (Model BF-UC180F, Olympus Medical Systems Corporation, Tokyo, Japan) was attached to the hard shaft, and a 23 Gauge needle extended out from the probe. The probe could be curved 90° in either direction to position the probe on the surface of the heart.
Manipulation establishment and safety evaluation
After a double sheath insertion, the rigid endoscopes were advanced through the Sh-S sheath, and forceps or needles were inserted through Sh-L. The pericardial space was inflated with air, as previously reported . Optimization of the devices and maneuvers for a clearer visualization of the pericardial space were performed. Furthermore, a cell transplant demonstration, epicardial pacemaker lead implantation, epicardial ablation, and a safety evaluation of a continuous carbon monoxide supply device were evaluated as described below. Complications such as hemorrhages, arrhythmia, and hemodynamic data were continuously monitored throughout the procedures.
Cell transplant demonstration
To simulate the clinical implications for cell transplantation, 24 μm silicon beads (Catalog No.35-5B, Thermo Scientific, Kanagawa, Japan) were transplanted by Nd-G and Nd-U in 2 canines. The silicon beads were mixed with indocyanine green, a contrast medium for fluoroscopy (Iomeron™, Eizai Co.,Ltd., Tokyo, Japan), and a contrast medium for ultrasonography (Levovist™, Bayer-Schering, Berlin, Germany) to identify the infiltration in the myocardium. A total of 5 ml of the beads medium (105-106 beads/mL) was transplanted by puncturing the myocardium 20 times. After the procedure, the animals were sacrificed and the hearts were evaluated macroscopically and microscopically with a fluorescence filter (Model PB0007, Asahi Spectra Co., Ltd. Tokyo, Japan).
Epicardial treatment for arrhythmia
In cardiac resynchronizing therapy, an epicardial approach with open chest surgery can be considered in cases where appropriate positioning of pacing leads via the coronary sinus is difficult. However, as a minimally invasive alternative, various pacemaker leads were implanted by rigid pericardial endoscopy. The intraventricular pacing lead (Model 5076, Medtronic, Minneapolis, MN, USA), the epicardial temporary pacing lead (Model 6491, Medtronic, Minneapolis, MN, USA) and the epicardial, sutureless, myocardial lead (Model 5071–35, Medtronic, Minneapolis, MN, USA) were used. The intraventricular pacing lead was implanted with a deflectable stylet (Model 6094–58, Medtronic, Minneapolis, MN, USA) and fixed in the left ventricular free wall of the canines. The epicardial temporary pacing lead was transplanted in the right ventricle with a needle holder. Implantation of the epicardial, sutureless, myocardial lead was attempted by screwing the lead into the myocardium through a 14 mm sized sheath in a swine. After implantation, the lead was connected to the stimulator (Model 5348, Medtronic, Minneapolis, MN, USA) and parameters of sensitivity and pacing threshold were measured.
Furthermore, applications for epicardial ablation were evaluated. Epicardial radiofrequency catheter ablation is currently the established, intracardiac approach for incurable arrhythmia . Attempts to obtain a clear visualization of the heart to perform ablation were evaluated in the canines. Complications caused by clamping the pulmonary veins were monitored.
Intrapericardial pressure monitoring
A 60 kg swine was used to examine the relationship between intrapericardial pressure and hemodymanics. After a single pericardiocentesis, the continuous carbon dioxide infusion device (Model UHI-3; Olympus Medical Systems Corporation, Tokyo, Japan) used for pneumoperitoneum during laparoscopic surgery was connected to 8 French sheaths. The intrapericardial pressure, vital signs, fluoroscopy, and the amount of air during the procedure were monitored.
Manipulation of pericardial endoscopic surgery
The manipulation of rigid pericardial endoscopes was intuitive even for cardiologists; however, stretching the pericardium with endoscopes and managing body position were important to clearly visualize the narrow pericardial space.
First, a rigid endoscope was inserted into Sh-S from the epicardial fossa to obtain the antero-posterior caudal view. This enabled us to look up the anterior base of the heart from the apical position. Stretching the pericardium by moving the endoscope like a lever allowed us to enlarge the space between the pericardium and the heart so that a better overview of the wide area of the heart could be obtained. By rotating the shaft of the endoscope, both sides of the lateral wall of the heart were observed. The base of the ventricles, atrial appendages and atriums were confirmed by carefully advancing the endoscopes. To observe the posterior and lateral sides of the heart for the pulmonary veins and the coronary sinuses, the body should be positioned in a recumbent position toward the opposite side of the desired target; for instance, a right lateral recumbent position is required for the left pulmonary veins. The shaft was then rotated and gently moved around the heart along the pericardium.
For a clearer visualization of the narrow pericardial space, air injection was the critical technique required, as described in the previous paper . The “air tamponade” gave us enough space for handling the endoscopes. However, detention of the guide wire in the pericardial space was also valuable in preserving the work space. The stiff guide wires served as struts in the pericardium, similar to pitching a tent. This procedure could reduce the amount of air inside of the pericardial space, which resulted in less risk of hemodynamic deterioration.
The images obtained from rigid pericardial endoscopes were vivid enough to identify not only anatomical structures but also small branches of coronary vessels and details of the heart tissue condition. Therefore, minor surgery was able to be performed as follows.
There are no accepted minimally invasive techniques to transplant cells into the myocardium for tissue engineering therapy; however, we have now established secure and precise transplantation tools.
Pacemaker lead implantation
Epicardial pacemaker leads are used for cardiac resynchronizing therapy when the anatomy of the coronary sinus is not suitable for the maximum pacing benefit, which currently requires open chest surgery. Therefore, we developed a methodology for minimally invasive endoscopic epicardial pacemaker lead implantation.
To demonstrate pericardial lead implantation, the intraventricular pacemaker lead was securely implanted at the base of the lateral wall of the left ventricle (Figure 4E). The endoscope was held to observe the basal free wall of the left ventricle. The lead was delivered by the deflectable stylet and placed at the base of the left ventricle under fluoroscopic guidance. The lead was screwed into the myocardium without fatal arrhythmia, hemodynamic change, or hemorrhage. Pacing from the implanted lead was successfully performed, and there were no signs of dislodging or pacing failures 30 min after the implantation. Furthermore, the epicardial temporary pacing lead was successfully implanted in the right ventricle (Figure 4F). The lead was delivered by a needle holder through Sh-L, and pacing worked properly (threshold 0.1 V/0.4 msec, sense >20 mV). The use of an epicardial sutureless myocardial pacing lead was also evaluated (Figure 4G). Although the implantation was successful, the maneuver was difficult, and the pacing condition was not sufficient (threshold: 5 V/0.4 msec, sense: 4 mV). Because the device itself was designed for open chest surgery, the outer diameter of the lead handle was so large (14 mm) that it was not suitable for this approach. Further modification of the lead designed for a rigid pericardial endoscope should be considered.
For implantation of epicardial pacing leads (Model 4968, Medtronic, Minneapolis, MN, USA), a suture attempt was successfully performed using round needles (3–0 Vicryl) and needle holders in Sh-L. This method could be utilized for ligation of the coronary artery to provide a myocardial infarction model for experimental use (Figure 4H).
Applications for epicardial catheter ablation
Safety of pneumoperitoneum device
Stretching the pericardium with a rigid endoscope could reduce the amount of air in the pericardium that is used for clear visualization. However, filling the pericardial space with air was efficient for this procedure. Similar to laparoscopic surgery, we evaluated the safety of using a continuous pressure-controlled carbon dioxide supply device. The baseline intrapericardial pressure was 2 mmHg, and an additional 3–5 mmHg did not cause any hemodynamic changes. However, an intrapericardial pressure of 6 mmHg induced a sudden systolic pressure drop of 10 mmHg. The fluoroscopy showed a remarkable enlargement of the pericardium, swinging of the heart, and collapse of the left lung. The total amount of supplied air was 700 ml, which suggested that air had leaked into the thoracic cavity and the mediastinum. The pressure drop quickly recovered upon removal of the air. The pressure-controlled air infusion device was not suitable for pericardial endoscopy and required modifications for safe use.
As a promising minimally invasive approach to the pericardial space, the safety of rigid pericardial endoscopy was investigated. With regards to acute complications, arterial blood pressure, and pulse oximetry data were stable during the entire session except for during the use of a continuous air supply device. Other mechanical injuries, such as major bleeding, pneumothorax, and fatal arrhythmia were not identified during these sessions.
Safer, easier, and clearer: The benefits compared to flexible pericardial endoscopes
Comparisons between rigid and flexible pericardial endoscopes
Lateral and Posterior approach
Easily performed by body position change
Difficult but performed with fluoroscopy
Risk of unintentional extraction
Easily controlled with fluoroscopy
Fixed by forceps
Reduced by forceps
One of the complications caused by flexible pericardial endoscopy was the air tamponade. For easier control, we tried to use the pneumoperitoneum device. However, because the device was maintained by pressure control, the amount of air required was large. An air leak could be lethal, inducing pneumomediastinum and pneumothorax. Therefore, stretching the pericardium with the rigid pericardial endoscopes and retaining the stiff guide wire in the pericardium were more effective in establishing a visual field with a reduced dependence on air.
Additional file 1: Video: rigid pericardial endoscopy. A rigid pericardial endoscope was inserted from the epicardial fossa to the pericardial space. Atrial appendages, coronary sinus and pulmonary veins were clearly identified. As the demonstration of cell transplantation, silicon beads, indocyanide green, ultrasonographic and fluoroscopic contrast medium were injected 20 times. Transient ventricular tachycardia did not deteriorate hemodynamic conditions. The use of an ultrasonography-attached needle was successful in avoiding intraventricular injection. For pacemaker implantation, the left ventricular free wall was clearly visualized in the left recombinant position. The intraventricular pacemaker lead was successfully implanted, and the epicardial sutureless lead and the suture trial were attempted. For epicardial ablation, the left pulmonary vein was identified and clamped by rigid forceps for the minimally invasive Maze procedure. (MPEG 14 MB)
Lead implantation using rigid pericardial endoscopy was possible without any major complications. Because the most appropriate site of lead transplantation in cardiac resynchronizing therapy may not be possible via the coronary sinus, a rigid pericardial endoscopic approach could supply another minimally invasive option. Testing 3 kinds of pacing leads gave us enough information to optimize a prototype for this procedure.
Furthermore, in the field of epicardial ablation, pulmonary veins could be clamped by forceps inserted from the epicardial fossa. The same kind of manipulation was previously performed by thoracoscopy . The less invasive Maze procedure could be performed by a rigid pericardial endoscope. Endoscopic-guided epicardial ablation is advantageous in obtaining gross information on the surface of the heart, especially in cases of arrhythmogenic foci located in the epicaridum near a coronary artery or scar, which requires open chest surgery. Further studies for complicated manipulation, such as penetrating the pericardium, and chronic complications, including pulmonary vein stenosis, should be evaluated.
The recent report showed that transmural atrial gene transfer by painting adenoviral vectors to the epicardial surface was successful . This methodology could be applicable to rigid pericardial endoscopy for safer manipulation. The combination of gene therapy and rigid pericardial endoscopy as a focal transgenic method could cure refractory genetic arrhythmias and ionic channel disorders.
Furthermore, the combination of pericardial endoscopy and thoracoscopy under general anesthesia might enable us to perform more complicated maneuvers; however, that procedure would no longer be minimally invasive. Approaching the epicardial space from the esophagus could be a practical application of natural orifice transluminal endoscopic surgery (NOTES) , which could avoid injury to the esophagus during left atrial ablation. Other devices, such as hemostatic devices, suture devices, and epicardial pacemaker lead kits, are now under development. This procedure might be performed after cardiac surgery, and successful adhesiotomy of the pericardium would broaden its indications. We predict that the rigid pericardial endoscopy technique in humans and the broadening of its indications are forthcoming.
The number of animals was too small to evaluate all of the advantages and disadvantages of this procedure. Because swine and canines were used for this experiment, the anatomical differences between the animal species might result in differences in safety and efficacy. All of the protocols were evaluated for acute complications. Further studies were necessary to evaluate chronic complications and the effectiveness of minor surgery. Furthermore, the risk of infection in cell transplantation and follow-up data of lead function in lead implantation should also be evaluated.
Rigid pericardial endoscopy is a promising manipulation procedure for minimally invasive cell transplantation, epicardial lead implantation, and epicardial ablation by providing a safe and clear visual field in the pericardial space.
We thank Olympus Medical Systems Corporation for device development and technical advice.
- Schachinger V, Erbs S, Elsasser A, Haberbosch W, Hambrecht R, Holschermann H, Yu J, Corti R, Mathey DG, Hamm CW: Intracoronary bone marrow-derived progenitor cells in acute myocardial infarction. N Engl J Med. 2006, 355: 1210-1221. 10.1056/NEJMoa060186.View ArticlePubMedGoogle Scholar
- Erbs S, Linke A, Adams V, Lenk K, Thiele H, Diederich KW, Emmrich F, Kluge R, Kendziorra K, Sabri O: Transplantation of blood-derived progenitor cells after recanalization of chronic coronary artery occlusion: first randomized and placebo-controlled study. Circ Res. 2005, 97: 756-762. 10.1161/01.RES.0000185811.71306.8b.View ArticlePubMedGoogle Scholar
- Fischer-Rasokat U, Assmus B, Seeger FH, Honold J, Leistner D, Fichtlscherer S, Schachinger V, Tonn T, Martin H, Dimmeler S, Zeiher AM: A pilot trial to assess potential effects of selective intracoronary bone marrow-derived progenitor cell infusion in patients with nonischemic dilated cardiomyopathy: final 1-year results of the transplantation of progenitor cells and functional regeneration enhancement pilot trial in patients with nonischemic dilated cardiomyopathy. Circ Heart Fail. 2009, 2: 417-423. 10.1161/CIRCHEARTFAILURE.109.855023.View ArticlePubMedGoogle Scholar
- Freyman T, Polin G, Osman H, Crary J, Lu M, Cheng L, Palasis M, Wilensky RL: A quantitative, randomized study evaluating three methods of mesenchymal stem cell delivery following myocardial infarction. Eur Heart J. 2006, 27: 1114-1122.View ArticlePubMedGoogle Scholar
- Hou D, Youssef EA, Brinton TJ, Zhang P, Rogers P, Price ET, Yeung AC, Johnstone BH, Yock PG, March KL: Radiolabeled cell distribution after intramyocardial, intracoronary, and interstitial retrograde coronary venous delivery: implications for current clinical trials. Circulation. 2005, 112: I150-I156.PubMedGoogle Scholar
- Gavira JJ, Perez-Ilzarbe M, Abizanda G, Garcia-Rodriguez A, Orbe J, Paramo JA, Belzunce M, Rabago G, Barba J, Herreros J: A comparison between percutaneous and surgical transplantation of autologous skeletal myoblasts in a swine model of chronic myocardial infarction. Cardiovasc Res. 2006, 71: 744-753. 10.1016/j.cardiores.2006.06.018.View ArticlePubMedGoogle Scholar
- Menasche P, Alfieri O, Janssens S, McKenna W, Reichenspurner H, Trinquart L, Vilquin JT, Marolleau JP, Seymour B, Larghero J: The Myoblast Autologous Grafting in Ischemic Cardiomyopathy (MAGIC) trial: first randomized placebo-controlled study of myoblast transplantation. Circulation. 2008, 117: 1189-1200. 10.1161/CIRCULATIONAHA.107.734103.View ArticlePubMedGoogle Scholar
- Pompilio G, Steinhoff G, Liebold A, Pesce M, Alamanni F, Capogrossi MC, Biglioli P: Direct minimally invasive intramyocardial injection of bone marrow-derived AC133+ stem cells in patients with refractory ischemia: preliminary results. Thorac Cardiovasc Surg. 2008, 56: 71-76. 10.1055/s-2007-989351.View ArticlePubMedGoogle Scholar
- Siminiak T, Fiszer D, Jerzykowska O, Grygielska B, Rozwadowska N, Kalmucki P, Kurpisz M: Percutaneous trans-coronary-venous transplantation of autologous skeletal myoblasts in the treatment of post-infarction myocardial contractility impairment: the POZNAN trial. Eur Heart J. 2005, 26: 1188-1195. 10.1093/eurheartj/ehi159.View ArticlePubMedGoogle Scholar
- Gavira JJ, Nasarre E, Abizanda G, Perez-Ilzarbe M, de Martino-Rodriguez A, de Jalon JA G, Mazo M, Macias A, Garcia-Bolao I, Pelacho B: Repeated implantation of skeletal myoblast in a swine model of chronic myocardial infarction. Eur Heart J. 2010, 31: 1013-1021. 10.1093/eurheartj/ehp342.View ArticlePubMedGoogle Scholar
- Psaltis PJ, Carbone A, Nelson AJ, Lau DH, Jantzen T, Manavis J, Williams K, Itescu S, Sanders P, Gronthos S: Reparative effects of allogeneic mesenchymal precursor cells delivered transendocardially in experimental nonischemic cardiomyopathy. JACC Cardiovasc Interv. 2010, 3: 974-983. 10.1016/j.jcin.2010.05.016.View ArticlePubMedGoogle Scholar
- Dib N, Dinsmore J, Lababidi Z, White B, Moravec S, Campbell A, Rosenbaum A, Seyedmadani K, Jaber WA, Rizenhour CS, Diethrich E: One-year follow-up of feasibility and safety of the first U.S., randomized, controlled study using 3-dimensional guided catheter-based delivery of autologous skeletal myoblasts for ischemic cardiomyopathy (CAuSMIC study). JACC Cardiovasc Interv. 2009, 2: 9-16. 10.1016/j.jcin.2008.11.003.View ArticlePubMedGoogle Scholar
- Leon MB, Kornowski R, Downey WE, Weisz G, Baim DS, Bonow RO, Hendel RC, Cohen DJ, Gervino E, Laham R: A blinded, randomized, placebo-controlled trial of percutaneous laser myocardial revascularization to improve angina symptoms in patients with severe coronary disease. J Am Coll Cardiol. 2005, 46: 1812-1819. 10.1016/j.jacc.2005.06.079.View ArticlePubMedGoogle Scholar
- Vale PR, Losordo DW, Milliken CE, Maysky M, Esakof DD, Symes JF, Isner JM: Left ventricular electromechanical mapping to assess efficacy of phVEGF(165) gene transfer for therapeutic angiogenesis in chronic myocardial ischemia. Circulation. 2000, 102: 965-974. 10.1161/01.CIR.102.9.965.View ArticlePubMedGoogle Scholar
- Perin EC, Dohmann HF, Borojevic R, Silva SA, Sousa AL, Silva GV, Mesquita CT, Belem L, Vaughn WK, Rangel FO: Improved exercise capacity and ischemia 6 and 12 months after transendocardial injection of autologous bone marrow mononuclear cells for ischemic cardiomyopathy. Circulation. 2004, 110: II213-II218.View ArticlePubMedGoogle Scholar
- Kornowski R, Hong MK, Gepstein L, Goldstein S, Ellahham S, Ben-Haim SA, Leon MB: Preliminary animal and clinical experiences using an electromechanical endocardial mapping procedure to distinguish infarcted from healthy myocardium. Circulation. 1998, 98: 1116-1124. 10.1161/01.CIR.98.11.1116.View ArticlePubMedGoogle Scholar
- Psaltis PJ, Carbone A, Nelson A, Lau DH, Manavis J, Finnie J, Teo KS, Mackenzie L, Sanders P, Gronthos S: An ovine model of toxic, nonischemic cardiomyopathy–assessment by cardiac magnetic resonance imaging. J Card Fail. 2008, 14: 785-795. 10.1016/j.cardfail.2008.06.449.View ArticlePubMedGoogle Scholar
- Vulliet PR, Greeley M, Halloran SM, MacDonald KA, Kittleson MD: Intra-coronary arterial injection of mesenchymal stromal cells and microinfarction in dogs. Lancet. 2004, 363: 783-784. 10.1016/S0140-6736(04)15695-X.View ArticlePubMedGoogle Scholar
- Kimura T, Miyoshi S, Takatsuki S, Tanimoto K, Fukumoto K, Soejima K, Fukuda K: Safety and efficacy of pericardial endoscopy by percutaneous subxyphoid approach in swine heart in vivo. J Thorac Cardiovasc Surg. 2011, 142: 181-190. 10.1016/j.jtcvs.2010.09.050.View ArticlePubMedGoogle Scholar
- Sacher F, Roberts-Thomson K, Maury P, Tedrow U, Nault I, Steven D, Hocini M, Koplan B, Leroux L, Derval N: Epicardial ventricular tachycardia ablation a multicenter safety study. J Am Coll Cardiol. 2010, 55: 2366-2372. 10.1016/j.jacc.2009.10.084.View ArticlePubMedGoogle Scholar
- Haissaguerre M, Jais P, Shah DC, Takahashi A, Hocini M, Quiniou G, Garrigue S, Le Mouroux A, Le Metayer P, Clementy J: Spontaneous initiation of atrial fibrillation by ectopic beats originating in the pulmonary veins. N Engl J Med. 1998, 339: 659-666. 10.1056/NEJM199809033391003.View ArticlePubMedGoogle Scholar
- Hamdi H, Planat-Benard V, Bel A, Puymirat E, Geha R, Pidial L, Nematalla H, Bellamy V, Bouaziz P, Peyrard S: Epicardial adipose stem cell sheets results in greater post-infarction survival than intramyocardial injections. Cardiovasc Res. 2011, 91: 483-491. 10.1093/cvr/cvr099.View ArticlePubMedGoogle Scholar
- Gerosa G, Bianco R, Buja G, di Marco F: Totally endoscopic robotic-guided pulmonary veins ablation: an alternative method for the treatment of atrial fibrillation. Eur J Cardiothorac Surg. 2004, 26: 450-452. 10.1016/j.ejcts.2004.04.043.View ArticlePubMedGoogle Scholar
- Kikuchi K, McDonald AD, Sasano T, Donahue JK: Targeted modification of atrial electrophysiology by homogeneous transmural atrial gene transfer. Circulation. 2005, 111: 264-270. 10.1161/01.CIR.0000153338.47507.83.View ArticlePubMedGoogle Scholar
- Fritscher-Ravens A, Patel K, Ghanbari A, Kahle E, von Herbay A, Fritscher T, Niemann H, Koehler P: Natural orifice transluminal endoscopic surgery (NOTES) in the mediastinum: long-term survival animal experiments in transesophageal access, including minor surgical procedures. Endoscopy. 2007, 39: 870-875. 10.1055/s-2007-966907.View ArticlePubMedGoogle Scholar