The successful introduction of an adapted form of the mini extra corporeal circulation used for cardiac surgery in an obese patient
© Sartini et al; licensee BioMed Central Ltd. 2012
Received: 4 January 2012
Accepted: 9 March 2012
Published: 9 March 2012
Obese patients undergoing cardiac surgery have been shown to have a high risk of developing postoperative complications, specifically: increased length of hospital stay, readmission to intensive care unit, acute renal failure, deep sternal wound infections and new onset of atrial fibrillation.
A custom-made circuit was created to allow the use of Mini Extra Corporeal Circulation (MECC) but permitting the switch to a closed siphon drainage system in the case of difficulties.
To limit artificial surface contact a small oxygenating device (Admiral, Eurosets) was employed in spite of the patients size. This adapted circuit permits a feasible and safer approach to using MECC. This report suggests that smaller oxygenators could be integrated into clinical practice in an adult MECC configuration, even for more obese patients, limiting artificial surface contact.
KeywordsCardiopulmonary bypass Obese patients Aortic valve replacement Mini extra corporeal circulation
Obese patients undergoing cardiac surgery have an increased risk of developing specific postoperative complications. Obesity (defined as BMI > 30) is associated with prolonged mechanical ventilation, readmission to intensive care unit, increased rates of deep sternal wound infection and a length of stay greater than 14 days. Both obesity and morbid obesity, especially when other risk factors such as diabetes mellitus and hypertension are present, may increase the rate of onset of acute renal failure. Nevertheless, an increased BMI does not increase the risk of early postoperative death or the risk of postoperative cerebrovascular accidents and the rates of postoperative bleeding in obese patients are significantly lower . The profile of patients undergoing cardiac surgery has changed in recent times, as rates of obesity increase, the proportion of obese patients requiring cardiac surgery has increased also.
The aim of this article is to propose a progressive approach to MECC, demonstrating its successful use in an obese patient.
A 73 year-old man was admitted to our unit with severe aortic stenosis requiring aortic valve replacement. The patient had numerous risk factors for cardiovascular disease including obesity (BMI = 40; BSA = 2,46), hypercholesterolemia, hypertriglyceridemia, Type II Diabetes Mellitus and being a cigarette smoker. The procedure was performed through a J-shaped mini-sternotomy  with routine ascending aorta and right atrial cannulation. After opening the aorta, we carefully removed all the debris on the valvular annulus that extended to the mitro-aortic junction. A tissue valve was then implanted. The operation was completed routinely and without incident. The patient was discharged from the Intensive Care Unit on post-operative day two. There were no major complications.
Cardio Pulmonary Bypass (CPB) was performed using a closed custom-made circuit designed to perform Mini-extracorporeal circulation (MECC). The circuit was designed such that the system could switch to a conventional closed siphon drainage system if necessary.
This dual purpose circuit employed a 1/2 inch venous pipe longer than the ones usually used in MECC and the same pipe was selected for use as the venous cannula suitable for siphon drainage (24 fr Aortic cannula; 44/36 fr right atrium cannula. DLP® Medtronic, Inc.). To minimise artificial surface contact and improve biocompatibility we employed the Admiral® oxygenator (Eurosets, Mirandola, Italy), which had the smallest available membrane and phosphorylcoline treatment of the lines and oxygenator. It is an integrated microporous hollow-fibre oxygenator, with a total membrane surface area of 1.35 m2, a heat exchanger surface area of 0,08 m2 and a low prime volume (190 ml) . The oxygenator and the circuit had a superficial phosphorylcoline (Pc) coating. This coating is reported to provide some protection against blood clot activation . The circuit was placed under the operating table to obtain a drainage pressure of approximately -25, mmHg. A pressure transducer monitored the negative pressure in venous line. Retrograde Autologous Priming (RAP) was employed to limit haemodilution .
Buckberg cardioplegic solution was delivered directly into the coronary ostia and into the coronary sinus during aortotomy. Mild hypothermia was induced. An auto-transfusion device (Electa, Soring Group, Italy) was used to collect, wash and store shed red blood cells, the remaining circuit perfusate was also treated by Electra and autologous red cell transfusion was initiated in the post-operative period . Hyperglycaemia was managed according to the modified Portland protocol.
Oxygenator performance was evaluated by monitoring of serial arterial blood gas measurments (pO2 150-200 mmHg). Adequacy of perfusion was monitored by Oxygen Delivery (DO2), lactate production and the ratio between the oxygen delivery and carbon dioxide production (DO2/VCO2). The DO2 formula can be calculated as follows: [pump flow × (Hb × O2Sat × 1,36) + 0,003 × PaO2 ] . Blood lactate monitoring is more appropriate for detecting the correct matching of oxygen supply and demand during CPB .
Carbon dioxide production (VCO2) was calculated as follows: VCO2 indexed (mL × min -1 × m-2): eCO2 (mmHg) × Ve (L/min) × 1000/760 × BSA (m2). Exhaled carbon dioxide from oxygenator (CO2) and gas flow into the oxygenator (Ve) were simultaneously recorded at each sampling aspiration.
Conduct of perfusion
Tubing clamps were placed in both A and C 3/8 inch pipes, starting from the venous bag. The centrifugal pump velocity was previously set to overcome the patient's systemic blood pressure avoiding back flow. At the onset of CPB the clamp in position A was partially opened obtaining drainage from the patient into a venous bag. Once the partial release of the "A segment clamp" permitted the venous drainage, the blood collected in the collapsible bag began to flow from the centrifugal pump and gradually increased with further release of the "A clamp segment". With this clamp configuration siphon drainage was obtained as it occurs in conventional closed system. In few minutes full CPB was established. At this point, in order to switch toward MECC, the "C segment clamp" was released and subsequently the "A segment" was clamped.
At this moment there was a transient diminution of systemic perfusion due to volume collection in the collapsible bag but this was readily corrected with light occlusion of the "C segment" until the collapsible bag emptied returning perfusate, restoring circulating volume.
Once the circulating volume correction was performed the desired flow was obtained using MECC conduction. In this kinetic assisted drainage configuration with a closed circuit the monitoring of central venous pressure (CVP) was helpful in the maintenance of desired blood flow and avoided over-distension of the right atrium. Excessive negative pressure and venous line chattering was overcome by partial clamping of the 1/2 inch venous line. The negative pressure in 1/2 inch venous line was kept to approximately -50 mmHg.
MECC optimizes the extracorporeal circulation, thus resulting in a reduction in the inflammatory reaction, blood exposure and trauma . The use of MECC as demonstrated in this study could be incorporated into daily practice without considerable difficulty at other centres  as this centre has done following the experience of others.
The use of the venous bag included in the circuit permits a rapid and safe switch from MECC to a closed conventional extracorporeal circulation. The oxygenator (Eurosets, Mirandola, Italy) performed satisfactorily throughout the procedure despite the small oxygenating surface . To our knowledge this case represents the patient with the largest body mass index to have been treated with this small oxygenating device.
The intra-operative [Figure 3] lactate production remained within a normal range throughout the procedure, providing reassurance about the adopted perfusion strategy, which permitted a prudent transfusion policy.
With this simple circuit the surgical team can gain the necessary confidence in MECC and experience can be gained safely. The integration of a smaller oxygenator than usual in a MECC circuit, could be a positive contribution in the continuous effort to increase perfusion optimization [16–19] and manufacturing companies could consider this for their own commercial development. The use of MECC in this way has been shown in this case to be successful in the obese patient.
Written informed consent was obtained from the patient for publication of this report and any accompanying images.
The authors do not receive honoraria, consultation fees or support from cited companies.
- Yap C-H, Zimmet A, Mohajeri M, Yii M: Effect of obesity on early morbidity following cardiac surgery. Heart Lung Circ. 2007, 16: 31-36. 10.1016/j.hlc.2006.09.007.View ArticlePubMedGoogle Scholar
- Perrotta S, Lentini S: Ministernotomy approach for surgery of the aortic root and ascending aorta. Interact Cardiovasc Thorac Surg. 2009, 9 (5): 849-858. 10.1510/icvts.2009.206904.View ArticlePubMedGoogle Scholar
- De Stefano E, Delay D, Horisberger H, von Segesser LK: Initial clinical experience with the Admiral oxygenator combined with separated suction. Perfusion. 2008, 23: 209-213. 10.1177/0267659108100467.View ArticlePubMedGoogle Scholar
- Sohn N, Marcoux J, Mycyk T, Krahn J, Meng QH, Gourlay T, Shedden Laurie: The impact of different biocompatible coated cardiopulmonary bypass circuits on inflammatory response and oxidative stress. Perfusion. 2010, 25: 427-10.1177/0267659110381453.View ArticleGoogle Scholar
- Durandy Y: Perfusionist strategies for blood conservation in pediatric cardiac surgery. World J Cardiol. 2010, 2 (2): 27-33. 10.4330/wjc.v2.i2.27.View ArticlePubMedPubMed CentralGoogle Scholar
- Slight RD, Lux D, Nzewi OC, McClelland DBL, Mankad PS: Oxygen delivery and hemoglobing concentration in cardiac surgery: When do we have Enough?. Artif Organs. 2008, Wiley Periodicals, Inc, 32 (12): 949-955. 10.1111/j.1525-1594.2008.00685.x.Google Scholar
- Scinde SB, Golam KK, Kumar P, Patil ND: Blood lactate levels during cardiopulmonary bypass. Ann Card Anaesth. 2005, 8: 39-44.Google Scholar
- Edmunds H: Inflammatory response to Cardiopulmonary Bypass. Ann Thorac Surg. 1998, 66: S12-S16. 10.1016/S0003-4975(98)00967-9.View ArticlePubMedGoogle Scholar
- Curtis N, Vohra HA, Ohri SK: Mini extracorporeal circuit cardiopulmonary bypass system: a review. Perfusion. 2010, 25: 115-10.1177/0267659110371705.View ArticlePubMedGoogle Scholar
- Issit R, Cumberland T, Clemens A, Mulholland J: Clinical evaluation of the Admiral 1,35 m2 hollow-fibre membrane oxygenator. Perfusion. 2008, 23: 33-38. 10.1177/0267659108093880.View ArticleGoogle Scholar
- Parolari A, Alamanni F, Gherli T, Bertera A, Dainese L, Costa C, Schena M, Sisillo E, Spirito R, Porqueddu M: Biglioli, Cardiopulmonary bypass and oxygen consumption: oxygen delivery and hemodynamics. Ann Thorac Surg. 1999, 67: 1320-1327. 10.1016/S0003-4975(99)00261-1.View ArticlePubMedGoogle Scholar
- Ranucci M, Romitti F, Isgro G, Cotza M, Brozzi S, Boncilli A, Ditta A: Oxygen delivery during cardiopulmonary bypass and acute renal failure after coronary operation. Ann Thorac Surg. 2005, 80: 2213-2220. 10.1016/j.athoracsur.2005.05.069.View ArticlePubMedGoogle Scholar
- Myers GJ, Morrison S, Merchant AK, Johnstone DR, Hawkins J, Swyer W, Gardner K: The significance of Artero-Venous Carbon Dioxide Gradient. A Review of this Guide to the Adequacy of Peripheral Tissue Perfusion. Canadian Perfusion Canadienne. 2008, 18 (2): 62-64.Google Scholar
- Ranucci M, Isgrò G, Romitti F, Mele S, Biagioli B, Giomarelli Pierpaolo: Anaerobic Metabolism During Cardiopulmonary Bypass: Predictive Value of Carbon Dioxide Derived Parameters. Ann Thorac Surg. 2006, 81: 2189-2195. 10.1016/j.athoracsur.2006.01.025.View ArticlePubMedGoogle Scholar
- Ranucci M, Biagioli B, Scolletta S, Grillone G, Cazzaniga A, Cattabriga I, Isgrò G, Giomarelli P: Lowest hematocrit on cardiopulmonary bypass impairs the outcome in coronary surgery. Tex Heart Inst J. 2006, 33: 300-305.PubMedPubMed CentralGoogle Scholar
- Shann KG, Likosky DS, Murkin JM, Baker RA, Baribeau JR, DeFoe GR, Dickinson TA, Gardner TJ, Grocott HP, O'Connor GT, Rosinski DJ, Selle FW, Willcox TW: An evidence-based review of the practice of cardiopulmonary bypass in adults: a focus on neurological injury, glicaemic control, haemodilution, and inflammatory response. J Thorac Cardiovasc Surg. 2006, 132: 283-290. 10.1016/j.jtcvs.2006.03.027.View ArticlePubMedGoogle Scholar
- Palanzo D, Qiu F, Baer L, Clark JB, Myers JL, Undar A: Evolution of the extracorporeal life support circuitry. Artificial Organs. 2010, Wiley Periodicals, Inc, 34 (11): 869-873. 10.1111/j.1525-1594.2010.01127.x.Google Scholar
- Murphy GS, Hessel EA, Groom R: Optimal perfusion during cardiopulmonary bypass: an evidente-based approach. Anesth Analg. 2009, 108: 1394-1417. 10.1213/ane.0b013e3181875e2e.View ArticlePubMedGoogle Scholar
- Vohra HA, Whistance R, Modi A, Ohri SK: The inflammatory response to miniaturized extracorporeal circulation: a review of the literature. Mediators Inflamm. 2009: 7-Article ID 707042Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.