Early experience with ovation endograft system in abdominal aortic disease
- Giovanni Nano†1,
- Daniela Mazzaccaro†1Email author,
- Silvia Stegher1,
- Maria Teresa Occhiuto1,
- Giovanni Malacrida1,
- Domenico G Tealdi1,
- Antonino Alberti2 and
- Pietro Volpe2
© Nano et al.; licensee BioMed Central Ltd. 2014
Received: 19 November 2013
Accepted: 4 March 2014
Published: 12 March 2014
We describe our initial experience with the use of the TriVascular Ovation endograft system for the treatment of abdominal aortic aneurysms (AAA).
We retrospectively reviewed data from patients treated for AAA using the Ovation endograft at two institutions from January 2011 to September 2012. Main outcomes included primary success, survival, complications, and device-related events. The mean follow-up period was 10 months (range 1–22 months).
Thirty-seven patients (male: 95%, mean age: 76 yr) were treated for AAA (mean diameter: 54 mm) with the Ovation endograft. Local or regional anesthesia was used in 86.5% of cases. Percutaneous access was utilized in 73% of cases. Primary success was 89.2% (33/37). Four adjunctive procedures were required including two distal extensions (type 1b endoleak and iliac limb disconnection resulting in type III endoleak) and two bypass surgeries (limb graft occlusion and gate cannulation failure). No deaths or major complications were reported during the procedure or in follow-up. No type I, III, or IV endoleak, AAA enlargement, AAA rupture, stent fracture, migration, or endovascular or surgical reintervention were reported during the follow-up period. Type II endoleak was observed in two patients. Asymptomatic narrowing of both iliac limbs was observed in one patient at 6 months.
Our initial experience with the Ovation endograft demonstrated encouraging results in patients with AAA.
KeywordsAbdominal aortic aneurysm Endovascular aneurysm repair Ovation Stent graft
The first endovascular repair of an abdominal aortic aneurysm (AAA) was performed in 1990 by Dr. Parodi , who implanted a large straight endograft into the abdominal aorta. Limitations of the use of a similar device soon became evident and new self-expandable materials and bifurcated stent-grafts, which had a smaller caliber than previous generations, rapidly became available. The latest generation of ultra-low profile endografts currently allows the treatment of AAAs in patients who have previously been excluded because of challenging aortic anatomies and small access vessels.
However, some have expressed concern that reducing the profile of a stent graft may compromise sealing. The search for an endograft with a smaller profile with no compromise on performance and durability of the device remains a therapeutic challenge. After having received the CE Mark on September 17th 2010, the low-profile TriVascular Ovation system (14 F) has been recently introduced also to our institutions for use in selected patients with relatively challenging necks or smaller iliac vessels. The purpose of this study was to retrospectively review our early experience with this novel stent graft.
The research described in this manuscript was approved by the Institutional Review Board at each respective institution and was conducted per the guidance set forth in the Declaration of Helsinki.
We retrospectively reviewed data of all patients who were treated for AAA using the Ovation endograft system from January 2011 to September 2012 in our two centres.
Patients who meet specific anatomic criteria of both proximal and distal landing zone (non-aneurysmal proximal aortic neck length of at least 7 mm, proximal neck luminal diameter between 16 and 30 mm, non-aneurysmal distal iliac artery length of at least 10 mm, distal iliac luminal diameters between 8 and 20 mm, possibility to preserve patency in at least one hypogastric artery, distance from the most distal renal artery to most superior internal iliac artery measurement of at least 13 cm) were considered suitable for the implantation of the Ovation Abdominal Stent Graft.
One patient was treated outside the IFU of the Ovation Stent Graft, with a pre-planned “chimney-technique” because of AAA involving both renal arteries. Due to small iliac access, however, it was not possible to treat the patient using a different type of endograft.
Patients with a dissecting or acutely ruptured aneurysm were excluded, as well as patients with history of connective tissue disease (e.g., Marfan's or Ehler's-Danlos syndrome) or with a known allergy or intolerance to polytetrafluorethylene (PTFE), PEG-based polymers, fluorinated ethylene propylene (FEP) or nitinol.
After treatment all patients were discharged on antiplatelet therapy (either ASA 100 mg daily or Clopidogrel 75 mg daily all lifelong) unless they were already on anticoagulant therapy for preoperative comorbidities.
Ovation stent graft
The polymer is radiopaque, allowing the filling of the graft to be observed under fluoroscopy. Once the main body of the graft is positioned and opened, the polymer is prepared on the operating table, and mixed in a syringe that will be joined to the graft introducer. The syringe itself is joined to an autoinjector which pushes the polymer through the ring-shaped channels in 20 minutes, during which it is possible to complete the placement of the contralateral iliac extension. When the time of the polymer’s filling is elapsed, the syringe and the autoinjector are removed together and it is possible to place the eventual ipsilateral iliac extension.
Because of the ultra-low profile, the Ovation system can be used in patients with iliac or femoral artery access of less than 7 mm. The Ovation device is contraindicated in patients who have allergies to the device materials, as well as in patients who are unable to undergo the necessary preoperative and postoperative imaging studies.
Patient records were reviewed for demographics, medical history, and aortoiliac morphology via computed tomography angiography. Procedural data included fluoroscopy time, operation time, amount of procedural contrast medium, blood loss, and complications. Follow-up data were obtained from outpatient visits and CT scan or duplex scans. Imaging was performed every 3 months during the first year after the operation and annually thereafter using mainly duplex scans. A CT scan was regularly performed at 1 year and in any case of inconclusive duplex scan (i.e. hostile abdomen) or finding of sac enlargement/graft occlusion/stenosis/endoleaks at duplex scan. Outcomes were analyzed to evaluate primary success, survival, complications, and device-related events. Primary success was defined as successful access, delivery, and implant of the endograft with absence of surgical conversion, mortality, type I or III endoleak, or graft limb obstruction.
Statistical analysis was performed using JMP® 5.1.2 (SAS Institute, Inc., Cary, NC, USA). Continuous variables are reported as mean ± SD and categorical variables are presented as n (%). Logistic regression using the Wald statistic was performed to identify predictors of adjunctive procedures. P values < 0.05 were considered statistically significant.
Baseline patient characteristics
Patients’ characteristics and anatomical data
n = 37
Mean age, years (Range)
Current or previous smoke
Anatomical data (Mean ± SD), mm
Proximal aortic neck diameter
25.9 ± 0.8
Proximal aortic neck lenght
18 ± 2.1
Proximal aortic neck angulation
52.2° ± 4.2°
Distal aortic neck diameter
27.7 ± 0.4
Distal aortic neck lenght
24 ± 1.2
12.3 ± 0.2
14.2 ± 0.5
14.2 ± 0.4
13.9 ± 0.7
8.1 ± 0.7
Sac diameter (range)
Intraprocedural and In-hospital data (Mean ± SD)
n = 37
Local + conscious sedation
Percutaneous brachial (additional)
Time of operation (min)
43.1 ± 3.2
Amount of contrast (cc)
28.3 ± 2.5
Fluoroscopy time (min)
11.3 ± 1.5
Blood loss (cc)
110 ± 16
Length of stay (days)
3.6 ± 0.9
Major adverse events
Major adverse events
Endoleak type II
Asintomatic narrowing of iliac limbs
An additional percutaneous brachial access was performed in 2 patients in which a “Chimney Technique” was used. In one of these cases, the AAA involved the origin of both renal arteries while in the remaining case the chimney graft was used as a bail-out technique, allowing a safe proximal landing zone without covering the right renal artery. In both patients, a Viabahn® covered stent was advanced antegrade through brachial access. The proximal rings of the Ovation graft were inflated while a balloon was kept inflated within the covered stent.
All grafts were oversized less than 20%; in both cases in which a “Chimney Technique” was used, the graft had a 10% oversize.
Closure of the access site was obtained with manual compression bilaterally in 2 out of 27 cases of percutaneous femoral access. A vascular closure device was used in the remaining cases of percutaneous femoral access.
Primary success was achieved in 33 cases (89.2%). Four patients required adjunctive procedures during the operation. In one case, given the twist of the device, which opened incompletely at the gate, we were not able to canulate the contralateral leg. Therefore, although a good proximal seal had been obtained, a conical stent graft (Talent, Medtronic) was placed, with an occluder at the level of the left common iliac artery, and a femoro-femoral right-to-left bypass was performed. In the second case, intraoperative graft occlusion of the right leg required an additional femoro-femoral left-to-right bypass. In the third case, the device failed despite proper deployment with enough overlap. Disconnection between the main body and the left leg occurred, causing an intraprocedural type III endoleak which was immediately corrected with a second iliac extension. In the fourth case, a type Ib endoleak at the end of the procedure required the placement of a distal iliac extension.
Logistic regression predicting risk of adjunctive procedures from aortoiliac characteristics
Proximal aortic neck diameter
2.36 - 24.17
Proximal aortic neck length
0.21 – 1.80
Distal aortic neck diameter
0.36 – 4.82
Distal aortic neck lenght
0.73 – 5.16
0.01 – 2.23
0.67 – 4.11
0.87 – 3.92
Distal LCIA diameter
0.09 – 3.2
0.58 – 3.64
0.77 – 2.14
0.42 – 1.23
Safety and effectiveness outcomes
Since Parodi’s initial experience , the need to have appropriate vascular access to deliver the aortic stent graft in EVAR has emerged as a main issue. As the endovascular experience grew, it soon became clear that when endovascular therapy was indicated for clinical criteria, it was often not feasible for anatomical reasons. The introduction of self-expandable materials permitted an extension of the method even in those patients who presented with small iliac-femoral access. Nevertheless, in the common practice, access site related complications are still observed, such as dissections, arterial ruptures and haematomas in case of percutaneous approaches. Reduced access site complications and lower overall procedural mortality rates after EVAR have been recently reported with the use of new lower-profile endograft systems [2–5].
The benefits of lower-profile devices are obvious. Not only is the risk of access site complications lower with smaller profile systems, but more patients with smaller access vessels can also be treated. However the downside of reducing the profile is the possible loss of proximal sealing, which can compromise the overall performance and durability of the device. For this purpose, not only were different anchoring systems introduced, but also prostheses with large free-flow. The theoretical assumption concerned the axiom that good sealing without migration meant absence of endoleak. It was realized that this axiom was true in part and that even in presence of graft migration sometimes there were no endoleaks.
The Ovation stent addresses these dual problems by combining a large free-flow with hooks with an inflatable ring begins 10 mm below the top of the covered part of the graft. The graft is fixed and the ring provides sealing to the arterial wall. By eliminating the need for a metallic endoframe entirely, the Ovation can keep a lower profile while achieving a good proximal sealing and likely a good durability over time.
In this sense our results are consistent with the purpose of the endograft since in the long-term we have not seen neither type I endoleaks nor graft migrations.
In our series, we observed some intraprocedural complications, all of which were successfully corrected with either a graft extension or fem-fem bypass surgery. On one hand, the reduction of the profile allowed the treatment of cases that would have been difficult or impractical to treat with other devices. On the other side, the reduction of the prosthesic structure might have increased the intraprocedural difficulties and also made cannulation of the gate more difficult, which in a patient of our case series led to a fem-fem bypass surgery.
Speculating on graft modifications is beyond the scope of our study, however industries should address improvements of the graft considering real life experiences.
To some degree, the problems associated with cannulation have been addressed by the Ovation PRIME, the newest iteration of the endoprosthesis, which received CE Mark in June, 2012. In this version of the device, the contralateral leg is connected to the delivery system via a trigger wire, which is designed to prevent the twisting that caused the failure to cannulate described previously.
Albeit the experience is only at the beginning, the results reported in our series are encouraging, consistent with those reported in the Ovation pre-market clinical trial  in terms of aneurysm exclusion and prevention of aortic rupture.
Metha et al.  recently published a prospective, multicenter, single-arm trial to evaluate the one-year results about safety and effectiveness of the Ovation stent graft in 161 patients with AAAs. They reported a 30-day major adverse event rate of 2.5%, while at 1 year, AAA-related and all-cause mortality were 0.6% and 2.5%, respectively. Similarly to our results, no stent graft migration, no AAA rupture or type I, III, or IV endoleaks were reported in the long term but type II endoleaks were identified in 34% of patients and AAA-related secondary procedures were performed in 10 patients (6.2%) for 6 endoleaks, 3 aortic main body stenosis and 3 iliac limb stenosis or occlusion.
Similar outcomes are expected in UE post-marketing study. Moreover, the ongoing prospective non-randomized multi-center pre-market clinical trial  will probably build further understanding on the use of this device and its durability over time.
Our initial experience with the Ovation endograft demonstrated encouraging early and long-term results in patients with AAA in terms of safety and effectiveness.
Abdominal aortic aneurysms
Fluorinated ethylene propylene
Intensive care unit
Endovascular aortic repair.
The authors thank Larry Miller, PhD for providing editorial assistance and critical review.
- Parodi JC, Palmaz JC, Barone HD: Transfemoral intraluminal graft implantation for abdominal aortic aneurysms. Ann Vasc Surg. 1991, 5: 491-499. 10.1007/BF02015271.View ArticlePubMed
- Verhoeven BA, Waasdorp EJ, Gorrepati ML, van Herwaarden JA, Vos JA, Wille J, Moll FL, Zarins CK, de Vries JP: Long-term results of Talent endografts for endovascular abdominal aortic aneurysm repair. J Vasc Surg. 2011, 53: 293-298. 10.1016/j.jvs.2010.08.078.View ArticlePubMed
- Bersin RM: The Evolution of Low-Profile Endograft Design. 2011, Evar Update: Endovascular Today, 54-58.
- Moulakakis KG, Dalainas I, Kakisis J, Giannakopolous TG, Liapis CD: Current knowledge on EVAR with the ultra-low profile Ovation Abdominal Stent-Graft System. J Cardiovasc Surg. 2012, 53: 427-432.
- Criado FJ: The EVAR Landscape in 2011. 2011, Evar Update: Endovascular Today, 40-44.
- Presentation at the LINC 2012 (Leipzig International Course) by Dr. Dierk Scheinert: Featured research: Ovation European prospective multi-center study – 1 year data. 2012, Leipzig, January 25–28, 2012. Accessed from http://www.leipzig-interventional-course.com/index.php?option=com_content&task=view&id=183&Itemid=286 on October 23rd
- Mehta M, Valdés FE, Nolte T, Mishkel GJ, Jordan WD, Gray B, Eskandari MK, Botti C: A Pivotal Clinical Study to Evaluate the Safety and Effectiveness of the Ovation Abdominal Stent Graft System” Investigators. One-year outcomes from an international study of the Ovation Abdominal Stent Graft System for endovascular aneurysm repair. J Vasc Surg. 2014, 59 (1): 65-73. 10.1016/j.jvs.2013.06.065. e3View ArticlePubMed
- Clinical Study of the TriVascular Ovation Abdominal Stent Graft System. From http://clinicaltrials.gov/ct2/show/NCT01092117. Accessed on October 23rd, 2012
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.