- Letters to the Editor
- Open Access
- Open Peer Review
Migration of superior vena cava stent
Journal of Cardiothoracic Surgery volume 3, Article number: 12 (2008)
There has been a recent increase in the use of endovascular prostheses resulting in complex surgical and interventional complications not previously recognised. We report a case of Superior vena cava stenosis treated with a wallstent which migrated to the right atrium, necessitating a combined radiological and surgical approach to retrieve it.
There has been a recent increase in the use of endovascular prostheses resulting in complex surgical and interventional complications not previously recognised. We report a case of Superior Vena Cava Stenosis treated with a Wallstent which migrated to the right atrium, necessitating a combined radiological and surgical approach to retrieve it.
Illustrative case history
A 78 year old haemodialysis patient was admitted under the care of renal team with a history of symptomatic bilateral upper limb and facial swelling unrelated to her renal impairment. In the past, her right groin had been irradiated for lymphoma and she had significant co-morbidity precluding any surgical intervention. She had duplex scanning, which showed evidence of Superior Vena Caval obstruction. She underwent angioplasty and Stenting of the Superior Vena Cava with a 10 mm × 46 mm Wallstent Uni endoprosthesis (Boston Scientific Corp., MA, USA). A right Internal Jugular Vascath was inserted proximal to the stent and good flows were achieved from all three lumen.
Two weeks later since her supracaval venous access became blocked and a tunnelled Permacath was placed in the left femoral vein. An Amplatz superstiff wire was placed across the SVC stent through the pre-existing vascath. Under fluoroscopy, the line was removed and a peel-away sheath was introduced over the wire, to facilitate the introduction of a tunnelled Split-Cath. The intersect between the two different length lumina caught on the superior margin of the SVC stent instantly dislodging it. The stent moved into and lay horizontally in the right atrium [Fig 1, 2].
A 45 cm 10 Fr SuperArrow-flex percutaneous sheath was inserted in the left femoral vein, immediately superior to the tunnelled split-cath. Through this a 25 mm Amplatz gooseneck snare (ev3 Inc, MI, USA) was inserted and several futile attempts were made to snare the stent. A sidewinder catheter (Cordis Europa N.V) over an angled Terumo wire (Terumo Corp, Japan) was used to access the lumen of the stent. The Terumo wire, having passed through the length of the stent, was then snared through the gooseneck [Fig 3]. The snare was pulled into the sheath and a loop consisting of the sidewinder and Terumo wire was tightened, shortening the stent along its length. The constrained stent was pulled down into the left groin, the sheath and the sidewinder catheter were removed and the loop of the wire was held externally with a surgical artery clamp [Fig 4].
Since the left femoral access was precious and the radiological options were exhausted the vascular surgeons were consulted. The patient was too unfit to be anaesthetised and it was decided that the best option in her case was to try to retrieve the stent from the common femoral vein under local anaesthesia.
Under intravenous sedation and local infiltration, the common femoral vein was controlled proximal and distal to the venous access and through a venotomy the stent was retrieved [Fig 5]. The tunnelled line was still in-situ and functional. Subsequent venography showed no visible damage to the right atrium, inferior vena cava, iliac or femoral veins. On review months later, the patient continues to be dialysed through the femoral line with no known ill effect as a result of the reported complication.
With an increasing number of central venous access procedures in recent years in oncology, renal failure and Nutrition, the incidence of upper extremity venous occlusion is also more prevalent. It is estimated that as many as 40% of patients who undergo subclavian vein catheterisation eventually develop venous stenosis . The postulated mechanisms involve turbulence, nonphysiologic increase in flow volume, platelet aggregation, fibrosis and stenosis of the lumen by central venous catheters .
Various interventional radiological procedures have evolved to treat central venous obstruction, such as balloon angioplasty, pharmacologic and mechanical thrombolysis, thrombectomy and stenting. The frequent use of stenting as therapy for venous occlusion is controversial. Dotter first described Stents in arterial system in 1969 . Zollikofer in 1988 described the clinical applications in the venous system . Stenting is thought to act as an adjuvant to venous angioplasty by limiting the elastic recoil in compliant veins, excluding the damaged and dissected vasculature and counteracting extrinsic compression. Both balloon-expandable stents like Palmaz stents and self-expandable flexible stents like Wallstents have been implanted in central venous stenoses , but because they mould better to the vessel wall, Wallstents are preferred by most Radiologists for restoring venous patency. A self-adjusting stent is advantageous since chronic venous occlusions may undergo progressive luminal enlargement after stent deployment .
Stenting symptomatic venous obstruction achieves temporary benefit but regular follow up and reinterventions may be required to maintain patency . Primary venoplasty patency rates can be increased by 10% at 12 months by Stenting with a reduction in procedures for restenosis .
Migration or misplacement of endoprostheses is being recognised more often . Stent migration can result in potentially serious complications such as lodgement in the right ventricle [10–12]. Percutaneous techniques for rescuing dislocated endovascular stents can be effective with few complications in most cases . In extreme cases; displaced stents could be left in-situ with no disastrous consequences .
The increasing complexity of the complications arising from endovascular procedures often require the vascular interventionist to be innovative in order to rescue vascular access which is precious both for the patients and their physicians . On this Occasion a combined radiological and Surgical Procedure was required.
Care must be taken where catheter are inserted through the lumina of such stents as dislodgement may result in due consequences.
Written informed consent was obtained from the patient for participation in this research.
A copy of the written consent is available for review by the Editor-in-Chief of this journal.
Surrat R, Picus D, Hicks M, Darcy MD, Kleinhoffer M, Jendrisak M: The importance of preoperative evaluation of the subclavian vein in dialysis planning. AJR. 1991, 156: 623-625.
Maintz D, Landwehr P, Gawenda M, Lackner K: Failure of Wallstents in the subclavian vein due to stent damage. Clinical Imaging. 25 (2): 133-137. 10.1016/S0899-7071(01)00261-3.
Dotter C: Transluminally placed coil-spring endoarterial tubegrafts: long-term patency in canine popliteal artery. Invest Radiol. 1969, 57: 5-10.
Zollikofer C, Largiader I, Bruhlmann W, Uhlschmid GK, Marty AH: Endovascular stenting of veins and grafts: preliminary clinical experience. Radiology. 1988, 167: 707-712.
Haage P, Vorwerk D, Piroth W, Schuermann K, Guenther RW: Treatment of haemodialysis related central venous stenosis or occlusion: Results of Primary Wallstent placement and follow up in 50 patients. Radiology. 1999, 212: 175-180.
Vorwerk D, Guenther RW, Mann H, Bohndorf K, Keulers P, Alzen G, Sohn M, Kistler D: Venous stenosis and occlusion in haemodialysis shunts: follow up results of stent placement in 65 patients. Radiology. 1995, 195: 140-146.
Oderich GS, Treiman GS, Schneider P, Bhirangi K: Stent placement for treatment of central and peripheral venous obstruction: A long-term multi-institutional experience. J Vasc Surg. 2000, 32: 760-769. 10.1067/mva.2000.107988.
Turmel-Rodrigues L, Pengloan J, Blanchier D, Abaza M, Birmele B, Haillot O, Blanchard D: Insufficient dialysis shunts improved patency rates with closed hemodynamic monitoring, repeated percutaneous angioplasties, and stent placement. Radiology. 1993, 187: 273-278.
Slonim SM, Dake MD, Razavi MK, Kee ST, Samuels SL, Rhee JS, Semba CP: Management of misplaced or migrated endovascular stents. J Vasc Interv Radiol. 1999, 10 (7): 851-859.
Dubois P, Mandieau A, Dolatabadi D, Chaumont P, Gurnet P, Thiriaux J, Delcour C, Creplet J: Right ventricular migration of a stent after endovascular treatment of a superior vena cava syndrome. Arch Mal Coeur Vaiss. 2001, 94 (11): 1180-3.
Srinathan S, McCafferty I, Wilson I: Radiological management of superior vena caval stent migration and infection. Cardiovasc Intervent Radiol. 2005, 28 (1): 127-30. 10.1007/s00270-003-0183-x.
Chiu KM, Chu SH, Chan CY: Dislodged caval stent in right pulmonary artery. Catheter Cardiovasc Interv. 70 (6): 799-800. 10.1002/ccd.21251. 2007 Nov 15
Gabelmann A, Krämer SC, Tomczak R, Görich J: Percutaneous techniques for managing maldeployed or migrated stents. J endovasc Ther. 2001, 8 (3): 291-302. 10.1583/1545-1550(2001)008<0291:PTFMMO>2.0.CO;2.
Sharma AK, Sinha S, Ali Bakran: Migration of intravascular metallic stent into pulmonary artery. Nephrol Dial Transplant. 2002, 17 (3): 511. 10.1093/ndt/17.3.511.
Taylor JD, Lehmann ED, Belli AM, Nicholson AA, Kessel D, Robertson IR, Pollock JG, Morgan RA: Strategies for the management of SVC stent migration into the right atrium. Cardiovasc Intervent Radiol. 2007, 30 (5): 1003-9. 10.1007/s00270-007-9109-3.
The author(s) declare that they have no competing interests.
All authors were involved in the care of the patient. NBB and PM drafted the manuscript. NJM, FM, GH critically reviewed and improved the manuscript. All authors read and approved the final manuscript.
About this article
Cite this article
Bagul, N.B., Moth, P., Menon, N.J. et al. Migration of superior vena cava stent. J Cardiothorac Surg 3, 12 (2008). https://doi.org/10.1186/1749-8090-3-12
- Venous Occlusion
- Common Femoral Vein
- Left Femoral Vein
- Superior Vena Caval Obstruction
- Palmaz Stents