This article has Open Peer Review reports available.
Is distortion of the bioprosthesis ring a risk factor for early calcification?
© Cereijo et al; licensee BioMed Central Ltd. 2010
Received: 13 April 2010
Accepted: 7 October 2010
Published: 7 October 2010
As the population ages, bioprosthesis are increasingly being used in cardiac valve replacement. Pericardial bioprosthesis combine an excellent hemodynamic performance with low thrombogenicity, but valve failure associated with calcification remains a concern with these valves. We describe distortion of the bioprosthesis ring as a risk factor for early calcification.
A total of 510 patients over the age of 70 years underwent isolated aortic valve replacement with the Mitroflow (A12) pericardial bioprosthesis. Thirty two patients (6,2%) have undergone a second aortic valve replacement due to structural valve dysfunction resulting from valve calcification. In all patients a chest radiography and coronary angiography was performed before reoperation. A 64 Multidetector Computed Tomography (MDCT) with retrospective ECG gating study was performed in four patients to evaluate the aortic bioprosthesis.
Chest radiography showed in all patients an irregular bioprosthesis ring. At preoperative coronary angiography a distorted bioprosthesis ring was detected in all patients. Macroscopic findings of the explanted bioprostheses included extensive calcification in all specimens.
There was a possible relationship between early bioprosthetic calcification and radiologic distortion of the bioprosthesis ring.
As the population ages, bioprosthesis are increasingly being used in cardiac valve replacement. Pericardial bioprosthesis combine an excellent hemodynamic performance  with low thrombogenicity , but valve failure associated with calcification remains a concern with these valves . We describe distortion of the bioprosthesis ring as a risk factor for early calcification.
Materials and methods
77 ± 4 (71 - 86)
Valve Size (mm)
In all patients a chest radiography and coronary angiography was performed before reoperation.
A 64 Multidetector Computed Tomography (MDCT) with retrospective ECG gating study was performed in four patients to evaluate the aortic bioprosthesis. In two patients the study was performed three weeks after aortic valve replacement with the Mitroflow A12 pericardial bioprosthesis. Both bioprosthesis were normal by echocardiographic study. Another two patients had bioprosthesis with SVD.
A Histopathologic analysis of the removed bioprosthesis was performed by one experienced pathologist.
Macroscopic findings of the explanted bioprostheses included extensive calcification in all specimens. In all cases, one leaflet was more calcified than the others; in our experience, the right coronary leaflet was commonly the most calcified (19/32. 59,3%).
In all patients, serum levels of cholesterol, triglycerides, lipoprotein A and calcium were within the normal range. No patients were treated with calcium supplementation.
The percentage of distortion of the bioprosthesis ring in patients without SVD was 76%, but no all patients had the same degree of distortion.
Bioprosthetic calcification is a multifactorial process; contributing factors include the type of implant used, mechanical stress, preservation method, patient age and technical correctness of the implantation procedure . The rate of structural valve dysfunction is lower in elderly patients and early calcification in this group of patients is an uncommon event.
There is a strong relationship between mechanical stress and calcification in bioprosthesis [4, 5]; in our study we noted that early calcification tended to develop in patients with radiologic distortion of the bioprosthesis ring. Liao and colleagues  observed similar results with bioprostheses removed from left ventricular assist devices. Distortion of the bioprosthesis ring may be due to technical problems at implantation  or compression around the ring. However we speculate that Dacron ring traction from the annular stitch may distort the normal planar geometry of Mitroflow pericardial bioprosthesis, leading to distortion of the pericardial leaflet mounted outside the stent and fixed to the Dacron ring, resulting in a higher mechanical stress.
In our patients other casual relationship was no found.
In our experience there was a possible relationship between bioprosthetic calcification and radiologic distortion of the bioprosthesis ring.
- Bengochea JG, Sierra J, Gonzalez - Juanatey JR, Rubio J, Vega M, Fernandez AL, Sanchez D: Left ventricular mass regressión after aortic valve replacement with the new Mitroflow 12 A pericardial bioprosthesis. J Heart Valve Dis. 2006, 15: 446-452.Google Scholar
- Corbineau H, De la Tour B, Verhoye JP, Langanay Th, Lelong B, Leguerrier A: Carpentier-Edwards supraannular porcine bioprosthesis in aortic position: 16-year experience. Ann Thorac Surg. 2001, 71: S228-S231. 10.1016/S0003-4975(01)02545-0.View ArticlePubMedGoogle Scholar
- Rubio J, Sierra J, Vega M, Adrio B, Martinez Comendador JM, Gude F, Martinez Cereijo JM, Garcia J: Early calcification of the aortic Mitroflow pericardial bioprosthesis in the elderly. Interact Cardio Vasc Thorac Surg. 2009, 9: 842-846. 10.1510/icvts.2009.204958.View ArticleGoogle Scholar
- Liao KK, Li X, Ranjit J, Amayta DM, Joyce LD, Park SJ, Bianco R, Bolman RM: Mechanical stress: An independent determinant of early bioprosthetic calcification in humans. Ann Thorac Surg. 2008, 86: 491-495. 10.1016/j.athoracsur.2008.03.061.View ArticlePubMedGoogle Scholar
- Bernacca GM, Fisher AC, Wilkinson R, Mackay TG, Wheatley D: Calcification and stress distribution in bovine pericardial heart valves. J Biomed Mater Res. 1992, 26: 959-966. 10.1002/jbm.820260710.View ArticlePubMedGoogle Scholar
- Saunders PC, Grossi EA, Esposito RA, Bizekis CS, Strong MD, Colvin SB: Failure of four bovine pericardial mitral prostheses. J Thorac Cardiovasc Surg. 2004, 127: 267-268. 10.1016/S0022-5223(03)00724-4.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.