- Research article
- Open Access
- Open Peer Review
Effects on heart pumping function when using foam and gauze for negative pressure wound therapy of sternotomy wounds
© Malmsjö et al; licensee BioMed Central Ltd. 2011
- Received: 4 August 2010
- Accepted: 13 January 2011
- Published: 13 January 2011
Negative pressure wound therapy (NPWT) has remarkable effects on the healing of poststernotomy mediastinitis. Foam is presently the material of choice for NPWT in this indication. There is now increasing interest in using gauze, as this has proven successful in the treatment of peripheral wounds. It is important to determine the effects of NPWT using gauze on heart pumping function before it can be used for deep sternotomy wounds. The aim was to examine the effects of NPWT when using gauze and foam on the heart pumping function during the treatment of a sternotomy wound.
Eight pigs underwent median sternotomy followed by NPWT at -40, -70, -120 and -160 mmHg, using foam or gauze. The heart frequency, cardiac output, mean systemic arterial pressure, mean pulmonary artery pressure, central venous pressure and left atrial pressure were recorded.
Cardiac output was not affected by NPWT using gauze or foam. Heart frequency decreased during NPWT when using foam, but not gauze. Treatment with foam also lowered the central venous pressure and the left atrial pressure, while gauze had no such effects. Mean systemic arterial pressure, mean pulmonary artery pressure and systemic vascular resistance were not affected by NPWT. Similar haemodynamic effects were observed at all levels of negative pressure studied.
NPWT using foam results in decreased heart frequency and lower right and left atrial filling pressures. The use of gauze in NPWT did not affect the haemodynamic parameters studied. Gauze may thus provide an alternative to foam for NPWT of sternotomy wounds.
- Cardiac Output
- Negative Pressure
- Central Venous Pressure
- Pulmonary Artery Pressure
Cardiac surgery is complicated by poststernotomy mediastinitis in 1-5% of all procedures ; a complication that is life-threatening . The reported early mortality using conventional therapy is between 8 and 25% [3, 4]. In 1999, Obdeijn and colleagues described the treatment of poststernotomy mediastinitis using vacuum-assisted closure , now called negative pressure wound therapy (NPWT). The technique entails the application of negative pressure to a sealed wound. NPWT has remarkable effects on the healing of poststernotomy mediastinitis, and has reduced the rate of mortality considerably . The organs in the mediastinum are haemodynamically crucial, and both vulnerable bypass grafts and heart function should be taken into consideration when performing NPWT after cardiac surgery.
NPWT is known to affect the heart pumping function, although the results from different studies on the subject are not consistent [7–11]. Using sonometry, Conquest and colleagues showed that subatmospheric pressure decreased the left ventricular volume and cardiac output in pigs by approximately 30% , but that this could be prevented by rotating a rectus muscle flap over the mediastinal wound. In a subsequent study by Petzina et al., using magnetic resonance imaging, NPWT of sternotomy wounds in pigs was shown to decrease the cardiac output to a lesser extent, 13% . In studies by Sjögren et al. and Steigelman et al., using thermodilution, cardiac output was found not to be affected by negative pressure [8, 11].
Foam is presently the material of choice when applying NPWT to deep sternotomy wounds. However, interest in the use of gauze has increased [12, 13], as this has proven successful in the treatment of other kinds of peripheral wounds . The advantages with gauze include its conformability and ease of application to large and irregular wounds . It has been suggested that gauze under negative pressure can tamponade superficial bleedings in sternotomy wounds . Furthermore, there are no reported problems with ingrowth of granulation tissue into gauze in NPWT . Before gauze is used for NPWT of sternotomy wounds, it is important to determine the effects on heart pumping function. Hitherto, the effect of NPWT on the pumping function of the heart has only been examined using foam [7–10]. The aim of the present study was therefore to compare the haemodynamic effects of NPWT using gauze and foam. Eight pigs underwent median sternotomy and the wounds were treated with NPWT at negative pressures of -40, -70, -120 or -160 mmHg, using foam or gauze. Haemodynamic parameters, including heart frequency, cardiac output, mean systemic arterial pressure, mean pulmonary artery pressure, central venous pressure and left atrial pressure, were recorded.
A porcine sternotomy wound model was used. Eight domestic landrace pigs with a mean body weight of 70 kg were fasted overnight with free access to water. The study was approved by the Ethics Committee for Animal Research, Lund University, Sweden. The investigation complied with the "Guide for the Care and Use of Laboratory Animals" as recommended by the U.S. National Institutes of Health and published by the National Academies Press (1996).
Anaesthesia and surgery
Premedication was performed with an intramuscular injection of xylazine (Rompun® vet. 20 mg/ml; Bayer AG, Leverkusen, Germany; 2 mg/kg) mixed with ketamine (Ketaminol® vet. 100 mg/ml; Farmaceutici Gellini S.p.A, Aprilia, Italy; 20 mg/kg). Before surgery, a tracheotomy was performed and an endo-tracheal tube was inserted. Anaesthesia was maintained with a continuous infusion of ketamine (Ketaminol® vet. 50 mg/ml; 0.4-.6 mg/kg/h). Complete neuromuscular blockade was achieved with a continuous infusion of pancuronium bromide (Pavulon; N.V. Organon, OSS, the Netherlands; 0.3-0.5 mg/kg/h). Fluid loss was compensated for by continuous infusion of Ringer's acetate at a rate of 300 ml/kg/h. Mechanical ventilation was established with a Siemens-Elema ventilator (Servo Ventilator 300, Siemens, Solna, Sweden) in the volume-controlled mode (65% nitrous oxide, 35% oxygen). Ventilatory settings were identical for all animals (respiratory rate: 15 breaths/min; minute ventilation: 8 l/min). A positive end-expiratory pressure of 5 cmH2O was applied. A Foley catheter was inserted into the urinary bladder through a suprapubic cystostomy. Upon completion of the experiments, the animals were euthanized with a lethal dose (60 mmol) of intravenous potassium chloride.
Wound preparation for NPWT
A midline sternotomy was performed. Three layers of paraffin gauze Jelonet® (Smith & Nephew, Hull, UK) were placed over the anterior surface of the heart to protect it from the sternal edges. The wound was filled with saline-soaked AMD gauze (RENASYS-G, St Petersberg, FL), or one of two kinds of foam: VAC foam size 18 × 12.5 × 3.3 cm (VAC® black GranuFoam®, KCI, San Antonio, TX), or RENASYS-F foam, 20 × 12.5 × 3 cm (RENASYS-F, St Petersberg, FL). One layer of foam or two rolls of gauze were placed between the sternal edges. A second layer of foam or two rolls of gauze were placed over the first layer, between the soft tissue wound edges, and secured to the surrounding skin. Two drains were inserted into the wound filler. One drain was placed between two layers of wound filler (foam or gauze) and the other was placed at the top of the wound filler The wound was sealed with a transparent adhesive drape and connected to a vacuum source (RENASYS-EZ, Smith & Nephew St Petersburg, FL). The vacuum source was set to deliver negative continuous pressures of -40, -70, -120 and -160 mmHg.
The mean systemic arterial pressure was monitored via a catheter in the left carotid artery. The mean pulmonary artery pressure was monitored via a catheter in the pulmonary artery. Double-lumen central venous catheters were inserted into the left external jugular vein and the left atrium to record the central venous pressure and left atrial pressure, respectively. A flow probe (CardioMed TraCe System, Medistim, Norway) was placed around the pulmonary artery to record the cardiac output. The catheter was connected to a cardiac output monitor (Oximetrix 3, Abbot Laboratories, North Chicago, IL, USA). The haemodynamic data were collected in a data acquisition system (PowerLab, AD Instruments Ltd., Castle Hill, Australia). Systemic vascular resistance and pulmonary vascular resistance were calculated from the above parameters.
After surgical preparation, the animal was allowed to stabilize for one hour. Baseline measurements of the above mentioned haemodynamic parameters were then recorded before applying NPWT. Negative pressures of -40, -70, -120 or -160 mmHg were applied, and the effects on the haemodynamic parameters were recorded after 2½ and 5 min of NPWT. The negative pressures were applied in a random order, with intervals of 10 min without any pressure.
The study was performed using an acute sternotomy wound model in pigs where the organs in the thoracic cavity can move freely. In patient s with mediastinitis there are typically adhesions around the heart that may restrict movement. In addition, following a short period of NPWT, the mediastinum becomes fixed and rigid with the development of granulation tissue and fibrous ingrowth. The hemodynamic effects of NPWT may differ between an acute porcine sternotomy wound model and patients with mediastinitis.
Calculations and statistics
Calculations and statistical analysis were performed using GraphPad 5.0 software (San Diego, CA, USA). The effect of NPWT on each haemodynamic parameter was calculated as a percent of the baseline value. Statistical analysis was performed using the Mann-Whitney test. Significance was defined as P < 0.05 (*), P < 0.01 (**), P < 0.001 (***) and P > 0.05 (not significant, n.s.). Values are presented as means ± the standard error on the mean (S.E.M.) unless otherwise stated.
The effects of NPWT on the haemodynamic parameters were immediate, and the recordings after 2½ and 5 min of therapy were similar (Figures 1, 2 and 3). Comparable haemodynamic effects were observed at all levels of negative pressure studied (-40, -70, -120 and -160 mmHg). There was no significant difference in the haemodynamic effects when using the two different types of foam.
The present findings regarding cardiac output agree largely with those in previous studies, although the results in these studies vary somewhat. Mohktari et al. reported that the haemodynamics in pigs with a sternotomy wound treated with NPWT at -75 mmHg was unaltered, using an ultraflow probe to measure the flow through the pulmonary artery . Furthermore, in a study by Steigelman et al., using theromdilution, cardiac output was unaltered by a negative pressure of -125 mmHg . In a study by Sjögren et al., using thermodilution, the cardiac output was found to be slightly increased when a sternotomy wound in pigs was treated with a negative pressure of -75, while other negative pressures (-50, -100, -125, -150 and -175 mmHg) had no effect . In a study by Petzina et al., cardiac output was reported to be slightly reduced during NPWT (-75, -125 and -175 mmHg) of sternotomy wounds in pigs, when measured using magnetic resonance imaging .
The effect of NPWT on cardiac output is especially important since many patients with deep sternal wound infections have impaired cardiac function and heart failure due to ischaemic heart disease. Their ability to compensate for a decrease in cardiac output during NPWT may thus be limited. It has been suggested that haemodynamic parameters should be carefully monitored in patients undergoing sternal NPWT . However, when considering the data from the present study and other reports [8–10], it appears that NPWT has no major effects on cardiac output, and invasive monitoring may therefore be unnecessary.
The results of the present study show that the heart frequency decreased when NPWT was applied using foam. Similar results have been found in an experimental study on NPWT of a sternotomy wound using foam, demonstrating a tendency towards decreased heart rated upon the application of negative pressures of -75, -125 and -175 mmHg . Interestingly, the present study shows that gauze has no such effects. The reason for this cannot be deduced from the present study, but may be due to differences in the mechanical effects by the two materials on the heart . The decrease in heart rate when using foam is only slight (~20%) and the heart pumping function may only be affected when compensatory mechanisms are compromised as a result of advanced heart failure.
Left and right atrial filling pressure
The results of this and previous studies indicate that careful consideration should be given to the choice of material used in NPWT, as both foam and gauze may have advantages and disadvantages . Both result in the formation of granulation tissue; foam leads to the growth of a thick layer of tissue, which may penetrate the foam, making it difficult to remove after NPWT. Gauze leads to the growth of thin, dense granulation tissue without ingrowth, commonly referred to by physicians as high-quality tissue. NPWT using gauze has proven to be successful in the treatment of other kinds of wounds .
This study was performed to compare the effects of NPWT of a sternotomy wound using foam and gauze on heart pumping function. Foam is presently the material of choice for NPWT in cardiac surgery. NPWT using foam results in a slight decrease in heart frequency and lower right and left atrial filling pressures. NPWT using gauze does not affect the heart pumping function. Gauze may provide an alternative to foam for the treatment of sternotomy wounds using NPWT.
This study was funded by Åke-Wieberg Foundation, the Magn Bergvall Foundation, the Swedish Medical Association, the Royal Physiographic Society in Lund, the Crafoord Foundation, the Swedish Heart-Lung Foundation, the Swedish Hypertension Society and Smith & Nephew Wound Management.
- Raudat CW, Pagel J, Woodhall D, Wojtanowski M, Van Bergen R: Early intervention and aggressive management of infected median sternotomy incision: a review of 2242 open-heart procedures. Am Surg. 1997, 63: 238-241. discussion 241-232PubMedGoogle Scholar
- El Oakley RM, Wright JE: Postoperative mediastinitis: classification and management. Ann Thorac Surg. 1996, 61: 1030-1036. 10.1016/0003-4975(95)01035-1.View ArticlePubMedGoogle Scholar
- Crabtree TD, Codd JE, Fraser VJ, Bailey MS, Olsen MA, Damiano RJ: Multivariate analysis of risk factors for deep and superficial sternal infection after coronary artery bypass grafting at a tertiary care medical center. Semin Thorac Cardiovasc Surg. 2004, 16: 53-61. 10.1053/j.semtcvs.2004.01.009.View ArticlePubMedGoogle Scholar
- Lu JC, Grayson AD, Jha P, Srinivasan AK, Fabri BM: Risk factors for sternal wound infection and mid-term survival following coronary artery bypass surgery. Eur J Cardiothorac Surg. 2003, 23: 943-949. 10.1016/S1010-7940(03)00137-4.View ArticlePubMedGoogle Scholar
- Obdeijn MC, de Lange MY, Lichtendahl DH, de Boer WJ: Vacuum-assisted closure in the treatment of poststernotomy mediastinitis. Ann Thorac Surg. 1999, 68: 2358-2360. 10.1016/S0003-4975(99)01159-5.View ArticlePubMedGoogle Scholar
- Sjogren J, Gustafsson R, Nilsson J, Malmsjo M, Ingemansson R: Clinical outcome after poststernotomy mediastinitis: vacuum-assisted closure versus conventional treatment. Ann Thorac Surg. 2005, 79: 2049-2055. 10.1016/j.athoracsur.2004.12.048.View ArticlePubMedGoogle Scholar
- Conquest AM, Garofalo JH, Maziarz DM, Mendelson KG, Su Sun Y, Wooden WA, Meadows WM, Nifong W, Chitwood WR: Hemodynamic effects of the vacuum-assisted closure device on open mediastinal wounds. J Surg Res. 2003, 115: 209-213. 10.1016/S0022-4804(03)00331-7.View ArticlePubMedGoogle Scholar
- Sjögren J, Gustafsson R, Wackenfors A, Malmsjö M, Algotsson L, Ingemansson R: Effects of vacuum-assisted closure on central hemodynamics in a sternotomy wound model. ICVTS. 2004, 3: 666-671.Google Scholar
- Mokhtari A, Gustafsson R, Sjogren J, Nilsson J, Lindstedt S, Malmsjo M, Ingemansson R: Haemodynamic effects of -75 mmHg negative pressure therapy in a porcine sternotomy wound model. Int Wound J. 2009, 6: 48-54. 10.1111/j.1742-481X.2008.00566.x.View ArticlePubMedGoogle Scholar
- Petzina R, Ugander M, Gustafsson L, Engblom H, Sjogren J, Hetzer R, Ingemansson R, Arheden H, Malmsjo M: Hemodynamic effects of vacuum-assisted closure therapy in cardiac surgery: assessment using magnetic resonance imaging. J Thorac Cardiovasc Surg. 2007, 133: 1154-1162. 10.1016/j.jtcvs.2007.01.011.View ArticlePubMedGoogle Scholar
- Steigelman MB, Norbury KC, Kilpadi DV, McNeil JD: Cardiopulmonary effects of continuous negative pressure wound therapy in swine. Ann Thorac Surg. 2009, 88: 1277-1283. 10.1016/j.athoracsur.2009.06.027.View ArticlePubMedGoogle Scholar
- Campbell PE: Surgical wound case studies with the versatile 1 wound vacuum system for negative pressure wound therapy. J Wound Ostomy Continence Nurs. 2006, 33: 176-185. discussion 185-190View ArticlePubMedGoogle Scholar
- Miller MS, Lowery CA: Negative pressure wound therapy: "a rose by any other name". Ostomy Wound Manage. 2005, 51: 44-46. 48-49PubMedGoogle Scholar
- Campbell PE, Smith GS, Smith JM: Retrospective clinical evaluation of gauze-based negative pressure wound therapy. Int Wound J. 2008, 5: 280-286. 10.1111/j.1742-481X.2008.00485.x.View ArticlePubMedGoogle Scholar
- Abu-Omar Y, Naik MJ, Catarino PA, Ratnatunga C: Right ventricular rupture during use of high-pressure suction drainage in the management of poststernotomy mediastinitis. Ann Thorac Surg. 2003, 76: 974-10.1016/S0003-4975(03)00180-2. author reply 974-975View ArticlePubMedGoogle Scholar
- Sartipy U, Lockowandt U, Gabel J, Jideus L, Dellgren G: Cardiac rupture during vacuum-assisted closure therapy. Ann Thorac Surg. 2006, 82: 1110-1111. 10.1016/j.athoracsur.2006.01.060.View ArticlePubMedGoogle Scholar
- Malmsjo M, Petzina R, Ugander M, Engblom H, Torbrand C, Mokhtari A, Hetzer R, Arheden H, Ingemansson R: Preventing heart injury during negative pressure wound therapy in cardiac surgery: assessment using real-time magnetic resonance imaging. J Thorac Cardiovasc Surg. 2009, 138: 712-717. 10.1016/j.jtcvs.2008.11.068.View ArticlePubMedGoogle Scholar
- Jeffery LC: Advanced wound therapies in the management of severe military lower limb trauma: a new perspective. Eplasty. 2009, 9: e28-PubMedPubMed CentralGoogle Scholar
- Sjogren J, Gustafsson R, Koul B, Ingemansson R: Selective mediastinal tamponade to control coagulopathic bleeding. Ann Thorac Surg. 2003, 75: 1311-1313. 10.1016/S0003-4975(02)04570-8.View ArticlePubMedGoogle Scholar
- Borgquist O, Gustafsson L, Ingemansson R, Malmsjo M: Tissue ingrowth into foam but not into gauze during negative pressure wound therapy. Wounds. 2009Google Scholar
- Malmsjö M, Borgquist O: NPWT settings and dressing choices made easy. Wounds International. 2010, 1 (3):Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.