This article has Open Peer Review reports available.
Degenerative processes in bioprosthetic mitral valves in juvenile pigs
© Honge et al; licensee BioMed Central Ltd. 2011
Received: 14 January 2011
Accepted: 15 May 2011
Published: 15 May 2011
Glutaraldehyde-treated bioprosthetic heart valves are commonly used for replacement of diseased heart valves. However, calcification and wear limit their durability, and the development of new and improved bioprosthetic valve designs is needed and must be evaluated in a reliable animal model. We studied glutaraldehyde-treated valves 6 months after implantation to evaluate bioprosthetic valve complications in the mitral position in juvenile pigs.
The study material comprised eight, 5-month old, 60-kg pigs. All pigs received a size 27, glutaraldehyde-treated, stented, Carpentier-Edwards S.A.V. mitral valve prosthesis. After six months, echocardiography was performed, and the valves explanted for gross examination, high resolution X-ray, and histological evaluation.
Five pigs survived the follow-up period. Preexplant echocardiography revealed a median peak and mean velocity of 1.61 m/s (range: 1.17-2.00) and 1.20 (SD = ±0.25), respectively, and a median peak and mean pressure difference of 10.42 mmHg (range: 5.83-16.55) and 6.51 mmHg (SD = ±2.57), respectively. Gross examination showed minor thrombotic depositions at two commissures in two valves and at all three commissures in three valves. High resolution X-ray imaging revealed different degrees of calcification in all explanted valves, primarily in the commissural and belly areas. In all valves, histological evaluation demonstrated various degrees of fibrous sheath formation, limited immunological infiltration, and no overgrowth of host endothelium.
Bioprosthetic glutaraldehyde-treated mitral valves can be implanted into the mitral position in pigs and function after 6 months. Echocardiographic data, calcification, and histological examinations were comparable to results obtained in sheep models and human demonstrating the suitability of the porcine model.
Glutaraldehyde-treated bioprosthetic heart valves are commonly used for replacement of diseased heart valves. In the mitral position, glutaraldehyde-treated valves are preferred in elderly patients (>65 years), in patients in whom successful repair is unlikely, in rheumatic disease and endocarditis, and when the use of anticoagulation therapy is contraindicated . The failure rate of current bioprosthetic mitral valves is much higher than that of aortic valves and reoperation is needed in 50% after 15 years . Therefore, the development of new and improved bioprosthetic valve designs is needed. To evaluate bioprosthetic heart valves before clinical implementation, animal testing is preferred. Several species (sheep, dog, and pig) have been chosen for valve evaluation, the sheep model being the most common [3–6]. The sheep model, however, has several limitations. It failed to demonstrate the thrombogenecity of the Medtronic Parallel mechanical valve, which, after implantation in patients, was associated with a high incidence of valve-related thrombosis. It also failed to show a strong inflammatory response toward a decellularized biological valve, Synergraft, which was seen after implantation in children [7, 8]. The anatomical and physiological similarities between pigs and humans in terms of heart size, cardiac output, blood pressure, and, in particular, platelet adhesive properties make this model suitable for heart valve evaluation [9–11].
To evaluate new bioprosthetic valves, it is essential to know how these perform in an animal model compared with standard glutaraldehyde-treated valves, which are considered the gold standard of bioprosthetic valves. We, therefore, chose the pig model to thoroughly evaluate standard glutaraldehyde-treated bioprosthetic stented porcine valves in the mitral position in a long-term model.
Materials and methods
This observational study was conducted in female Danish Landrace/Yorkshire pigs. The study material comprised eight, 5-month old, 60-kg pigs. All pigs received a size 27, glutaraldehyde-treated, stented, Carpentier-Edwards S.A.V. mitral valve prosthesis.
All animal experiments were conducted according to the guidelines given by the Danish Inspectorate for Animal Experimentation and after specific approval from this institution. Qualified animal caretaker personnel monitored the health status of the animals daily during the study period. Analgesics were administered if animals exhibited any sign of pain. In the case of refractory pain or failure to thrive, the animals were euthanized. At the end of the study, the animals were euthanized under anaesthesia.
Mitral valve implantation
The operative technique and anesthetic treatment have been described elsewhere . In brief, after sedation, intubation and median sternotomy, a cardiopulmonary bypass was performed and the heart arrested with cold crystalloid cardioplegia. A size 27 Carpentier-Edwards mitral bioprosthesis was then implanted through a left atriotomy. Next, the atrium was closed and a DC countershock was given. After approximately 45 minutes of reperfusion, the animal was weaned from the cardiopulmonary bypass and the chest closed in a standard manner. The chest drains were removed when satisfactory hemodynamics were obtained and drain production was below 50 ml/h. The animals were awakened after 4 hours and transported to the farm in the evening. Importantly, no antithrombotic therapy was given during the follow-up period.
Echocardiography was performed using a commercially available system (Vivid E9, General Electric, Horten, Norway) in supine pigs under anesthesia. Parasternal echocardiograms were obtained by a transthoracic approach, and apical echocardiograms were obtained through a minimal abdominal incision. The study protocol included 2-dimensional echocardiograms with color Doppler images (apical four chamber, two chamber, long axis, as well as parasternal long axis) to evaluate left ventricular ejection fraction and mitral regurgitation using visual assessment by an experienced observer . To measure blood velocities and pressure differences over the artificial valves, continuous-wave Doppler echocardiograms through the center of the mitral ostium were obtained . All echocardiograms were recorded twice including three consecutive heart beats, and analyzed offline using commercial software (Echopac, General Electric-Vingmed, Horten, Norway).
The animals were euthanized after 6 months, and the valves were explanted after administration of an intravenous dose of 10.000 IU unfractionated heparin. Animals that died during the first postoperative week were excluded from the study. If the animals exhibited failure to thrive later than 1 week postoperatively, they were euthanized and the valve was explanted.
All valves were inspected in situ and photographed after removal. Inspection included gross assessment of fenestration, thrombotic material, and vegetations. The amount of thrombotic material was quantified in terms of size, appearance, and location.
High Resolution X-ray was performed to evaluate locations and distribution of calcifications using an Xpert 40™, Kubtec technologies, Milford, CT, USA.
Cusps and housing site tissues were removed from the stent and transected radially. Next, the tissue was fixated in formaldehyde and embedded in paraffin. Hematoxylin-eosin (H&E), vimentin (monoclonal mouse antivimentin), elastin trichrome, von Willebrand factor (polyclonal rabbit antihuman, DAKO) (endothelial cell), smooth muscle cell actin (monoclonal mouse antihuman, DAKO) (smooth muscle cell), and von Kossa stains were used to evaluate areas of recellularization, cell type, structural changes in the trilaminar cusp architecture, and both intrinsic and extrinsic calcification foci. Cells with a clear reduction in basophilia and increase in eosinophilia in H&E stains as well as a negative vimentin stain were considered donor cells. Vimentin-positive cells were therefore considered to be host cells and were afterwards correlated with cell-specific stains to verify their phenotype in order to differentiate between host and donor cells. The specimens were evaluated as a single observer, nonblinded assessment. The equipment used for histological assessment and image capturing was an Olympus BX50 microscope with Olympus Power View II camera.
Standard descriptive statistics (means, standard deviations, medians, and ranges) were used to characterize the investigated valves. Blood velocity was expressed as median peak and mean velocity. Pressure difference was expressed as median peak and mean pressure difference.
Eight Carpentier-Edwards S.A.V. mitral valves were implanted into the mitral position. One pig was euthanized after 5 days because of failure to thrive and respiratory insufficiency. The valve was found to be competent without alterations compared with the same valve at the time of implantation. One pig was euthanized after 3 months because of sudden unexplained lower limb paralysis. Gross anatomy and histologic evaluation of the valve from this pig revealed a well-functioning valve prosthesis with a minor thrombosis in the commissural area and no endocarditis. One pig died suddenly after 4 months. Autopsy revealed severe thrombosis of the valve and histological evaluation showed infective endocarditis and calcification of all cusps. Five pigs survived the 6-month follow-up.
The median peak and mean velocities over the valves were 1.61 m/s (range: 1.17-2.00) and 1.20 (SD = ±0.25), respectively, and the median peak and mean pressure differences over the valves were 10.42 mmHg (range: 5.83-16.55) and 6.51 mmHg (SD = ±2.57), respectively. All velocities and pressure differences were within the normal ranges for Carpentier-Edwards mitral valves implanted into humans (14). Three animals had mild central mitral regurgitation, and all animals had normal left ventricular ejection fractions. No paravalvular leaks were observed.
In this study, we evaluated bioprosthetic glutaraldehyde-treated valves in the mitral position in juvenile pigs for valve complications 6 months after implantation. Few have studied bioprosthetic valves in the mitral position in pigs, and the present study is the first to thoroughly evaluate the performance of glutaraldehyde-treated valves in a long-term porcine model [15–17].
Using high resolution X-ray imaging and von Kossa stain, we found that calcification was apparent in all explanted valves. Calcification of the commissures could be seen in all valves to various degrees and involved the belly area of the cusps in two valves. Thrombotic deposits were especially prominent in connection with the commissural calcifications. Weber et al. found calcification in the aortic area in six of nine Mosaic valves explanted after 20 weeks in a sheep model, and in no cases was cusp calcification observed. The Hancock Standard valve was tested in the same study, and here calcification was focused in the aortic wall and commissures and occasionally at the base of the cusps . Flaming et al. found that calcification were preferentially located at the commissures and present in two of six Perimount valves and four of nine Trilogy valves after implantation in sheep . Additionally, cusp and wall-adjacent calcifications were less frequent. These findings by other groups correspond very well with those of the present study, and this demonstrates the suitability of the porcine model for the evaluation of calcification in bioprosthetic valves.
In a large study in sheep, Flameng et al. found by means of quantitative calcium content analysis significantly lower calcification of the cusps than in the aortic wall portions of the valve . Furthermore, the highest calcium levels were found in the commissures, and the lowest calcium levels were found in the edges of the cusps. Although we did not perform any quantitative calcium content analysis, these results correlate well to the spatial distribution of calcification observed in the present study by means of high resolution X-ray.
Histological evaluation of the investigated valves revealed the presence of non-vital donor cell remnants and very scarce amounts of host cell ingrowth. Fibroblasts were only found in the fibrous sheath or as a part of an inflammatory response toward the outer part of the stent-adjacent area, and endothelial cells could only be identified superimposing the fibrous sheath or as von Willebrand-positive cells in very limited numbers. A host cell inflammatory response was apparent in all valves but limited to the stent-adjacent areas except for one valve in which cusp infiltration was observed. Most of the explanted valves in this study presented with well-preserved trilaminar cusp tissue. Inflammatory cells were infrequently seen in the cusps and mostly limited to the stent-adjacent areas and muscle shelf tissues. The tissue preservation of glutaraldehyde-treated valves was also observed in recent sheep studies in which the cusps appeared well-preserved and contained wavy collagen [6, 18]. However, Duarte et al. reported trilaminar structure disruption of the Mosaic valve both with and without alpha amino oleic acid (AOA) treatment . Despite meticulous histological examination, we observed almost no endothelial cells except for those located on the surface of fibrous sheaths, and fibroblast ingrowth was limited to the inflammatory host response seen in the stent-adjacent areas of most of the valves. Therefore, recellularization of glutaraldehyde-treated valves in pigs is extremely limited after an implantation period of 6 months, which corresponds well with human studies of bioprosthetic valve endothealization .
We found that both blood velocities and pressure differences across the valve after 6 months were comparable with human studies of the Carpentier-Edwards S.A.V. valve in the mitral position as well as previous sheep studies by Duarte et al. and Irwin et al. [20, 22]. Flaming et al. found peak velocites (m/s) of 1.34 (0.89, 1.47) and 1.11 (0.75, 1.54) and mean gradients (mmHg) of 3.6 (3.0, 5.2) and 2.4 (1.7, 5.4) for the Perimount and Trilogy valves, respectively, after 5 months in a sheep model. Although a higher mean pressure difference was found in the present study, most likely caused by the thrombotic deposits in some of the valves, we consider our results to be comparable. Additionally, we observed trivial regurgitation in three valves, a finding which was also noted by Weber et al. in another study . In our study, the thrombotic depositions could have caused some degree of commissural fusion leading to a limitation in cusp movement and minimal valve insufficiency.
Our aim was to evaluate the porcine model for long-term testing of bioprosthetic mitral heart valves. The porcine mitral valve anatomy has already been verified to be very similar to human mitral valve anatomy [23, 24]. The present study demonstrates that the porcine model is a reliable animal model for long-term bioprosthetic heart valve evaluation and can be used in the future as a relevant, important, and demanding animal model. The findings in the present study speak for the pig as being an animal model for heart valve evaluation that can provide a satisfactory answer to the question of animal testing that will avoid later tragic clinical incidents such as those seen after preclinical sheep experiments. Thorough evaluation of heart valve bioprostheses is critical before any clinical use. Therefore, we consider the present animal model to be very suitable for preclinical bioprosthetic mitral valve testing to ensure proper patient care. Further validation of the porcine model from other groups to support our findings should be performed to verify the pig as a clinically valid animal model. Especially interesting would be the evaluation of other tissue valves, decellularized valves, and tissue-engineered polymer valves because of the recent focus on this topic.
A small number of animals were included in this study. However, because of the comparable results between the animals, we consider the number to be sufficient for the evaluation of the porcine model as a tool for bioprosthetic mitral valve evaluation. No anticoagulation therapy was used during the follow-up period, and this could have resulted in the thrombotic deposits observed. A more aggressive anticoagulation strategy might have limited this problem. No baseline echocardiography was performed postoperatively, and hemodynamic comparison with pre-explantation echocardiography was therefore not possible. The growth potential of the pig could result in a patient-prosthesis mismatch, however, we do not consider this a limiting factor, since all surviving pigs thrived well at euthanization, and none of the three animal deaths before the six months follow-up could be related to mitral stenosis.
Bioprosthetic glutaraldehyde-treated mitral valves can be implanted into the mitral position in pigs and function after a period of 6 months. Echocardiographic data, calcification, and histological examinations were comparable to results obtained in sheep and humans, and we therefore consider the porcine model an appropriate animal model for bioprosthetic mitral valve testing.
We are indebted to the staff at the Institute of Clinical Medicine, Tanja Thomsen, Kira Sonnichnsen, and Henrik Sørensen, as well as Walther Gyldenløve, Påskehøjgaard, for skillful assistance. Thanks to Professor Ulrik Baandrup, MD, PhD, Department of Pathology, Sygehus Vensyssel and Søren Redsted, MD, Department of Radiology, Aarhus University Hospital, for pathology and radiology assistance. Also thanks to scholarship students, Dept. T, for anesthesia assistance. Financial support was provided by the Danish Heart Association, Sophus Jacobsen Foundation, and Dagmar Marshalls Foundation.
- Bonow RO, Carabello BA, Chatterjee K, de Leon ACJ, Faxon DP, Freed MD: Focused update incorporated into the ACC/AHA 2006 guidelines for the management of patients with valvular heart disease: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (Writing Committee to Revise the 1998 Guidelines for the Management of Patients With Valvular Heart Disease): endorsed by the Society of Cardiovascular Anesthesiologists, Society for Cardiovascular Angiography and Interventions, and Society of Thoracic Surgeons. Circulation. 2008, 118: e523-e661.View ArticlePubMedGoogle Scholar
- Hammermeister K, Sethi GK, Henderson WG, Grover FL, Oprian C, Rahimtoola SH: Outcomes 15 years after valve replacement with a mechanical versus a bioprosthetic valve: final report of the Veterans Affairs randomized trial. J Am Coll Cardiol. 2000, 36: 1152-1158. 10.1016/S0735-1097(00)00834-2.View ArticlePubMedGoogle Scholar
- Gallegos RP, Nockel PJ, Rivard AL, Bianco RW: The current state of in-vivo pre-clinical animal models for heart valve evaluation. J Heart Valve Dis. 2005, 14: 423-432.PubMedGoogle Scholar
- Hasenkam JM, Nygaard H, Terp K, Riis C, Paulsen PK: Hemodynamic evaluation of a new bileaflet valve prosthesis: an acute animal experimental study. J Heart Valve Dis. 1996, 5: 574-580.PubMedGoogle Scholar
- Smerup M, Pedersen TF, Nyboe C, Funder JA, Christensen TD, Nielsen SL: A long-term porcine model for evaluation of prosthetic heart valves. Heart Surg Forum. 7: E259-E264.Google Scholar
- Flameng W, Meuris B, De VG, Cunanan C, Lane E, Verbeken E: Trilogy pericardial valve: hemodynamic performance and calcification in adolescent sheep. Ann Thorac Surg. 2008, 85: 587-592. 10.1016/j.athoracsur.2007.09.041.View ArticlePubMedGoogle Scholar
- Bodnar E: The Medtronic Parallel valve and the lessons learned. J Heart Valve Dis. 1996, 5: 572-573.PubMedGoogle Scholar
- Simon P, Kasimir MT, Seebacher G, Weigel G, Ullrich R, Salzer-Muhar U: Early failure of the tissue engineered porcine heart valve SYNERGRAFT in pediatric patients. Eur J Cardiothorac Surg. 2003, 23: 1002-1006. 10.1016/S1010-7940(03)00094-0.View ArticlePubMedGoogle Scholar
- Swindle MM, Horneffer PJ, Gardner TJ, Gott VL, Hall TS, Stuart RS: Anatomic and anesthetic considerations in experimental cardiopulmonary surgery in swine. Lab Anim Sci. 1986, 36: 357-361.PubMedGoogle Scholar
- Goodman SL: Sheep, pig, and human platelet-material interactions with model cardiovascular biomaterials. J Biomed Mater Res. 1999, 45: 240-250. 10.1002/(SICI)1097-4636(19990605)45:3<240::AID-JBM12>3.0.CO;2-C.View ArticlePubMedGoogle Scholar
- Pelagalli A, Belisario MA, Tafuri S, Lombardi P, d'Angelo D, Avallone L: Adhesive properties of platelets from different animal species. J Comp Pathol. 2003, 128: 127-131. 10.1053/jcpa.2002.0615.View ArticlePubMedGoogle Scholar
- McGowan JH, Cleland JG: Reliability of reporting left ventricular systolic function by echocardiography: a systematic review of 3 methods. Am Heart J. 2003, 146: 388-397. 10.1016/S0002-8703(03)00248-5.View ArticlePubMedGoogle Scholar
- Hatle L, Brubakk A, Tromsdal A, Angelsen B: Noninvasive assessment of pressure drop in mitral stenosis by Doppler ultrasound. Br Heart J. 1978, 40: 131-140. 10.1136/hrt.40.2.131.View ArticlePubMedPubMed CentralGoogle Scholar
- Rosenhek R, Binder T, Maurer G, Baumgartner H: Normal values for Doppler echocardiographic assessment of heart valve prostheses. J Am Soc Echocardiogr. 2003, 16: 1116-1127. 10.1067/S0894-7317(03)00638-2.View ArticlePubMedGoogle Scholar
- Gross DR: Thromboembolic phenomena and the use of the pig as an appropriate animal model for research on cardiovascular devices. Int J Artif Organs. 1997, 20: 195-203.PubMedGoogle Scholar
- Gross DR, Dewanjee MK, Zhai P, Lanzo S, Wu SM: Successful prosthetic mitral valve implantation in pigs. ASAIO J. 1997, 43: M382-M386. 10.1097/00002480-199709000-00003.View ArticlePubMedGoogle Scholar
- Honge JL, Funder JA, Jensen H, Dohmen PM, Konertz WF, Hasenkam JM: Recellularization of decellularized mitral heart valves in juvenile pigs. J Heart Valve Dis. 2010, 19: 584-592.PubMedGoogle Scholar
- Weber PA, Jouan J, Matsunaga A, Pettenazzo E, Joudinaud T, Thiene G: Evidence of mitigated calcification of the Mosaic versus Hancock Standard valve xenograft in the mitral position of young sheep. J Thorac Cardiovasc Surg. 2006, 132: 1137-1143. 10.1016/j.jtcvs.2006.06.027.View ArticlePubMedGoogle Scholar
- Flameng W, Meuris B, Yperman J, De VG, Herijgers P, Verbeken E: Factors influencing calcification of cardiac bioprostheses in adolescent sheep. J Thorac Cardiovasc Surg. 2006, 132: 89-98. 10.1016/j.jtcvs.2006.02.036.View ArticlePubMedGoogle Scholar
- Duarte IG, MacDonald MJ, Cooper WA, Schmarkey SL, Gott JP, Brown WM: In vivo hemodynamic, histologic, and antimineralization characteristics of the Mosaic bioprosthesis. Ann Thorac Surg. 2001, 71: 92-99. 10.1016/S0003-4975(00)01989-5.View ArticlePubMedGoogle Scholar
- Ishihara T, Ferrans VJ, Jones M, Boyce SW, Roberts WC: Occurrence and significance of endothelial cells in implanted porcine bioprosthetic valves. Am J Cardiol. 1981, 48: 443-454. 10.1016/0002-9149(81)90071-0.View ArticlePubMedGoogle Scholar
- Irwin E, Lang G, Clack R, St CJ, Runge W, Foker J: Long-term evaluation of prosthetic mitral valves in sheep. J Invest Surg. 1993, 6: 133-141. 10.3109/08941939309141604.View ArticlePubMedGoogle Scholar
- Crick SJ, Sheppard MN, Ho SY, Gebstein L, Anderson RH: Anatomy of the pig heart: comparisons with normal human cardiac structure. J Anat. 1998, 193: 105-119. 10.1046/j.1469-7580.1998.19310105.x.View ArticlePubMedPubMed CentralGoogle Scholar
- Kunzelman KS, Cochran RP, Verrier ED, Eberhart RC: Anatomic basis for mitral valve modelling. J Heart Valve Dis. 1994, 3: 491-496.PubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.