- Research article
- Open Access
- Open Peer Review
Application of Regent mechanical valve in patients with small aortic annulus: 3-year follow-up
© Zhao et al.; licensee BioMed Central Ltd. 2012
- Received: 16 May 2012
- Accepted: 17 September 2012
- Published: 21 September 2012
Aortic valve replacement (AVR) with a small aortic annulus is always challenging for the cardiac surgeon. In this study, we sought to evaluate the midterm performance of implantation with a 17-mm or 19-mm St. Jude Medical Regent (SJM Regent) mechanical valve in retrospective consecutive cohort of patients with small aortic annulus (diameter ≤ 19 mm).
From January 2008 to April 2011, 40 patients (31 female, mean age = 47.2 ± 5.8 years) with small aortic annulus (≤19 mm in diameter) underwent aortic valve replacement with a 17-mm or 19-mm St. Jude Medical Regent (SJM Regent) mechanical valve. Preoperative mean body surface area, New York Heart Association class, and mean aortic annulus were 1.61 ± 0.26 m2, 3.2 ± 0.4, and 18 ± 1.4 mm respectively. Patients were divided into two groups, according to the implantation of 17 mm SJM Regent mechanical valve (group 1, n = 18) or 19 mm SJM Regent valve (group 2, n = 22). All patients underwent echocardiography examination preoperatively and at one year post-operation.
There were no early deaths in either group. Follow-up time averaged 36 ± 17.6 months. The mean postoperative New York Heart Association class was 1.3 ± 0.6 (p < 0.001). By echocardiography, in group 1, the left ventricular ejection fraction (LVEF), left ventricular fraction shortening (LVFS), and the indexed effective orifice area (EOAI) increased from 43.7% ± 11.6%, 27.3% ± 7.6%, and 0.70 ± 0.06 cm2/m2 to 69.8 ± 9.3%, 41.4 ± 8.3%, and 0.92 ± 0.10 cm2/m2 respectively (P < 0.05), while the left ventricular mass index (LVMI), and the aortic transvalvular pressure gradient decreased from 116.4 ± 25.4 g/m2, 46.1 ± 8.5 mmHg to 86.7 ± 18.2 g/m2 , 13.7 ± 5.2 mmHg respectively. In group 2, the LVEF, LVFS and EOAI increased from 45.9% ± 9.7%, 30.7% ± 8.0%, and 0.81 ± 0.09 cm2/m2 to 77.4% ± 9.7%, 44.5% ± 9.6%, and 1.27 ± 0.11 cm2/m2 respectively, while the LVMI, and the aortic transvalvular pressure gradient decreased from 118.3 ± 27.6 g/m2, 44.0 ± 6.7 mmHg to 80.1 ± 19.7 g/m2, 10.8 ± 4.1 mmHg as well. The prevalence of PPM was documented in 2 patients in Group 1.
Patients with small aortic annulus and body surface area, experienced satisfactory clinical improvement after aortic valve replacement with modern SJM Regent bileaflet prostheses.
- Small aortic annulus
- St.Jude Medical Regent mechanical valve
- Left cardiac function
- Prothesis-Patient Mismatch
Aortic valve replacement (AVR) with a small aortic annulus is always challenging for the cardiac surgeon. Traditional small mechanical valves, especially if implanted in patients with big body surface area, may show high aortic transvalvular pressure gradients, the so-called patient–prosthesis mismatch (PPM) phenomenon . Several surgical strategies can be applied to avoid PPM, among them aortic root enlargement procedure seems a preventive potential strategy to minimize mismatch, since it showed good early and long-term results in experienced centers . However, this approach is complex and time-consuming that may cause serious bleeding and infection, and can be difficult to perform in patients with a small, calcified aortic annulus and root.
On the other hand, modern St.Jude Medical Regent (SJM Regent) mechanical valve is much more favored for the unique design and the bigger effective orifice area (EOA). By shifting the sewing cuff and the retaining ring to the supra-annular position, the SJM Regent valve achieve a greater geometric orifice for a fixed diameter, which may prevent PPM phenomenon in patients with small aortic annulus as well [3–5]. In our present study, we sought to review the midterm performance of implantation with a 17-mm or 19-mm SJM Regent mechanical valve in consecutive patients with small aortic annulus (diameter ≤ 19 mm), and evaluate whether it is a preventive strategy to eliminate PPM.
Aortic valve characteristics
Group 1 (17 mm Regent, n = 18)
Group 2 (19 mm Regent, n = 22)
Aortic stenosis (congenital)
Aortic stenosis (degenerative)
Aortic insufficiency (rheumatic)
Aortic stenosis and insufficiency (rheumatic)
The surgical procedures were performed through a midline sternotomy under cardiopulmonary bypass and mild hypothermia. Myocardial protection was achieved with cold blood cardioplegia perfusion into coronary arteries or the aortic root, and the surface of heart was covered with ice. Valve prostheses were implanted at the valve ring level with non-everting interrupted mattress sutures. After the second postoperative day, patients received oral anticoagulation with sodium warfarin at daily updated dosages according to international normalized ratios. The target international normalized ratio value was in accordance with American Heart Association guidelines.
A complete M-mode, two-dimensional, and Doppler evaluation was performed at the probe frequency of 2.5/2.7 MHZ using commercially available ultrasonographic equipment (HP Sonos5500) in Zhongshan Hospital, Fudan University. Left ventricular (LV) end-diastolic and end-systolic diameters and volumes, as well as left ventricular ejection fraction (LVEF), left ventricular fraction shortening (LVFS), were calculated using the area-length method. Left ventricular mass index (LVMI) was calculated from Reichek’s formula. As regard to aortic valve evaluation, we measured indexed aortic valve area (EOAI), aortic annulus diameter, and the aortic transvalvular pressure gradient. Echocardiographic evaluation with the same variables was repeated at follow-up.
All patients were followed in the outpatient center with clinical visits and echocardiography performed on a 6-month basis. A telephone interview was required only for patients with follow-up visits in excess of 6 months or for those lost to ambulatory follow-up. The follow-up was 100% complete, and the mean time to last follow-up was 36 ± 17.6 months (ranging from 12 to 51 months).
Statistics were calculated using stata 7.0. The continuous varieties were expressed as average ± standard deviation, the Wilcoxon method was used to test the difference between two continuous varieties, and t-test was used to test paired samples. P < 0.05 means there is statistical differences, and P < 0.01 represents significant statistical differences.
Clinical results and follow-up
Concomitant operative characteristics
Group 1 (17 mm Regent, n = 18)
Group 2 (19 mm Regent, n = 22)
Total bypass time
78 ± 6 minutes
81 ± 7 minutes
45 ± 6 minutes
51 ± 4 minutes
There were no early deaths in either group. One early nonfatal cerebrovascular accident was observed in group 2. No severe arrhythmia, low cardiac output syndrome, pericardial tamponade, respiratory failure, hepatic failure and renal failure were observed. Follow-up time averaged 36 ± 17.6 months. The mean postoperative New York Heart Association class was 1.3 ± 0.6 (p < 0.001), and no further 1-year mortality occurred.
Evaluation of heart function
All patients underwent echocardiography examination preoperatively and at one year post-operation.
Comparison of cardiac function in two groups
43.7 ± 11.6
27.3 ± 7.6
116.4 ± 25.4
46.1 ± 8.5
0.70 ± 0.06
69.8 ± 9.3*
41.4 ± 8.3*
86.7 ± 18.2#
13.7 ± 5.2*
0.92 ± 0.10#
45.9 ± 9.7
30.7 ± 8.0
118.3 ± 27.6
44.0 ± 6.7
0.81 ± 0.09
77.4 ± 9.7*
44.5 ± 9.6*
80.1 ± 19.7#
10.8 ± 4.1*
1.27 ± 0.11#$
In both groups, left ventricular mass regression occurred. In group 1, LVMI decreased from 116.4 ± 25.4 g/m2 pre-operatively to 86.7 ± 18.2 g/m2 at follow-up (P < 0.05), while in group 2, it decreased from118.3 ± 27.6 g/m2 pre-operatively to 80.1 ± 19.7 g/m2 at follow-up (P < 0.05).
Using a 17-mm or 19-mm SJM Regent, also shown in Table 3 in both groups, hemodynamic performance improved as well. In group 1, aortic transvalvular pressure gradient decreased from 46.1 ± 8.5 mmHg to 13.7 ± 5.2 mmHg (P < 0.01).In group 2, aortic transvalvular pressure gradient decreased from 44.0 ± 6.7 mmHg to 10.8 ± 4.1 mmHg (P < 0.01). The hemodynamic performances in the two groups showed no significant difference (13.7 ± 5.2 mmHg versus 10.8 ± 4.1 mmHg).
On the other hand, though EOAI increased in both groups at one-year follow-up as compared to pre-operation, a 19-mm SJM Regent showed superior EOAI as compared to that in group 1(1.27 ± 0.11 cm2/m2 versus 0.92 ± 0.10 cm2/m2, P < 0.01).
The prevalence of PPM, defined as an indexed effective orifice area (EOA) of less than 0.85 cm2/m2, was documented in 2 patients (11%, the EOA of the one patient was 0.77 cm2/m2, the other was 0.83 cm2/m2) in group 1, and 0 in group 2. These two patients were both male with larger body surface area. However, the presence of PPM did not affect early and mid-term survival or postoperative functional status. A significant reduction of aortic transvalvular pressure gradient occurred in both patients.
PPM, a concept widely accepted by many scholars [1, 6–16], refers to a series of complications or potential risks that occurred when the size of a implanted valve prosthesis mismatches the patients’ BSA. Several surgical strategies can be applied to avoid PPM, among them aortic root enlargement procedure seems a preventive potential strategy to minimize mismatch. However, this approach is complex and time-consuming that may cause serious bleeding and infection, and can be difficult to perform in patients with a small, calcified aortic annulus and root. In the present study, we applied modern SJM Regent valves in patients with small aortic annulus, and sought to evaluate whether SJM Regent valve could be an alternate choice to minimize PPM.
It is no doubt that PPM has a significant negative impact on the regression of left ventricular hypertrophy, the improvement of left cardiac function, and a long life survival, as Castro point out . To minimize PPM after AVR, we need to understand the most determinant factors of PPM. In 1997, Sawant pointed out BSA ≥ 1.9 m2 as a risk factor for patients replaced with small mechanical valve (outer diameter < 21 mm), and may lead to sudden death . Later in 2000, Pibarot  held the view that whether there will be PPM after operation depends on BSA, as well as effective orifice area of the valve prosthesis. More recently, Botzenhardt  found out that the ratio of EOA of the valve to BSA (EOAI) is an indicator which is more important than the aortic transvalvular pressure gradient. As they concluded, when EOAI > 0.85 cm2/m2, no PPM is presented; when the EOAI is between 0.65-0.85 cm2/m2, a moderate PPM can be seen; and if the EOAI is lower than 0.65 cm2/m2, the PPM can be severe [6, 18]. From this view, given that the BSA of the patient is stable, to minimize PPM after AVR, we need to increase the EOA of the implanted valve.
Nowadays, there are three strategies to increase the EOA of the implanted valve when treating patients with small aortic annulus : (1) aortic annulus enlargement; (2) stentless bioprothesis valve; (3) mechanical valve of same diameter with bigger EOA, including SJM Regent mechanical valve, Carbomedics Top-hat valve, Sorin Slimline valve, and ATS AP valve. Although annulus/root enlargement procedures, as well as stentless valve implantation, performed in experienced centers, showed good early and long-term results [2, 20], these two procedures were complex and time-consuming. By shifting the sewing cuff and the retaining ring to the supra-annular position, the SJM Regent possesses a larger EOA. In our present study, most (38/40) patients got a satisfactory heart function improvement and life quality after implantation of a 17 mm or 19 mm Regent valve with two patients of PPM. The good results are persistent with the superior dynamic aspects of Regent valve as others showed. Kon’s research on Regent valve revealed that the blood flows through the valve is laminar, the transvalvular pressure gradient is low, and the EOA and EOAI are satisfying . The report showed that the mean/peak aortic transvalvular pressure gradient were 3/7 mmHg for healthy people at rest and 4/9 mmHg in motion, and the application of Regent valve could meet the standard . Due to the aspect of hemodynamics, Okumiya hold the view that the cardiac indexes of patients replaced with SJM Regent mechanical valve are far more better than those of the patients replaced with conventional mechanical valves .
In our present study, using 17 mm or 19 mm Regent valve as prosthesis, we performed the operations with mean crossclamp time of about 50 minutes. This procedure was much simpler as compared to aortic annulus enlargement and implantation of stentless bioprothesis valve. Further, no operation related death or serious postoperative complications were presented in our study. Furthermore, the heart function of all patients (LVEF and LVFS) at one year after operation improved with left ventricular mass regression and decrease of aortic transvalvular pressure gradient. It suggests that using small Regent valve be an alternate choice for cardiac surgeons when facing patients with small aortic annulus.
In our present study, 2 PPMs were documented in our group 1. In our present study, the incidence of PPM is relatively low (only 11% in group-1 and 0% in group-2). According to the information of St Jude Company, the 17 mm Regent valve is suitable for patients with BSA <1.7 m2, and 19 mm Regent valve is suitable for patients with BSA < 1.9 m2. In our study, the BSA of the two groups was 1.61 ± 0.26 m2. Actually, the BSA in group 1 (17 mm Regent) was 1.53 ± 0.12 m2 (data not shown). That is perhaps the main reason for our low PPM incidence. Besides, Dr. Minardi G, and Sezai A also showed similar low incidence of PPM to our study (10.5% and 10.3%,respectively) [12, 23]. Fortunately, as for these two patients, the presence of PPM did not affect early and 3-year survival or postoperative functional status. Further, a significant reduction of aortic transvalvular pressure gradient occurred in both patients. It might be attributed to that the EOA of the two patients (0.77 cm2/m2 and 0.83 cm2/m2) were little smaller than 0.85 cm2/m2. The long-term follow-up needs to be carefully monitored.
In conclusion, for patients with small aortic annulus and body surface area, implantation of SJM Regent bileaflet prostheses would bring along heart function improvement and clinical satisfaction of the patients, and would be a better choice as compared to aortic annulus enlargement.
Written informed consent was obtained from the patient for publication of this report and any accompanying images.
- Rahimtoola SH: The problem of valve prosthesis-patient mismatch. Circulation. 1978, 58 (1): 20-10.1161/01.CIR.58.1.20.View ArticlePubMedGoogle Scholar
- Castro LJ, Arcidi JM, Fisher AL, Gaudiani VA: Routine enlargement of the small aortic root: a preventive strategy to minimize mismatch. Ann Thorac Surg. 2002, 74: 31-36. 10.1016/S0003-4975(02)03680-9.View ArticlePubMedGoogle Scholar
- Emery RW: St Jude Medical Regent valve. J Thorac Cardiovasc Surg. 2006, 131 (3): 760-761.View ArticlePubMedGoogle Scholar
- Sudkamp M, Lercher AJ, Muller-Riemenschneider F: Transvalvular in vivo gradients of the new generation bileaflet heart valve prosthesis St. Jude Medical Regent in aortic position. Thorac Cardiovasc Surg. 2003, 51 (3): 126-129.View ArticlePubMedGoogle Scholar
- Gelsomino S, Morocutti G, Da Col P: Preliminary experience with the St. Jude Medical Regent mechanical heart valve in the aortic position: early in vivo hemodynamic results. Ann Thorac Surg. 2002, 73 (6): 1830-1836. 10.1016/S0003-4975(01)03603-7. discussion 1836View ArticlePubMedGoogle Scholar
- Pibarot P, Dumesnil JG: Hemodynamic and clinical impact of prosthesis-patient mismatch in the aortic valve position and its prevention. J Am Coll Cardiol. 2000, 36: 1131-10.1016/S0735-1097(00)00859-7.View ArticlePubMedGoogle Scholar
- Ellis JT, Yoganathan AP: A comparison of the hinge and near-hinge flow fields of the St Jude medical hemodynamic plus and regent bileaflet mechanical heart valves. J Thorac Cardiovasc Surg. 2000, 119 (1): 83-93. 10.1016/S0022-5223(00)70221-2.View ArticlePubMedGoogle Scholar
- Tzikas A, Piazza N, Geleijnse ML: Prosthesis-patient mismatch after transcatheter aortic valve implantation with the Medtronic corevalve system in patients with aortic stenosis. Am J Cardiol. 2010, 106: 255-260. 10.1016/j.amjcard.2010.02.036.View ArticlePubMedGoogle Scholar
- Yoshikawa K, Fukunaga S, Arinaga K: Long-term results of aortic valve replacement with a small St.Jude medical valve in Japanese patients. Ann Thorac Surg. 2008, 85: 1303-1309. 10.1016/j.athoracsur.2007.12.031.View ArticlePubMedGoogle Scholar
- Moon MR, Pasque MK, Munfakh NA: Prosthesis-patient mismatch after aortic valve replacement:impact of age and body size on late survival. Ann Thorac Surg. 2006, 81: 481-489. 10.1016/j.athoracsur.2005.07.084.View ArticlePubMedGoogle Scholar
- Okamura H, Yamaguchi A: The 17-mm St. Jude Medical Regent valve is a valid option for patients with a small aortic annulus. Ann Thorac Surg. 2009, 87 (1): 90-94. 10.1016/j.athoracsur.2008.09.051.View ArticlePubMedGoogle Scholar
- Sezai A, Kasamaki Y, Abe K: Assessment of the St. Jude Medical Regent prosthetic valve by continuous-wave Doppler and dobutamine stress echocardiography. Ann Thorac Surg. 2010, 89 (1): 87-92. 10.1016/j.athoracsur.2009.09.013.View ArticlePubMedGoogle Scholar
- Lam BK, Chan V, Hendry P: The impact of patient-prosthesis mismatch on late outcomes after mitral valve replacement. J Thorac Cardiovasc Surg. 2007, 133 (6): 1464-1473. 10.1016/j.jtcvs.2006.12.071. Epub 2007 Apr 27View ArticlePubMedGoogle Scholar
- Peterson MD, Borger MA, Feindel CM: Aortic annular enlargement during aortic valve replacement: improving results with time. Ann Thorac Surg. 2007, 83 (6): 2044-2049. 10.1016/j.athoracsur.2007.01.059.View ArticlePubMedGoogle Scholar
- Bakhtiary F, Schiemann M, Dzemali O: Impact of patient-prosthesis mismatch and aortic valve design on coronary flow reserve after aortic valve replacement. J Am Coll Cardiol. 2007, 49 (7)): 790-796. Epub 2007 Feb 5View ArticlePubMedGoogle Scholar
- Howell NY, Keogh BE, Barnet V: Patient-prosthesis mismatch does not affect survival following aortic valve replacement. Eur J Cardiothorac Surg. 2006, 30 (1): 10-14. 10.1016/j.ejcts.2006.03.046. Epub 2006 May 24View ArticlePubMedGoogle Scholar
- Sawant D, Singh AK, Feng WC: Nineteen-millimeter aortic St.Jude medical heart valve prosthesis:up to sixteen years' follow-up. Ann Thorac Surg. 1997, 63: 964-970. 10.1016/S0003-4975(96)01145-9.View ArticlePubMedGoogle Scholar
- Botzenhardt F, Eichinger WB, Bleiziffer S: Hemodynamic comparison of bioprostheses for complete supra-annular position in patients with small aortic annulus. J Am Coll Cardiol. 2005, 45: 2054-2060. 10.1016/j.jacc.2005.03.039.View ArticlePubMedGoogle Scholar
- Takaseya T, Kawara T, Tokunaga S: Aortic valve replacement with 17-mm St.Jude medical prostheses for a small aortic root in elderly patients. Ann Thorac Surg. 2007, 83: 2050-2053. 10.1016/j.athoracsur.2007.01.044.View ArticlePubMedGoogle Scholar
- Doss M, Martens S, Wood JP: Performance of stentless versus stented aortic valve bioprostheses in the elderly patient: a prospective randomized trial. Eur J Cardiothorac Surg. 2003, 23: 299-304. 10.1016/s1010-7940(02)00815-1.View ArticlePubMedGoogle Scholar
- Kon ND, Cordell AR, Adair SM: Aortic root replacement with the freestyle stentless porcine aortic root bioprosthesis. Ann Thorac Surg. 1999, 67: 1609-1615. 10.1016/S0003-4975(99)00370-7.View ArticlePubMedGoogle Scholar
- Okumiya, Toshika IN, Mitsuko D: Evaluation of intravascular hemolysis with erythrocyte creatine in patients with cardiac valve prostheses. Chest. 2004, 1: 13-Google Scholar
- Minardi G, Manzara C, Creazzo V: Evaluation of 17-mm St. Jude Medical Regent prosthetic aortic heart valves by rest and dobutamine stress echocardiography. J Cardiothorac Surg. 2006, 1: 27-10.1186/1749-8090-1-27.View ArticlePubMedPubMed CentralGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.