- Research article
- Open Access
The ascending aortic image quality and the whole aortic radiation dose of high-pitch dual-source CT angiography
Journal of Cardiothoracic Surgery volume 8, Article number: 228 (2013)
Aortic dissection is a lift-threatening medical emergency associated with high rates of morbidity and mortality. The incidence rate of aortic dissection is estimated at 5 to 30 per 1 million people per year. The prompt and correct diagnosis of aortic dissection is critical. This study was to compare the ascending aortic image quality and the whole aortic radiation dose of high-pitch dual-source CT angiography and conventional dual-source CT angiography.
A total of 110 consecutive patients with suspected aortic dissection and other aortic disorders were randomly divided into two groups. Group A underwent traditional scan mode and Group B underwent high-pitch dual-source CT scan mode. The image quality and radiation dose of two groups were compared.
Close interobserver agreement was found for image quality scores (κ = 0.87). The image quality of ascending aorta was significantly better in the high-pitch group than in the conventional group (2.78 ± 0.46 vs 1.57 ± 0.43, P < 0.001). There was no significant difference of the CT attenuation values, the aortic image noise and SNR between two groups. The mean radiation dose of high-pitch group was also significantly lower than that of conventional group (2.7 ± 0.6 mSv vs. 3.9 ± 0.9 mSv, P < 0.001).
High-pitch dual-source CT angiography of the whole aorta can provide motion-artifact-free imaging of the ascending aorta at a low radiation dose compared to conventional protocol.
With the rapid development of spiral technology and scan speed, the multi-slice spiral CT angiography (CTA) is more widely used for aortic emergencies, and has become the preferred method in aortic diseases [1–8]. However, the images of intimal tear location and intimal flap in type I and type II aortic dissection involving the ascending aorta often cannot satisfy clinical diagnosis because of the proximal segment of the pulsation artifact [9, 10]. In addition, images of the ascending aorta are blurred using 64-slice CT or dual-source CT (DSCT) with earlier technology due to insufficient temporal resolution. On the other hand, due to the long scanning range and the patients’ preoperative assessment and postoperative follow-up for many times, the X-ray radiation hazards are also of concern [11, 12]. How to reduce aortic CTA scanning radiation dose and how to improve image quality of ascending aorta has important meaning for clinical practice.
DSCT is characterized by two X-ray tubes and two corresponding detectors offset by approximately 90°, whose time resolution increases two times compared to single source CT [13, 14]. The second generation DSCT system is equipped with a high-pitch data acquisition mode with pitch values of up to 3.4 by filling the gaps with the data acquired with the second detector system [15, 16]. This high pitch acquisition mode is fast, reducing the scan time for the entire aorta to only about 2 seconds [17, 18] and allowing for low radiation dose. Also, this CT system has improved gantry rotation time up to 0.28 s, which leads to a high temporal resolution of 75 ms for each acquired slice . Because of this fast volume coverage and high temporal resolution, it might be feasible to acquire diagnostic image quality of ascending aorta even without the application of ECG-gating and to reduce radiation dose at the same time. However, this has not been investigated so far. Also, the CTA scan with ECG-gated high-pitch mode always need the heart rate with limit of 70 beats per minute, however, these critically ill patients with aortic dissection always have high heart rate and are not suitable for waiting to control the heart rate. This study was to evaluate the aortic radiation dose and image quality in the ascending aorta using high-pitch mode without ECG-trigger, compared with conventional aortic scan mode of DSCT, aiming to reduce radiation dose and improve the image quality of the ascending aorta at the same time.
One hundred ten consecutively registered patients (74 male, 36 female; age 57.7 ± 11.5 years, range 27~83 years) suspected aortic dissection and other aortic disorders were included during the study time frame (June 2011 through July 2012). They all presented with a sudden severe chest pain with a tearing or ripping sensation. The patients were randomly divided into Group A (n = 55) and Group B (n = 55), and underwent the DSCT aortic CTA examination. General exclusion criteria for contrast-enhanced CTA were nephropathy (defined as a serum creatinine level >150 mol/l) or a known hypersensitivity to iodinecontaining contrast agents. The study protocol was approved by the institutional review board of Xijing Hospital, Fourth Military Medical University, and written informed consent was obtained from all subjects. Also, the study was conducted in accordance with the Declaration of Helsinki.
CT data acquisition
All examinations were performed with a second-generation DSCT system (Somatom Definition Flash, Siemens Healthcare, Germany). Thoracoabdominal scout view and contrast scan were performed from thoracic inlet to the pubic symphysis encompassing the chest. Each patient received an injection of 65 mL of iopromide (Ultravist 370, 370 mg I/mL, Bayer Schering Pharma) at a flow rate of 5 mL/s followed by 30 mL saline solution. A craniocaudal scan direction was chosen, and all CT imaging data were acquired with a breath hold in deep inspiration to eliminate respiratory motion artifacts. In Group A, conventional single tube scan was used with gantry rotation time 0.5 s, pitch 1.2 and a tube current time product of 110 reference mAs/rotation. Contrast administration was controlled with bolus tracking. The region of interest was placed in the aortic root. In Group B, high-pitch CTA scan was used with gantry rotation time 0.28 s, pitch 3.2 and quality reference mAs of 110. Contrast administration also was controlled with bolus tracking. The region of interest was placed in the common iliac artery bifurcation. The automatic trigger scan started 6 seconds after the attenuation threshold of 100 HU in both groups. All the other scanning parameters in both groups were as follows: detector collimation 2 × 64 × 0.6 mm, slice acquisition 2 × 128 × 0.6 mm by means of a z-flying focal spot, gantry rotation time 280 ms, tube voltage 100 kV, matrix 512 × 512, and FOV 200~320 mm. Automatic exposure control, CareDose4D, was used in all exams to adjust scanner output according to patient size.
Postprocessing and image reconstruction
All the original scan data were transmitted to the Siemens workstation (SyngoMMWP VE40A, Siemens Medical Solutions) for image postprocessing. Images were reconstructed using multiplanar reconstruction (MPR) with a slice thickness of 1.0 mm and an increment of 0.8 mm with a medium smooth tissue convolution kernel (B26f).
Evaluation of image quality
Subjective evaluation method
Image quality of ascending aorta was assigned separately by two independent experienced attending radiologists. The reviewers gave the examinations an overall grade according to the following scale: 3, excellent quality without limitation; 2, moderate quality with mild blurring; or 1, significant blurring or doubled appearance of structures . Motion artifacts were assessed for the ascending aorta.
Objective evaluation methods
The aortic CT enhancement values, the aortic image noise and background noise were measured in the region of interest with 30 mm2 in the aortic arch and iliac artery bifurcation, respectively. The signal to noise ratio (SNR) was calculated according to the following equation: SNR = SIaorta /SD, where SIaorta is the aortic CT enhancement values and SD is the background noise.
Evaluation of radiation dose
The volume CT dose index (CTDIvol) and the dose length product (DLP) were recorded, which were measured automaticlly by CT scanner. According to the DLP was used to calculate the effective dose (ED). ED = k × DLP. k coefficient was 0.015 according to the trunk value .
All data analyses were performed with statistical software (SPSS 17.0, SPSS). Quantitative data were expressed as mean ± SD and categorical variables were expressed as frequencies or percentages. T test was used to determine whether there was a significant difference between radiation dose and image noise of two groups. Interobserver agreement on the image quality was evaluated with Cohen’s Kappa statistics. A κ value greater than 0.81 was interpreted as excellent interobserver agreement, values of 0.61-0.80 were interpreted as good, values of 0.41-0.60 as moderate, values of 0.21-0.40 as fair and values less than 0.20 as poor agreement. Pairwise comparison between groups was performed by using the Mann-Whitney paired-sample test. A value of P < 0.05 was considered statistically significant.
Patients and image findings
The final diagnosis of 110 patients was summarized in Table 1. Ninety three patients (59 male, 34 female; mean age: 56.2 ± 11.1 years; range 27 to 81 years) were diagnosed aortic dissection by aortic CTA in this study, including 45 cases in Group A and 48 cases in Group B. They were all divided into 3 types according to DeBakey classification. The intimal tear of Type I and II occured in the the ascending aorta near the sinotubular junction, which coincided with intraoperative findings. In Group A, there were 13 cases of type I aortic dissection, 5 cases of type II and 27 cases of type III. True and false lumen and intimal flap were clearly displayed and the display rate was 100% (45/45) compared to surgical findings. Also, the display rate of aortic dissection tear was 93.3% (42/45). Due to the pulsation artifacts of ascending aortic root, the intimal tear of three cases could not be clearly shown or measured. Moreover, the coronary arteries could not be clearly shown. In Group B, there were 16 cases of type I, 4 cases of type II and 28 cases of type III. The display rate of true and false lumen and intimal flap was also 100% (48/48). The display rate of aortic dissection tear was 97.9% (47/48). Only one intimal tear could not be clearly shown because of the pulsation artifacts of ascending aortic root. Only 18 openings of coronary arteries could be clearly displayed in Group B, in which there were 5 artery dissection of left main trunk and 1 coronary artery dissection of right coronary artery.
Scanning parameters and radiation dose
The basic scanning parameters and radiation dose of two groups were shown in Table 2. With the same basic scanning parameters as Group A, the patients in Group B received less radiation dose and shorter scanning time, which were suitable for emergency and patients who were hard to cooperate with.
Interobserver agreement and image quality
All interobserver agreements for image quality assessment were good (κ = 0.87). No systematic difference between the two observers’ results was found during data analysis. Pulsation artifacts of ascending aorta were significantly reduced in Group B, and there was significant difference of image quality score between two groups (1.57 ± 0.43 vs 2.78 ± 0.46, P < 0.001, Figure 1 and Figure 2).
The aortic CT enhancement values and the aortic image noise were shown in Table 3. There was no significant difference of the CT attenuation values, the aortic image noise and SNR between two groups.
Aortic dissection is the most frequently diagnosed lethal condition of the aorta, especially in the acute setting [22–25]. Prompt and correct diagnosis of aortic dissection is critical for rapid and timely treatment and prognosis, which can decrease mortality and increase the survival rate. Currently, multislice CT imaging is the method of choice for diagnosis of aortic dissection due to its high spatial and temporal resolution with nearly 100% sensitivity and specificity [26–28]. However, when involving the ascending aorta, type I and type II dissection imaging, even with 64 slices spiral CT and earlier DSCT, the pulsation artifact of the root-proximal ascending level cannot be eliminated [9, 10]. Thus, the intimal tear location and intimal flap cannot be clearly displayed and cannot accurately measured. Some aortic type II dissections with limited intimal flap tear and close to the sinus are even misdiagnosed for an artifact because of limited intimal flap.
One recent study by Bolen et al. has shown that imaging of the thoracoabdominal aorta with ECG-triggered high-pitch CTA provides higher quality images of the aortic root and ascending aorta with sufficient contrast enhancement and decreased estimated radiation dose compared with non-ECG-synchronized standard-pitch helical CT . However, the fast CTA scan speed of the whole aorta without the application of ECG-gating is especially important for the difficult patients, such as patients with urgent cardiovascular problems, restless patients and patients who cannot hold their breath for long. In this study, we compared the ascending aortic image quality and the whole aortic radiation dose of high-pitch DSCT angiography without ECG-gating and conventional DSCT angiography. The results of our study showed that the image quality of ascending aorta was much better with high-pitch mode than with the conventional scan mode, though the CTA image with pitch values up to 1.5 and without ECG is often degraded by pulsation artifacts at the aortic root. Recently, Beeres M et al.  reported that high-pitch CTA without ECG synchronization can provide motion-free aortic imaging, which our results totally agreed with. The motion-free imaging of the aorta without ECG synchronization may meet the clinical needs particularly for diagnosis. In addition, with high-pitch scanning mode and the application of the 100 kV low tube voltage, the patients in Group B received the effective radiation dose of only 2.7 ± 0.6 mSv, which was significantly lower than that of conventional protocol in Group A (3.9 ± 0.9 mSv).
Recent studies  showed that high-pitch mode can get coronary artery images with the diagnostic value, however, the heart rate had to be lower than 65 beats per minute and the heart rate variability had to be low. In our study, because of the low radiation dose and the high noise, only some openings of coronary arteries could be clearly displayed in the high-pitch mode group.
The main limitation of the present experiment was that a qualitative image quality scoring system was used in this study, which may be influenced by a subjectivity bias. On the other hand, our image quality kappa value of 0.87 corresponded to a good inter-observer agreement. Moreover, the effective dose calculated as the product of the DLP is based on phantom data and mathematic modeling and does not represent a patient-specific dose. However, effective dose determined in this way provides a reasonable estimate in clinical practice and aids in determining and comparing radiation risk for different CT protocols.
Aortic dissection is the most frequently diagnosed lethal condition of the aorta. It is critical to make a prompt diagnosis of aortic dissection, which can decrease mortality and increase the survival rate. High-pitch scanning mode for DSCT can gain low radiation dose and excellent ascending aortic image quality at the same time in the whole aortic imaging, particularly in the aortic dissection imaging involving the ascending aorta.
Maximum intensity projection
Curved planar reformation
Volume rendering technique
Signal to noise ratio
Volume CT dose index
Dose length product
Fujioka C, Horiguchi J, Kiguchi M, Yamamoto H, Kitagawa T, Ito K: Survey of aorta and coronary arteries with prospective ECG-triggered 100-kV 64-MDCT angiography. AJR Am J Roentgenol. 2009, 193: 227-233. 10.2214/AJR.08.1722.
Tsang W, Bateman MG, Weinert L, Pellegrini G, Mor-Avi V, Sugeng L, Yeung H, Patel AR, Hill AJ, Iaizzo PA, Lang RM: Accuracy of aortic annular measurements obtained from three-dimensional echocardiography, CT and MRI: human in vitro and in vivo studies. Heart. 2012, 98: 1146-1152. 10.1136/heartjnl-2012-302074.
Nayeemuddin M, Pherwani AD, Asquith JR: Imaging and management of complications of open surgical repair of abdominal aortic aneurysms. Clin Radiol. 2012, 67: 802-814. 10.1016/j.crad.2011.12.005.
Duquette AA, Jodoin PM, Bouchot O, Lalande A: 3D segmentation of abdominal aorta from CT-scan and MR images. Comput Med Imaging Graph. 2012, 36: 294-303. 10.1016/j.compmedimag.2011.12.001.
Khan NU, Yonan N: Does preoperative computed tomography reduce the risks associated with re-do cardiac surgery?. Interact Cardiovasc Thorac Surg. 2009, 9: 119-123. 10.1510/icvts.2008.189506.
Singh C, Gupta M, Sharma S: Compression of trachea due to double aortic arch: demonstration by multi-slice CT scan (MSCT). Heart Lung Circ. 2006, 15: 332-333. 10.1016/j.hlc.2006.02.006.
Willoteaux S, Lions C, Gaxotte V, Negaiwi Z, Beregi JP: Imaging of aortic dissection by helical computed tomography (CT). Eur Radiol. 2004, 14: 1999-2008. 10.1007/s00330-004-2441-y.
Robinson SJ, Manns RA: Thoracic aortic iIntra-mural haematoma progression to dissection demonstrated on multi-slice CT imaging: don’t forget the follow-up scan!. Clin Radiol. 2003, 58: 739-741. 10.1016/S0009-9260(03)00228-9.
Smith AD, Schoenhagen P: CT imaging for acute aortic syndrome. Cleve Clin J Med. 2008, 75: 7-9. 10.3949/ccjm.75.1.7. 12, 15-17 passim
Roos JE, Willmann JK, Weishaupt D, Lachat M, Marincek B, Hilfiker PR: Thoracic aorta: motion artifact reduction with retrospective and prospective electrocardiography-assisted multi-detector row CT. Radiology. 2002, 222: 271-277. 10.1148/radiol.2221010481.
Rybicki FJ, Otero HJ, Steigner ML, Vorobiof G, Nallamshetty L, Mitsouras D, Ersoy H, Mather RT, Judy PF, Cai T, Coyner K, Schultz K, Whitmore AG, Di Carli MF: Initial evaluation of coronary images from 320-detector row computed tomography. Int J Cardiovasc Imaging. 2008, 24: 535-546. 10.1007/s10554-008-9308-2.
Sodickson A, Baeyens PF, Andriole KP, Prevedello LM, Nawfel RD, Hanson R, Khorasani R: Recurrent CT, cumulative radiation exposure, and associated radiation-induced cancer risks from CT of adults. Radiology. 2009, 251: 175-184. 10.1148/radiol.2511081296.
Flohr TG, McCollough CH, Bruder H, Petersilka M, Gruber K, Süss C, Grasruck M, Stierstorfer K, Krauss B, Raupach R, Primak AN, Küttner A, Achenbach S, Becker C, Kopp A, Ohnesorge BM: First performance evaluation of a dual-source CT (DSCT) system. Eur Radiol. 2006, 16: 256-268. 10.1007/s00330-005-2919-2.
Matt D, Scheffel H, Leschka S, Flohr TG, Marincek B, Kaufmann PA, Alkadhi H: Dual-source CT coronary angiography: image quality, mean heart rate, and heart rate variability. AJR Am J Roentgenol. 2007, 189: 567-573. 10.2214/AJR.07.2078.
Achenbach S, Marwan M, Schepis T, Pflederer T, Bruder H, Allmendinger T, Petersilka M, Anders K, Lell M, Kuettner A, Ropers D, Daniel WG, Flohr T: High-pitch spiral acquisition: a new scan mode for coronary CT angiography. J Cardiovasc Comput Tomogr. 2009, 3: 117-121. 10.1016/j.jcct.2009.02.008.
Petersilka M, Bruder H, Krauss B, Stierstorfer K, Flohr TG: Technical principles of dual source CT. Eur J Radiol. 2008, 68: 362-368. 10.1016/j.ejrad.2008.08.013.
Goetti R, Baumüller S, Feuchtner G, Stolzmann P, Karlo C, Alkadhi H, Leschka S: High-pitch dual-source CT angiography of the thoracic and abdominal aorta: is simultaneous coronary artery assessment possible?. AJR Am J Roentgenol. 2010, 194: 938-944. 10.2214/AJR.09.3482.
Stinn B, Stolzmann P, Fornaro J, Hibbeln D, Alkadhi H, Wildermuth S, Leschka S: Technical principles of computed tomography in patients with congenital heart disease. Insights Imaging. 2011, 2: 349-356. 10.1007/s13244-011-0088-1.
Karlo C, Leschka S, Goetti RP, Feuchtner G, Desbiolles L, Stolzmann P, Plass A, Falk V, Marincek B, Alkadhi H, Baumüller S: High-pitch dual-source CT angiography of the aortic valve-aortic root complex without ECG-synchronization. Eur Radiol. 2011, 21: 205-212. 10.1007/s00330-010-1907-3.
Bolen MA, Popovic ZB, Tandon N, Flamm SD, Schoenhagen P, Halliburton SS: Image quality, contrast enhancement, and radiation dose of ECG-triggered high-pitch CT versus non-ECG-triggered standard-pitch CT of the thoracoabdominal aorta. AJR Am J Roentgenol. 2012, 198: 931-938. 10.2214/AJR.11.6921.
Bongartz G, Golding SJ, Jurik AG, Leonardi M, Van Persijn Van Meerten E, Rodríguez R, Schneider K, Calzado A, Geleijns J, Jessen KA, Panzer W, Shrimpton PC, Tosi G: European Guidelines for Multislice Computed Tomography. 2004, Brussels, Belgium: European Commission
Verhoeven EL, Paraskevas KI, Oikonomou K, Yazar O, Ritter W, Pfister K, Kasprzak P: Fenestrated and branched stent-grafts to treat post-dissection chronic aortic aneurysms after initial treatment in the acute setting. J Endovasc Ther. 2012, 19: 343-349. 10.1583/12-3860R.1.
Svensson LG, Kouchoukos NT, Miller DC, Bavaria JE, Coselli JS, Curi MA, Eggebrecht H, Elefteriades JA, Erbel R, Gleason TG, Lytle BW, Mitchell RS, Nienaber CA, Roselli EE, Safi HJ, Shemin RJ, Sicard GA, Sundt TM, Szeto WY, Wheatley GH, Society of Thoracic Surgeons Endovascular Surgery Task Force: Expert consensus document on the treatment of descending thoracic aortic disease using endovascular stent-grafts. Ann Thorac Surg. 2008, 85: S1-S41. 10.1016/j.athoracsur.2007.10.099.
Zipfel B, Czerny M, Funovics M, Coppi G, Ferro C, Rousseau H, Berti S, Tealdi DG, Riambau V, Mangialardi N, Sassi C, RESTORE Investigators: Endovascular treatment of patients with types A and B thoracic aortic dissection using Relay thoracic stent-grafts: results from the RESTORE patient registry. J Endovasc Ther. 2011, 18: 131-143. 10.1583/10-3233MR.1.
Tsai TT, Trimarchi S, Nienaber CA: Acute aortic dissection: perspectives from the International Registry of Acute Aortic Dissection (IRAD). Eur J Vasc Endovasc Surg. 2009, 37: 149-159. 10.1016/j.ejvs.2008.11.032.
Chung JW, Park JH, Im JG, Chung MJ, Han MC, Ahn H: Spiral CT angiography of the thoracic aorta. Radiographics. 1996, 16: 811-824. 10.1148/radiographics.16.4.8835973.
Sebastià C, Pallisa E, Quiroga S, Alvarez-Castells A, Dominguez R, Evangelista A: Aortic dissection: diagnosis and follow-up with helical CT. Radiographics. 1999, 19: 45-60. 10.1148/radiographics.19.1.g99ja0945.
Castañer E, Andreu M, Gallardo X, Mata JM, Cabezuelo MA, Pallardó Y: CT in nontraumatic acute thoracic aortic disease: typical and atypical features and complications. Radiographics. 2003, 23: S93-S110. 10.1148/rg.23si035507.
Beeres M, Schell B, Mastragelopoulos A, Herrmann E, Kerl JM, Gruber-Rouh T, Lee C, Siebenhandl P, Bodelle B, Zangos S, Vogl TJ, Jacobi V, Bauer RW: High-pitch dual-source CT angiography of the whole aorta without ECG synchronisation: initial experience. Eur Radiol. 2012, 22 (1): 129-137. 10.1007/s00330-011-2257-5.
We would like to thank Dr. Hongliang Zhao and Dr. Bei E for their help in postprocessing and image reconstruction of the manuscript. The authors received no financial support for the research, authorship, and/or publication of this article.
One of the authors Hongtao Liu is Siemens employee.
YL and MWZ carried out the conception and design. JX and JL carried out the analysis and interpretation. JR, JQX, MQW and YWH participated in the data collection. YL and HTL participated in the writing the article, the critical revision of the article and the statistical analysis. All authors read and approved the final manuscript.
Ying Liu, Jian Xu contributed equally to this work.
About this article
Cite this article
Liu, Y., Xu, J., Li, J. et al. The ascending aortic image quality and the whole aortic radiation dose of high-pitch dual-source CT angiography. J Cardiothorac Surg 8, 228 (2013). https://doi.org/10.1186/1749-8090-8-228
- Radiation dose
- Image quality
- Dual-source CT