- Case report
- Open Access
- Open Peer Review
Successful bridge to recovery using two-stage HeartWare LVAD explantation approach after embolic stroke
© Sabashnikov et al.; licensee BioMed Central Ltd. 2013
- Received: 30 September 2013
- Accepted: 26 December 2013
- Published: 30 December 2013
A two-stage explantation of a left ventricular assist device (LVAD) was performed on 47 year old afro-american gentlemen with non-ischemic dilated cardiomyopathy (DCM) who was successfully bridged to recovery. After he suffered a stroke caused by a VAD thrombosis with embolisation, the VAD outflow graft was first ligated using minimally-invasive approach. Two months later, the device was explanted and a manufactured titanium plug was placed into the sewing ring. This stepwise procedure might be beneficial in cases of high thromboembolic risk and in patients who suffered a thromboembolic event previously.
- Left ventricular assist device (LVAD)
- LVAD thrombosis
- Bridge to recovery
Ventricular assist devices (VADs) provide an established mechanical support of patients with end-stage heart failure as bridge to transplantation, bridge to recovery and chronic support .
However, successful LVAD explantation can only be performed in a restricted number of patients who meet the criteria of myocardial recovery [2, 3]. In this respect, our institution has reported encouraging results with pharmacological therapy combined with continuous-flow or pulsatile-flow LVADs as bridge to recovery [2, 3].
The new miniaturized centrifugal HVAD® Pump (HeartWare Inc, Framingham, MA) is a novel third generation VAD. One of the most significant design advantages of the HeartWare HVAD Pump is its size, intrapericardial placement and versatility in terms of implant options . Due to the miniaturized dimensions it can be implanted over a minimal invasive approach without sternotomy .
Full sternotomy LVAD explantation is shown to be feasible; however it might be associated with increased risk of intraoperative bleeding, need for blood transfusions and postoperative ventricular dysfunction. This problem might be partially resolved by LVAD explantation via minimally invasive lateral thoracotomy . However, in cases of high thromboembolic risk, both approaches may lead to embolic stroke due to manipulation and handling of LVAD during surgery. In this report, we present our experience with staged LVAD explantation in a patient with myocardial recovery who suffered an embolic stroke following LVAD thrombosis. This approach was used with the view to preventing potential flush of clots from the thrombosed LVAD reducing the risk of further potential stroke.
A 47-years-old afro-american gentleman diagnosed with non-ischemic dilated cardiomyopathy (DCM) underwent a continuous-flow LVAD (HeartWare) implantation as a bridge to transplantation. Postoperatively, the device flow was 6.6 L/min, left ventricular ejection fraction (EF) was 16%, and the left ventricular systolic (LVESD) and diastolic (LVEDD) diameters were 63 and 71 mm, respectively. After an uncomplicated postoperative course, the patient was discharged home. Six month later, his cardiac parameters improved significantly (LVESD: 26 mm, LVEDD: 46 mm, EF: 63%). The exercise test according to Modified Bruce Protocol  revealed that the maximum volume of oxygen (VO2 max) on support was almost constant at 2,700 rpm and 1,800 rpm. A transthoracic echocardiogram (TTE) confirmed no serious heart dilatation after 15 min of exercise. Due to these evident signs of myocardial recovery the patient was admitted for an elective LVAD removal.
After two months, the LVAD pump was electively explanted through median sternotomy in off-pump technique. A manufactured titanium plug was placed into the inflow sewing ring which was left in place. Postoperative examination of the pump confirmed a dense thrombus formation with a soft red thrombus cast around the impeller of the device. With a quick postoperative recovery, the patient was discharged home with stable condition, satisfactory cardiac parameters and improving neurological status. From the myocardial recovery point of view, there have been no acute heart failure symptoms, no orthopnea and no peripheral oedema over 6 months follow-up.
The two-stage LVAD explantation in patients with myocardial recovery and high risk of thromboembolic events offers several advantages compared to the conventional approach. The ligature of the VAD outflow graft through a small intercostal incision might prevent thrombotic flush from the VAD which can occur during extensive dissections during conventional LVAD explantation. Also, patients who already suffered an embolic stroke might benefit from additional time for recovery from previous stroke before LVAD is explanted. Furthermore, such a stepwise approach might reduce intra- and postoperative complications, such as right ventricular dysfunction and bleeding with the need for massive transfusions.
In our case, the sewing ring of the LVAD was not removed but was closed using a manufactured titanium plug. As previously described, this approach may reduce the risk of perioperative bleeding and infections . Also, the operating time is shortened due to no need for explantation of the ring and closing the left ventricle. It also preserves the geometry of the left ventricle.
Previous research describes thrombolysis also as a suitable treatment of VAD thrombosis. However, performed in emergency and in the setting of thrombolysis and anticoagulation introduced for VAD thrombosis, it may lead to severe bleeding complications. Furthermore, in our particular case, the VAD explantation was the best option due to recovery of myocardial function.
The risk of perioperative complications might be minimized even more if the VAD explantation is also performed without cardiopulmonary bypass . This technique might reduce the risk of infection, hemodynamic instability, and can be done without complete mobilization of the heart. However, the feasibility of this approach should be considered in individual cases dependent on topographical location and accessibility of the LVAD.
The two-stage VAD explantation is a well reasonable option in cases of high thromboembolic risk and in patients who suffered a thromboembolic event previously.
A written informed consent was obtained from the patient’s next of kin for publication of this case report and accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
None of the authors have any conflicts of interest, disclosures or funding sources.
- Popov AF, Hosseini MT, Zych B, Mohite P, Hards R, Krueger H, Bahrami T, Amrani M, Simon AR: Clinical experience with HeartWare left ventricular assist device in patients with end-stage heart failure. Ann Thorac Surg. 2012, 93 (3): 810-815. 10.1016/j.athoracsur.2011.11.076.View ArticlePubMedGoogle Scholar
- Birks EJ, Tansley PD, Hardy J, George RS, Bowles CT, Burke M, Banner NR, Khaghani A, Yacoub MH: Left ventricular assist device and drug therapy for the reversal of heart failure. N Engl J Med. 2006, 355: 1873-1884. 10.1056/NEJMoa053063.View ArticlePubMedGoogle Scholar
- Birks EJ, George RS, Hedger M, Bahrami T, Wilton P, Bowles CT, Webb C, Bougard R, Amrani M, Yacoub MH, Dreyfus G, Khaghani A: Reversal of severe heart failure with a continuous-flow left ventricular assist device and pharmacological therapy: a prospective study. Circulation. 2011, 123: 381-390. 10.1161/CIRCULATIONAHA.109.933960.View ArticlePubMedGoogle Scholar
- Sabashnikov A, Mohite PN, Simon AR, Popov AF: HeartWare miniaturized intrapericardial ventricular assist device: advantages and adverse events in comparison to contemporary devices. Expert Rev Med Devices. 2013, 10 (4): 441-452. 10.1586/17434440.2013.811851.View ArticlePubMedGoogle Scholar
- Popov AF, Hosseini MT, Zych B, Simon AR, Bahrami T: HeartWare left ventricular assist device implantation through bilateral anterior thoracotomy. Ann Thorac Surg. 2012, 93 (2): 674-676. 10.1016/j.athoracsur.2011.09.055.View ArticlePubMedGoogle Scholar
- GarcíaSáez D, Mohite PN, Zych B, Sabashnikov A, Hards R, Simon AR, Bahrami T: Minimally invasive access for off-pump HeartWare left ventricular assist device explantation. Interact Cardiovasc Thorac Surg. 2013, 17: 581-58. 10.1093/icvts/ivt239.View ArticleGoogle Scholar
- Schweiger M, Potapov E, Vierecke J, Stepanenko A, Hetzer R, Krabatsch T: Expeditious and less traumatic explantation of a heartware LVAD after myocardial recovery. ASAIO J. 2012, 58: 542-544. 10.1097/MAT.0b013e3182640dce.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.