- Research article
- Open Access
- Open Peer Review
Hemodynamic behavior of stentless aortic valves in long term follow-up
© Christ et al.; licensee BioMed Central. 2014
- Received: 25 September 2014
- Accepted: 10 December 2014
- Published: 20 December 2014
Stentless aortic valve replacements show improved hemodynamics due to larger orifice area and lower transvalvular gradients in short and mid-term follow-up. Hemodynamic long-term behavior and the adaptation of the left ventricle as well as valve-durability in patients aged ≤60 years remains unclear.
7 to 16 years after aortic valve replacement, 54 patients (mean age at operation 53.1 ± years) received echocardiography and clinical examination. Mean follow-up time was 10.8 ± 2.2 years. Evaluated were NYHA class, transvalvular gradients, estimated aortic valve orifice area, degree of aortic valve insufficiency, left ventricular mass and function.
At follow-up only one patient presented with NYHA class III. All other patients were in NYHA class I or II. Maximum and mean pressure gradients of the prostheses were 16.3 ± 7.4 mmHg and 9.1 ± 4.2 mmHg, respectively. Compared to echocardiography at discharge the mean pressure gradients dropped 18.0% (2.0 ± 0.9 mmHg) and stayed stable until 14 years after the operation. Only 5 patients showed relevant regurgitation (at 13–16 years after valve replacement), 49 showed no or trivial regurgitation. Left ventricular mass had decreased 26.5% (107.9 ± 18.5 g). Left ventricular ejection fraction (LVEF) had increased in most patients and decreased in only one. For patients with preoperatively impaired left ventricular function an increase of LVEF of 13.1 ± 3.1% was seen.
Porcine stentless aortic valves provide excellent hemodynamic long-term results without significant rise of transvalvular pressure gradients or relevant insufficiencies until 14 years after implantation, leading to sustained decrease of left ventricular mass and improvement of left ventricular function.
- Aortic Valve
- Left Ventricular Mass
- Aortic Valve Replacement
- Hemodynamic Performance
- Transvalvular Gradient
Stentless aortic valve replacements (SAVR) were designed by avoiding an obstructive stent . Due to this difference, SAVR show a larger orifice area, lower transvalvular gradients and improved hemodynamics compared to stented aortic valve replacement . The occurrence of a patient-prosthesis-mismatch could be ruled out with SAVR . These advantages led in comparison to stented bioprostheses to reduced left ventricular mass and improved survival in short and midterm follow-up . It may also lead to improved survival and left ventricular function in long-term follow-up. The main disadvantage of bioprostheses is their limited durability, especially in younger patients. Therefore the American Heart Association and the American College of Cardiology recommend the use of bioprosthetic aortic valve replacements in patients above the age of 60 years, for patients under the age of 60 years the use of mechanical aortic valve replacements is recommended . Nonetheless, in long term follow-up survival between mechanical and stented bioprostheses slightly favored the latter , despite the higher risk of reoperation. The superior hemodynamic properties of SAVR may lead to a higher durability compared to stented bioprostheses. We already reported intermediate to long term results of SAVR in patients under 60 years ,. Yet no long-term follow-up of the hemodynamic performance of SAVR with its implication on durability in younger patients has been published.
The study was approved by the Ethikausschuss institutional review board on the local Ethics Committee.
We identified patients who had received a porcine SAVR in our center 7 to 16 years ago and were 60 years old or younger at the time of operation. Of the identified 64 patients with the originally implanted valve still in place 54 patients gave informed consent to the clinical trial and received clinical examination and echocardiography. Mean age of the patients at operation was 53.1 ± 7.7 years. Mean follow-up time was 10.8 ± 2.2 years. Captured and analyzed were NYHA class, maximum and mean transvalvular pressure gradient, effective aortic valve orifice area (EOA), degree of aortic valve insufficiency, left ventricular mass and left ventricular function. Standard techniques were used to obtain echocardiographic measurements, in accordance with the guidelines of the German Society of Ultrasound in Medicine . Pulsed wave Doppler was used to measure mean and maximum systolic blood flow velocities in the left ventricular outflow tract, and continuous wave Doppler was used to measure systolic blood flow velocities across the aortic valve. Peak and mean transvalvular gradients were obtained using the modified Bernoulli equation. The EOA and left ventricular mass were calculated with the continuity equation  and the ASE equation , respectively.
All data were analyzed with IBM SPSS Statistics version 21 (IBM Corporation). Descriptive statistics are reported as the mean ± standard deviation for continuous variables and as frequencies and percentages for categorical variables, unless otherwise noted. Normal distribution was tested by the Shapiro-Wilk-test. Normally distributed data were analyzed using the Students t-test, while the Wilcoxon test was used for non-normally distributed data. All p values were two-sided. Statistical significance was set at a p value of less than 0.05.
Five different porcine SAVR were implanted in the study population. Implanted valves were: 36 Edwards Prima Plus, 12 St. Jude Medical Toronto SPV™, 3 Vascutek Elan™, 2 Medtronic Freestyle® and 1 Shelhigh stentless bioprosthesis. Implanted valves sizes varied between 25 mm and 29 mm in diameter. Mean valve size was 27.4 ± 1.3 mm.
At follow-up only one patient presented with NYHA class III. All other patients were in NYHA class I or II.
One patient showed a severe regurgitation. The prosthesis was implanted 15 years ago and 3 months after the follow-up a successful reoperation was performed. Additionally, we found 4 cases with moderate insufficiencies. These patients received the SAVR 13–14 years ago and showed no clinical symptoms of the aortic insufficiency. The remaining 49 patients showed no or trivial insufficiencies.
Stentless aortic valves promised excellent hemodynamic behavior mediated by a larger orifice area and a more physiological flow pattern. Short-term and mid-term follow-up, recapitulated in a meta-analysis by Kunadian , confirmed this excellent hemodynamic behavior in comparison to stented bioprostheses. The author showed that the use of a SAVR resulted in a larger effective orifice area and lower transvalvular gradients compared to stented valves. Our long term results showed constantly low transvalvular gradients up to 14 years after implantation. EOAs were calculated 1.6, 1.8 and 2.1 cm2 for 25, 27 and 29 mm valve sizes, respectively. However, captured pressure gradients were lower than for most stented bioprostheses, regardless of intraannular or supraannular implantation. For example, the Edwards Lifesciences Perimount Magna™ and Medtronic Hancock II® showed for the labeled sizes 25 mm, 27 mm, 29 mm EOAs of 1.4 cm2, 1.5 cm2, 1.8 cm2 and 1.3 cm2, 1.4 cm2, 1.8 cm2, respectively . For the stented bioprosthesis St. Jude Medical Trifecta™ lower pressure gradients or higher EOA’s for the 3 different valve sizes have been reported . This newer bioprosthesis, however, still has to stand the test of time.
The hemodynamic properties have an impact on the regression of left ventricular mass and left ventricular remodeling. Left ventricular hypertrophy develops in patients with aortic stenosis as an adaptation to increased pressure load. Incomplete regression of left ventricular hypertrophy after aortic valve replacement is associated with an increased mortality . The observed regression of left ventricular hypertrophy in our study is a consequence of the hemodynamic properties of the SAVR. The remodeling process is also leading to an improvement of left ventricular function, which was observed in our study. Additionally, no relevant insufficiency could be shown till 14 years after operation. This lack of volume overload also supports the left ventricular remodeling.
The limited durability of biological valves and consecutive reoperations are the major concern for the use of bioprostheses in younger patients. Therefore current guidelines recommend the use of mechanical AVR in patients under 60 years . The superior hemodynamic performance of SAVR could lead to a better durability due to less stress on the cusps of the SAVR. The previously published freedom from reoperation with a median of 14.43 ± 0.54 years in our cohort  supports this hypothesis. Literature research showed that most other biological aortic valves in younger patients tend to fail earlier ,. Our data showed an at least similar durability of SAVR compared to stented bioprostheses in younger patients for the first 14 years . However, durability is inferior to mechanical valves. The risk of reoperation in case of bioprostheses must be weighed against the risk of thromboembolic or bleeding events with a mechanical valve . Ruel et al.  showed that these different risks regarding valve replacements led to slightly superior survival in patients with bioprostheses during 25 years follow-up.
Aortic valve repair with various valve sparing techniques is an alternative to valve replacement and often used in younger patients. These procedures may be a useful option in carefully selected patients, mostly with isolated aortic valve regurgitation. Minakata et al.  published reoperation rates of 11% and 15% at 5 and 7 years, respectively. A systematic review by Carr et al.  reported 5 and 10 year freedom from reoperation of 89% and 64%. These data led to concerns about long term durability after aortic valve repair and are inferior to the presented results with stentless bioprostheses.
The impact of the hemodynamic performance of SAVR on patient-survival has been shown in midterm follow-up . Figure 1 shows the long-term survival of our study population. Detailed characteristics of this study cohort were previously published . For patients with isolated aortic valve stenosis and normal left ventricular function survival was comparable to the general population . Additionally, comparison with the literature showed that mortality in our study population was comparable to mechanical or stented biological aortic valves ,. Thus, SAVR led to the same survival during long-term follow-up with better hemodynamic properties.
A limitation is the small study population, especially at 15 and 16 years after the operation. The implantation of SAVR in patients aged 60 years or younger, however, was not the standard procedure and no other long-term follow-up for such a cohort is available. Cardiac sonographers and echocardiography systems have changed over the 16 years of follow-up, which may lead to a bias of data.
SAVR in patients aged 60 years or younger provides excellent hemodynamic long-term results without significant rise of transvalvular pressure gradients or profound regurgitation until 14 years after the operation. Consequently, reoperation rates remained low. The hemodynamic performance led to left ventricular mass regression and improvement of left ventricular function in the study population, particularly in patients with advanced myocardial hypertrophy or poor left ventricular function. Thus, SAVR is an alternative to mechanical valves or valve sparing techniques in younger patients.
The authors gratefully thank Mrs. Katrin Krüger for her assistance in contacting the patients and organizing data.
- Borger MA, Carson SM, Ivanov J, Rao V, Scully HE, Feindel CM, David TE: Stentless aortic valves are hemodynamically superior to stented valves during mid-term follow-up: a large retrospective study. Ann Thorac Surg. 2005, 80: 2180-2185. 10.1016/j.athoracsur.2005.05.055.View ArticlePubMedGoogle Scholar
- Kunadian B, Vijayalakshmi K, Thornley AR, de Belder MA, Hunter S, Kendall S, Graham R, Stewart M, Thambyrajah J, Dunning J: Meta-Analysis of Valve Hemodynamics and Left Ventricular Mass Regression for Stentless Versus Stented Aortic Valves. Ann Thorac Surg. 2007, 84: 73-78. 10.1016/j.athoracsur.2007.02.057.View ArticlePubMedGoogle Scholar
- Pepper J, Cheng D, Stanbridge R, Ferdinand FD, Jamieson WRE, Stelzer P, Berg G, Sani G, Martin J: Stentless Versus Stented Bioprosthetic Aortic Valves: A Consensus Statement of the International Society of Minimally Invasive Cardiothoracic Surgery (ISMICS) 2008. Innov Phila Pa. 2009, 4: 49-60. 10.1097/IMI.0b013e3181a34872.View ArticleGoogle Scholar
- Lehmann S, Walther T, Kempfert J, Leontjev S, Rastan A, Falk V, Mohr FW: Stentless Versus Conventional Xenograft Aortic Valve Replacement: Midterm Results of a Prospectively Randomized Trial. Ann Thorac Surg. 2007, 84: 467-472. 10.1016/j.athoracsur.2007.02.017.View ArticlePubMedGoogle Scholar
- Rahimtoola SH: Choice of prosthetic heart valve in adults an update. J Am Coll Cardiol. 2010, 55: 2413-2426. 10.1016/j.jacc.2009.10.085.View ArticlePubMedGoogle Scholar
- Ruel M, Chan V, Bédard P, Kulik A, Ressler L, Lam BK, Rubens FD, Goldstein W, Hendry PJ, Masters RG, Mesana TG: Very long-term survival implications of heart valve replacement with tissue versus mechanical prostheses in adults <60 years of age. Circulation. 2007, 116 (11 Suppl): I294-I300.PubMedGoogle Scholar
- Christ T, Grubitzsch H, Claus B, Konertz W: Stentless aortic valve replacement in the young patient: long-term results. J Cardiothorac Surg. 2013, 8: 68-10.1186/1749-8090-8-68.View ArticlePubMedPubMed CentralGoogle Scholar
- Christ T, Grubitzsch H, Claus B, Konertz W: Long-term follow-up after aortic valve replacement with Edwards Prima Plus stentless bioprostheses in patients younger than 60 years of age. J Thorac Cardiovasc Surg. 2014, 147: 264-269. 10.1016/j.jtcvs.2012.10.032.View ArticlePubMedGoogle Scholar
- Hagendorff A: Die transthorakale Echokardiografie bei Patienten im Erwachsenenalter – Ablauf einer standardisierten Untersuchung. Ultraschall Med - Eur J Ultrasound. 2008, 29: 344-374. 10.1055/s-2008-1027491.View ArticleGoogle Scholar
- Baumgartner H, Hung J, Bermejo J, Chambers JB, Evangelista A, Griffin BP, Iung B, Otto CM, Pellikka PA, Quiñones M: American Society of Echocardiography, European Association of Echocardiography: Echocardiographic assessment of valve stenosis: EAE/ASE recommendations for clinical practice. J Am Soc Echocardiogr Off Publ Am Soc Echocardiogr. 2009, 22: 1-23. 10.1016/j.echo.2008.11.029. quiz 101–102View ArticleGoogle Scholar
- Lang R, Bierig M, Devereux R, Flachskampf F, Foster E, Pellikka P, Picard M, Roman M, Seward J, Shanewise J: Recommendations for chamber quantification☆. Eur J Echocardiogr. 2006, 7: 79-108. 10.1016/j.euje.2005.12.014.View ArticlePubMedGoogle Scholar
- Borger MA, Nette AF, Maganti M, Feindel CM: Carpentier-Edwards Perimount Magna valve versus Medtronic Hancock II: a matched hemodynamic comparison. Ann Thorac Surg. 2007, 83: 2054-2058. 10.1016/j.athoracsur.2007.02.062.View ArticlePubMedGoogle Scholar
- Bavaria JE, Desai ND, Cheung A, Petracek MR, Groh MA, Borger MA, Schaff HV: The St Jude Medical Trifecta aortic pericardial valve: Results from a global, multicenter, prospective clinical study. J Thorac Cardiovasc Surg. 2014, 147: 590-597. 10.1016/j.jtcvs.2012.12.087.View ArticlePubMedGoogle Scholar
- He GW, Grunkemeier GL, Gately HL, Furnary AP, Starr A: Up to thirty-year survival after aortic valve replacement in the small aortic root. Ann Thorac Surg. 1995, 59: 1056-1062. 10.1016/0003-4975(95)00075-V.View ArticlePubMedGoogle Scholar
- Hammermeister K, Sethi GK, Henderson WG, Grover FL, Oprian C, Rahimtoola SH: Outcomes 15 years after valve replacement with a mechanical versus a bioprosthetic valve: final report of the Veterans Affairs randomized trial. J Am Coll Cardiol. 2000, 36: 1152-10.1016/S0735-1097(00)00834-2.View ArticlePubMedGoogle Scholar
- Minakata K, Schaff HV, Zehr KJ, Dearani JA, Daly RC, Orszulak TA, Puga FJ, Danielson GK: Is repair of aortic valve regurgitation a safe alternative to valve replacement?. J Thorac Cardiovasc Surg. 2004, 127: 645-653. 10.1016/j.jtcvs.2003.09.018.View ArticlePubMedGoogle Scholar
- Carr JA, Savage EB: Aortic valve repair for aortic insufficiency in adults: a contemporary review and comparison with replacement techniques. Eur J Cardiothorac Surg. 2004, 25: 6-15. 10.1016/j.ejcts.2003.09.018.View ArticlePubMedGoogle Scholar