- Research article
- Open Access
- Open Peer Review
Fifteen-year experience with the Bicarbon heart valve prosthesis in a single center
© Misawa et al. 2015
- Received: 12 February 2015
- Accepted: 19 June 2015
- Published: 28 June 2015
The purpose of this study was to evaluate retrospectively the clinical performance of the Bicarbon valve (Sorin Biomedica Cardio, Saluggia, Italy) implanted at our center in Japan.
Between January 1997 and December 2011, 415 patients in our institution were implanted with the Bicarbon valve. Nine of these recipients were excluded from the study because they had already undergone valve implantation and received a Bicarbon valve in a different position. The remaining patients were analyzed for evaluation of the postoperative clinical outcomes. Of the 406 patients (mean age 60.2 ± 11.7 years), 179 underwent aortic valve replacement (AVR), 149 mitral valve replacement (MVR), and 78 both aortic and mitral valve replacement (DVR).
There were 10 early deaths (2.5 %: 4 in the AVR group and 6 in the MVR group). Three hundred eighty-nine patients were followed up (95.8 % completeness of follow-up) with a mean follow-up of 6.6 ± 4.2 years overall (AVR 6.8 ± 4.2, MVR, 6.7 ± 4.4, and DVR 5.7 ± 3.4 years) and a cumulative follow-up of 2661 patient-years (1214, 1001, and 446 patient-years for AVR, MVR, and DVR, respectively). Ninety-nine patients died (3.7 % per patient-year: 22 valve-related and 77 valve-unrelated deaths). Survival at 10 years was 74.1 ± 4.0 % in the AVR group, 73.7 ± 4.2 % in the MVR group, and 61.0 ± 7.9 % in the DVR group. The linearized incidence of thromboembolic complications, bleeding complications, prosthetic valve endocarditis, paravalvular leaks, and sudden death in all patients was 0.5 %, 0.5 %, 0.2 %, 0.2 %, and 0.4 % per patient-year, respectively. The incidence of valve-related complications and reoperation was 1.6 % and 0.4 %, respectively. No other valve-related complications were observed.
The Bicarbon prosthetic heart valve has shown excellent clinical results and is associated with a low incidence of valve-related complications.
- Bicarbon valve
- Heart valve disease
- Valve replacement
- Valve-related complication
- Bileaflet prosthetic valve
- Heart valve surgery
Many types of tissue and mechanical heart valve prostheses are commercially available worldwide. Lifelong anticoagulant therapy is inevitable for patients with mechanical prosthetic valves, and those with tissue valves have higher risks of structural valve dysfunction than those with mechanical ones. The purpose of this study was to evaluate retrospectively the clinical performance of the Bicarbon valve (Sorin Biomedica Cardio, Saluggia, Italy) implanted at our center in Japan.
Etiology of valve diseases
aortic valve (cases)
mmitral valve (cases)
prosthetic valve dysfunction
Distribution of implanted valve sizes
1) Aortic position
2) Mitral position
coronary artery bypass grafting
tricuspid valve annuloplasty
The mid-term results of 105 patients with a Bicarbon valve as of the end of January 2000 and the prospective follow-up study of the same patients at the end of December 2005 have already been reported [1, 2]. Clinical data from additional patients was evaluated on the basis of mortality and morbidity analysis.
Operative procedures and postoperative management have been described in the previous study . The international normalization ratio (INR) was controlled to maintain values between 1.8 and 3.3 for patients having MVR and DVR, and was between 1.3 and 1.8 for AVR patients before 2000. Since then, the INR has been between 1.8 and 3.0 for all patients. Postoperative anticoagulation therapy did not differ with regard to rhythm status. An antiplatelet agent such as dipyridamole or aspirin was added to the anticoagulation therapy.
As of the end of June 2012, 406 patients participated in the follow-up study through an office interview, personal phone call, or information from patients’ family physicians. Seventeen patients were lost to follow-up (95.8 % completeness of follow-up) with a mean follow-up of 6.6 ± 4.2 years overall (AVR, 6.8 ± 4.2 years; MVR, 6.7 ± 4.4 years; DVR, 5.7 ± 3.4 years) and a cumulative follow-up of 2661 patient-years (1214, 1001, and 446 patient-years for AVR, MVR, and DVR, respectively).
Analyses were performed overall on the implanted population and stratified according to implant site (aortic, mitral, and both aortic and mitral). Morbidity analysis included all cardiovascular complications as defined by Edmunds et al. . Mortality data (cumulative survival) and incidence of clinical adverse events (freedom from events) were analyzed using the Kaplan-Meier actuarial method and 95 % confidence interval (CI)) by implant site.
Valve-related complications included thrombosis, embolism, anticoagulant-related bleeding, endocarditis, and valve dysfunction, and are presented as linearized rates. For each linearized rate, upper confidence limits (95 % CI) are provided, according to the method reported by Grunkemeier and Anderson . For continuous variables, descriptive statistics (mean, standard deviation, range) are provided. Qualitative variables are presented as number of patients and occurrence. All analyses were performed using StatView v.5 (SAS Institute Inc., Cary, NC, USA).
Survival and clinical functional class
Preoperatively, 66.6 % of the patients were in the New York Heart Association (NYHA) functional class III or IV, and 95.0 % of the surviving patients were in class I or II at follow-up.
Linearized rates and upper confidence limits of events
The linearized rates and 95 % CI of each event overall were 0.5 % (0.3–0.9 %) per patient-year for thromboembolic events, 0.5 % (0.3–0.9 %) per patient-year for bleeding, 0.2 % (0.1–0.4 %) per patient-year for prosthetic endocarditis, 0.2 % (0.1–0.4 %) per patient-year for paravalvular leak, 0.4 % (0.2–0.7 %) per patient-year for sudden death, and 1.4 % (0.8–2.5 %) per patient-year for all valve-related complications.
Linearized rates of the AVR group were 0.5 % (95 % CI: 0.2–1.0 %) per patient-year for thromboembolic events, 0.5 % (0.2–1.0 %) per patient-year for bleeding, and 1.6 % (1.1–2.7 %) per patient-year for all valve-related complications. Those of the MVR group were 0.4 % (0.1–1.0 %) for thromboembolic events, 0.4 % (0.1–1.0 %) for bleeding, and 1.6 % (0.8–3.2 %) for all valve-related complications. Those of the DVR group were 0.5 % (0.1–1.6 %) for thromboembolic events, 0.7 % (0.2–1.9 %) for bleeding, and 1.6 % (0.6–3.5 %) for all valve-related complications.
Tissue valves are more likely than mechanical valves to be implanted in Japan, with 11,656 tissue and 5050 mechanical heart valves implanted in 2012 . Prolonged durability of the tissue valves, an aging population, and an increasing number of plastic surgery techniques have led to a decrease in the use of mechanical valves. Nevertheless, mechanical valves appear to be a better option for some patients. At our institution we choose a mechanical valve for young adults and patients with end-stage renal disease because of the rapid structural dysfunction of tissue valves .
Excellent early and mid-term clinical results of the Bicarbon valve have been previously reported [1, 7–11]. We also reported early and mid-term results with the prosthetic valve in 2002 and 2007. Only long-term follow-up studies, however, will be sufficient enough to properly evaluate its clinical performance. Here we report our 15-year clinical experience with the Bicarbon valve.
With regard to single-center mid-term results, Goldsmith and colleagues reported in 1998 that the Bicarbon valve has a satisfactory clinical performance with low complication rates . A 2004 multicenter study of the Bicarbon valve in Europe with a mean follow-up of 2.2 ± 1.5 years showed rates of 5 % early death and 4.4 % late death . In addition, the linearized incidence of valve thrombosis was between 0.06 % and 0.69 % per patient-year among the AVR, MVR, and DVR groups, and that of embolic episodes was between 1.13 % and 2.14 % per patient-year. Bleeding complications occurred at a rate of 0.69–1.26 % per patient-year.
Other studies have shown that freedom from valve thrombosis at 7–9 years was between 97 % and 99.4 %, freedom from embolic episodes was between 64 % and 93 %, and freedom from bleeding complications was between 82 % and 98.6 % [9–11]. Actuarial analysis at 7–9 years showed an overall survival of between 63.9 % and 88 %.
Regarding long-term multicenter clinical results, Azarnoush and colleagues reported actuarial survival at 15 years of 61.4 % for AVR, 63.4 % for MVR, and 56.4 % for DVR . They added that actuarial freedom from thromboembolism, anticoagulant-related hemorrhage, and endocarditis at 15 years was 88.8 %, 77.5 %, and 96.8 %, respectively, without any cases of structural failure of the prosthesis.
Long-term clinical experience with the St. Jude Medical and Carbomedics bileaflet mechanical valves for AVR and MVR have shown rates of thrombosis of between 0.73 % and 3.4 % per patient-year, and 10-year freedom from thrombosis of between 77 % and 94.2 % [12–21]. In addition, the rates of bleeding were between 0.52 % and 2.7 % per patient-year, and the 10-year freedom from bleeding ranged from 77 % to 96.4 %. Our previous reports concluded that the results obtained were similar to those for other mechanical valves associated with a low incidence of valve-related morbidity and mortality [1, 2].
Jeong and colleagues reported long-term clinical results with the ATS mechanical valve in 1382 consecutive patients, revealing that the survival rates at 12 years were 87.0 ± 3.8 % in the AVR group and 71.4 ± 6.3 % in the DVR group . In addition, there was a higher cardiovascular event-free survival at 12 years in the AVR group than the DVR group (82.3 ± 4.7 % versus 65.1 ± 7.3 %). Another long-term study was conducted by Sezai, who reported bleeding events at a rate of 0.19 % and thromboembolic events of 0.44 % per patient-year in 231 cases over 15 years of observation .
In the present era, a bileaflet valve is more likely than a tilting disc valve to be implanted. In 2001 we revealed our 14-year experience with the Omnicarbon tilting disc valve in 168 consecutive patients of mean age 53 years . Freedom from thromboembolism at 10 years was 94 % in the AVR group, 80 % in the MVR group, and 96 % in the DVR group, and that from hemorrhagic complications was 86 %, 92 %, and 100 %, respectively. The Ominicarbon valve showed good clinical performances similar to the Bicarbon valve. The increasing age of patients with a narrow aortic valve annulus and posterior leaflet preservation techniques in MVR have led us to choose bileaflet mechanical valves . However, one must be aware of the possible risk of the valve becoming stuck because of the posterior orientation of the major orifice.
Although structural dysfunction has been overcome, valve-related complications after heart valve replacement with mechanical valves occur at unacceptable rates. Thromboembolic and hemorrhagic events related to anticoagulant therapy should be considered during lifelong follow-up. Non-structural prosthetic valve dysfunctions such as paravalvular leak and pannus ingrowth are also issues to be resolved .
The present study gives additional evidence of low rates of valve-related complications after Bicarbon valve implantation. Our anticoagulation therapy is associated with an antiplatelet agent. Patients with INR >3.0 sometimes develop lethal bleeding complications, so we maintain the INR between 1.8 and 3.0. The rate of thromboembolic events in this study is excellent and the rates of bleeding complications are also acceptable. We confirm that anticoagulation for patients with a mechanical valve should be implemented in association with an antiplatelet agent. Postoperative NYHA functional improvement was satisfactory in all groups. Low cardiac output syndrome led to lethal complications in those patients with advanced stage IV NYHA classification. To obtain better clinical outcomes, an operation before advanced functional class is reached should be considered, and more intensive perioperative management may be mandatory for patients with heart failure that is far advanced.
Many patients have been followed by their family physicians for several months after surgery for reasons of patients’ convenience. Some visit our outpatient clinic once or twice a year, while others do not. We sometimes experience poorly controlled anticoagulant therapy among such patients, which might lead to thromboembolic and bleeding complications. Unfortunately, we were unable to evaluate the INR of some patients suffering from thromboembolic and/or bleeding events. In addition, deaths from unknown causes are included in the category of sudden death, meaning that the rate of cases of sudden death may be higher than its true incidence.
This single-center study of a 15-year follow-up of the Bicarbon prosthetic heart valve shows excellent clinical results associated with a low incidence of valve-related mortality and morbidity.
Ms. Ueno E and Kobayashi N contributed to acquisition of data and analysis of the data.
- Misawa Y, Saito T, Konishi H, et al. Single-center experience with the Bicarbon bileaflet prosthetic heart valve in Japan. Artif Organs. 2002;26:856–61.View ArticlePubMedGoogle Scholar
- Misawa Y, Saito T, Konishi H, Ohki S, Kaminishi Y, Sakano Y, et al. Clinical experience with the Bicarbon heart valve prosthesis. J Cardiothorac Surg. 2007;2:8.View ArticlePubMedPubMed CentralGoogle Scholar
- Edmunds LH, Clark RE, Cohn LH, Grunkemeier GL, Miller DC, Weisel RD. Guideline for reporting morbidity and mortality after cardiac valvular operations. Ann Thorac Surg. 1996;62:932–5.View ArticlePubMedGoogle Scholar
- Grunkemeier GL, Anderson RP, Starr A. Actuarial and actual analysis of surgical results: empirical validation. Ann Thorac Surg. 2001;71:1885–7.View ArticlePubMedGoogle Scholar
- Masuda M, Kuwano H, Okumura M, Amano J, Arai H, Endo S, et al. Thoracic and cardiovascular surgery in Japan during 2012. Gen Thorac Cardiovasc Surg. 2014;62:734–64.View ArticlePubMedPubMed CentralGoogle Scholar
- Misawa Y. Heart valve replacement for patients with end-stage renal disease in Japan. Ann Thorac Cardiovasc Surg. 2010;16:4–8.PubMedGoogle Scholar
- Goldsmith I, Lip GY, Patel RL. Evaluation of the Sorin Bicarbon bileaflet valve in 488 patients (519 prostheses). Am J Cardiol. 1999;83:1069–74.View ArticlePubMedGoogle Scholar
- Borman JB, Brands WGB, Camilleri L, et al. Bicarbon valve-European multicenter clinical evaluation. Eur J Cardiothorac Surg. 1998;13:685–93.View ArticlePubMedGoogle Scholar
- Whitaker DC, James SE, Walesby RK. A single-center experience of the Sorin Bicarbon heart valve prosthesis: long-term clinical, hematological and hemodynamic results. J Heart Valve Dis. 2004;13:97–102.PubMedGoogle Scholar
- Borman JB, De Riberolles C. Sorin Bicarbon bileaflet valve: a 10-year experience. Eur J Cardiothorac Surg. 2003;23:86–92.View ArticlePubMedGoogle Scholar
- Bortolotti U, Milano A, D'Alfonso A, et al. Evaluation of valve-related complications in patients with Sorin Bicarbon prosthesis: a seven-year experience. J Heart Valve Dis. 2001;10:795–801.PubMedGoogle Scholar
- Azarnoush K, Laborde F, de Riberolles C. The Sorin Bicarbon over 15 years clinical outcomes: multicenter experience in 1704 patients. Eur J Cardiothorac Surg. 2010;38:759–66.View ArticlePubMedGoogle Scholar
- Lund O, Nielsen SL, Arildsen H, Ilkjaer LB, Pilegaard HK. Standard aortic St. Jude valve at 18 years: performance profile and determinants of outcome. Ann Thorac Surg. 2000;69:1459–65.View ArticlePubMedGoogle Scholar
- Ikonomidis JS, Kratz JM, Crumbley 3rd AJ, et al. Twenty-year experience with the St Jude Medical mechanical valve prosthesis. J Thorac Cardiovasc Surg. 2003;126:2022–31.View ArticlePubMedGoogle Scholar
- Emery RW, Krogh CC, Arom KV, et al. The St. Jude Medical cardiac valve prosthesis: a 25-year experience with single valve replacement. Ann Thorac Surg. 2005;79:776–82.View ArticlePubMedGoogle Scholar
- Tominaga R, Kurisu K, Ochiai Y, et al. A 10-year experience with the Carbomedics cardiac prosthesis. Ann Thorac Surg. 2005;79:784–9.View ArticlePubMedGoogle Scholar
- Kang CH, Ahn H, Kim KH, Kim KB. Long-term result of 1144 CarboMedics mechanical valve implantations. Ann Thorac Surg. 2005;79:1939–44.View ArticlePubMedGoogle Scholar
- Aagaard J, Tingleff J. Fifteen years’ clinical experience with the CarboMedics prosthetic heart valve. J Heart Valve Dis. 2005;14:82–8.PubMedGoogle Scholar
- Bryan AJ, Rogers CA, Bayliss K, Wild J, Angelini GD. Prospective randomized comparison of CarboMedics and St. Jude Medical bileaflet mechanical heart valve prostheses: ten year follow-up. J Thorac Cardiovasc Surg. 2007;133:614–22.View ArticlePubMedGoogle Scholar
- Toole JM, Stroud MR, Kratz JM, Crumbley AJ, Bradley SM, Crawford Jr FA, et al. Twenty-five year experience with the St. Jude Medical mechanical valve prosthesis. Ann Thorac Surg. 2010;89:1402–9.View ArticlePubMedGoogle Scholar
- Jeong DS, Kim WS, Sung K, Yang JH, Jun TG, Lee YT, et al. Long-term hemodynamic performance of ATS valves in the aortic position: impact on the progression of late tricuspid regurgitation. J Heart Valve Dis. 2013;22:794–803.PubMedGoogle Scholar
- Sezai A, Hata M, Niino YI, Kasamaki Y, Hirayama A, Minami K. Fiftenn years of experience with ATS mechanical heart valve prostheses. J Thorac Cardiovasc Surg. 2010;139:1494–500.View ArticlePubMedGoogle Scholar
- Misawa Y, Fuse K, Saito T, Konishi H, Oki S. Fourteen year experience with the Ominicarbon prosthetic heart valve. ASAIO J. 2001;47:677–82.View ArticlePubMedGoogle Scholar
- Kaminishi Y, Misawa Y, Kobayashi J, Konishi H, Miyata H, Motomura N, et al. Patient-Prosthesis Mismatch in Patients with Aortic Valve Replacement. Gen Thorac Cardiovasc Surg. 2013;61:274–9.View ArticlePubMedPubMed CentralGoogle Scholar
- Misawa Y. Valve-related complications after mechanical heart valve implantation. Surg Today. 2014. doi: 10.1007/s00595-014-1104-0.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.