- Research article
- Open Access
Albumin-glutaraldehyde glue for repair of superficial lung defect: an in vitro experiment
Journal of Cardiothoracic Surgery volume 11, Article number: 63 (2016)
Albumin-glutaraldehyde glue gained a widespread acceptance in repair of superficial lung defects associated with alveolar air leaks (AAL). As its sealing efficacy has not yet been thoroughly corroborated by clinical studies, we sought to assess the properties of commercially available albumin-glutaraldehyde glue (BioGlue™) in an in vitro lung model.
The lower lobe of freshly excised swine lung (n = 10) was intubated and ventilated. A focal superficial parenchymal defect (40 × 25 mm) was created on the inflated lung. AAL was assessed with increasing inspired tidal volume (TVi). After glue application, AAL was assessed until burst failure occurred. To evaluate glue elasticity, the length of defect was recorded in the inflated lung.
Superficial parenchymal defects resulted in AAL increasing with ascending TVi. Multiple linear regression analysis revealed strong correlation between AAL and maximal inspiratory pressure. There was one application error. At TVi = 400, 500, 600, 700, 800 and 900 ml, BioGlue™ achieved complete sealing in nine, six, five, four two and one specimens, respectively. Mean burst pressure was 38.0 ± 4.2 cmH2O. All sealant failures were cohesive. BioGlue™ allowed an expansion of covered lung defects of 1.5 ± 1.7 mm.
Our in vitro tests demonstrated a high sealing efficacy of BioGlue™ for repair of superficial lung defects. Due to the rigid nature, caution should be taken to use this kind of sealant in trapped lungs.
Die Wirksamkeit von chirurgischen Klebstoffen zur Behandlung der alveolo-pleuralem Luftleck (AAL) ist immer noch kontrovers diskutiert. Wir evaluierten in dieser Studie die Abdichtungswirksamkeit von einem Albumin-Glutaraldehyd-Kleber BioGlue™ mittels eines in vitro Lungenmodels.
Der Unterlappen von frisch entnommenen Schweinlungen (n = 10) wurde intubiert und beatmet. Eine pleurale Läsion (40 × 25 mm) wurde erstellt und APL mit steigendem inspiratorischem Tidalvolumen (TVi) untersucht. Nach Applikation von BioGlue™ wurde AAL auf die gleiche Weise gemessen bis zur Auftritt von Kleberbruch. Zur Untersuchung der Elastizität des Klebers wurde die Länge der pleuralen Läsion gemessen.
Die oberflächliche pleurale Läsion führte bei aufsteigendem TVi zum Anstieg von AAL. Multiple lineare Regressionsanalyse ergab eine starke Korrelation zwischen AAL und dem maximalen inspiratorischen Druck. Applikationsfehler trat bei einem Test auf. Bei TVi = 400, 500, 600, 700, 800 und 900 ml führte BioGlue™ zur kompletten Versieglung in neun, sechs, fünf, vier, zwei und einem Test. Der mittlere Bruchdruck betrug 38.0 ± 4.2 cmH2O. Es wurde nur kohäsiver Kleberbruch gefunden. Nach der Kleberapplikation hat sich die Länge der pleuralen Läsion um 1.5 ± 1.7 mm gedehnt.
Unsere in vitro Versuche zeigten eine zuverlässige Versiegelung der oberflächlichen Lungendefekte durch BioGlue™. Aufgrund der Streifigkeit des Klebers ist Vorsicht bei der Verwendung derartigen Klebers an gefesselter Lunge geboten.
Superficial lung defect is a common intraoperative complication of lung surgery, particularly following pleural decortication, dissection of firm pleural adhesions and division of incomplete fissures [1–3]. It results in alveolar air leaks (AAL) associated with delayed removal of chest tubes, prolonged hospital stays, increased postoperative morbidity and patient discomfort [4, 5]. In the past decade surgical sealants have been increasingly used in treating AAL as adjuncts to conventional closing techniques including suturing and stapling. The commonly used sealants are human thrombin-fibrinogen sponge (TachoSil®; Takeda Pharmaceutical Company Limited, Osaka, Japan), albumin-glutaraldehyde glue (BioGlue™; CryoLife Europa Ltd., Surrey, UK) and synthetic polymer sealants (TissuePatch™, Tissuemed Ltd, Leeds, UK; CoSeal®, Baxter Healthcare, Fremont, CA) [5, 6]. However, a body of clinical trials assessing sealant efficacy yielded inconsistent results, thus raising questions about evidence-based sealant use in lung surgery [5–7]. This might be attributed to a number of factors like small sample sizes, inherent flaws in study design or heterogeneity of clinical characteristics in study and control groups [5, 8].
Bioglue™ (CryoLife Europa Ltd., Surrey, UK) is a two-component sealant composed of bovine serum albumin and glutaraldehyde . The aldehyde groups of glutaraldehyde react with the amine groups of bovine serum albumin and those present in the extracellular matrix and cell surface, resulting in strong cross-linking after 2–3 min. Bioglue™ has been widely used in cardiovascular surgery due to its high hemostatic efficacy [10–12]. In recent years BioGlue™ gained widespread acceptance as an adjunct in treating AAL in lung surgery . So far, the pneumostatic efficacy of BioGlue™ has not been thoroughly evaluated. In the present study we sought to examine sealing efficacy and elasticity of BioGlue™ for repair of superficial lung defects by means of an established in vitro lung model .
Lungs of German Landrace pigs were freshly excised in a local slaughterhouse. Within two hours after harvest, the lungs were dissected along the trachea until the tracheal bifurcation was reached. The lower lobe was selectively intubated, ventilated and immersed in water to ensure impermeability. After being connected to the ventilation machine (Evita, Dräger, Lübeck, Germany), the lower lobe was ventilated in volume-controlled mode with a PEEP of 5 cmH2O, an I:E ratio of 1:2 and a frequency of 12/min. The lower lobe was fully inflated when inspiratory tidal volume (TVi) was ≥ 400 ml. Over-inflation of the lobe was observed with TVi ≥ 800 ml. To create a standardized superficial parenchymal lesion, a rectangle measuring 40 × 25 mm was marked on the fully inflated lower lobe with a marker pen. Using a small, conic headed drill, a defect was created by carefully applying pressure to the marked site, starting at the edges and advancing toward the middle of the designated area. Marker spots were then added to the cranial and caudal edge of the lesion (Fig. 1). TVi was increased in steps, recording respiratory parameters including expiratory tidal volume (TVe), maximal inspiratory pressure (Pmax), plateau pressure (Pplat), resistance and compliance for each subsequent step until a Pmax of 40 cmH2O was reached. AAL was calculated as difference between TVi and TVe.
Thereafter, BioGlue™ was applied to the lesion by carefully meandering along the surface, in compliance with user guidelines, and respecting a safety margin of one cm to all sides (Fig. 2). In the present experiment BioGlue™ was applied exclusively in two mL pre-filled syringes, using only one sample for each lesion. After 60 s full sealant adhesion was achieved and the glue hardened. The lower lobe was then ventilated again with TVi rising slowly from 100 ml. Commencing at 400 ml TVi the same ventilation parameters as before were recorded. AAL at the site of pleural defect were assessed by submersion tests and graded according to Macchiarini scale as grade 0 (no leak), grade 1 (countable bubbles), grade 2 (stream of bubbles) and grade 3 (coalescent bubbles) . Sealing was considered successful, if no bubbles were observed (Macchiarini grade 0). The distance between both marker spots was measured to evaluate sealant elasticity. In case of sealant failure Pmax was recorded as burst pressure (BP). Pmax tolerated at the TVi, at which sealing was last achieved, was considered maximally tolerated pressure (MTP). Burst failure was furthermore categorized into adhesive or cohesive failure. Adhesive failure occurred at the interface between sealant and parenchymal defect. Cohesive failure was defined as failure within the sealant. Application error occurred, when cohesive or adhesive failure was observed before starting the test at 400 ml TVi. Two independent investigators noted all results. A third investigator would arbitrate any disagreement.
Finally, lung specimen containing the parenchymal lesion along with attached sealant were resected and fixed with glutaraldehyde. The specimen were embedded in paraffin and processed to obtain sections for haematoxylin-eosin staining.
Normality of variables was tested using the Kolmogorov-Smirnov one-sample-test. Descriptive statistics are presented as mean ± standard deviation in case of normal distribution. Multiple linear regression was used to determine the ventilation parameters’ correlation with AAL. Statistical significance was assumed if p < 0.05. All statistical analysis was performed using SPSS (version 16.0 for Windows; SPSS Inc., Chicago, Illinois, USA).
A total of five pilot tests were performed prior to starting the experiment to ensure proper application of sealant and assessment of sealing efficacy. The results of these tests were not included in the data set or statistical analysis. Thereafter ten consecutive tests were undertaken. In the assessment before sealant application AAL increased with ascending TVi. Multiple linear regression analysis revealed strong correlation between AAL and Pmax (p < 0.001).
We observed one application error. Following sealant application, nine lungs were still considered airtight at TVi of 400 ml, six at TVi of 500 ml, seven at TVi of 600 ml and four at TVi of 700 ml. Two lungs remained sealed even at TVi of 800 ml, with one lung remaining sealed until 1000 mL TVi. Graphic interpretation of AAL with ascending TVi is presented in Fig. 3. Mean BP was 38.0 ± 4.2 cmH2O and mean MTP was 36.3 ± 6.1 cmH2O. All burst failures were cohesive. BioGlue™ allowed an average length expansion of 1.5 ± 1.7 mm in relation to length at TVi of 400 ml.
Superficial lesions of lung parenchyma occur regularly in lung surgery, especially in pleural decortation, dissection of firm pleural adhesions and division of incomplete fissures. One of the consequences is AAL resulting in delayed removal of chest tubes, prolonged hospital stays as well as higher postoperative morbidity [1, 3, 15]. The utility of surgical sealants as adjunct to conventional sealing techniques has been progressively recognized in lung surgery over the past decade [16, 17].
BioGlue™ is a two-component sealant consisting of bovine serum albumin and glutaraldehyde, and has been widely used for achieving haemostasis in cardiovascular surgery. In the recent past BioGlue™ has also seen an increase in use for repair of superficial lung defects. However, its true sealing efficacy remains unclear, which is at least partly due to inconsistent results of clinical trials. In a prospective, randomized controlled trial, Tansley et al. examined 52 patients with post-thoracotomy AAL and found significant differences in duration of intercostal drainage for patients treated with BioGlue™ over those in whom only surgical closing techniques were used . However, this advantage could not be confirmed in another multicenter prospective randomized clinical trial, in which Allen and co-workers randomized 161 patients in a 2:1 ratio to receive topical application of BioGlue™ or control for at least one significant AAL following lung resection . They found no significant difference in the duration to chest tube removal.
The present in vitro experiment aimed to test the sealing efficacy of BioGlue™ using an established lung model, which simulated a real scenario in lung surgery . The results showed a high efficacy of BioGlue™ in treating AAL. BioGlue™ withstood very high inspiratory pressures, as reflected by the mean BP of 38.0 cmH2O – a respiratory pressure higher than in routine clinical settings. This suggests that BioGlue™ might be a useful tool for treatment of AAL, particularly in patients requiring aggressive mechanical ventilation. Albeit, the overall clinical benefits of BioGlue™ need to be further evaluated by well-designed prospective, randomized clinical trials. The authors are going to launch an experiment to compare BioGlue™ with other commonly used surgical sealants regarding sealing efficacy by means of the present in vitro lung model.
Additionally, our experiment analysed elastic properties of BioGlue™, which seem to comprise this sealant’s main weakness. After application of BioGlue™ only very little elasticity remained in the affected lung tissue, which might be of concern in patients with already trapped lung. Beside local effects, sealant run-off during application and ensuing restriction after glue hardening can further augment its rigidity. It appears that strong cross-links between albumin and glutaraldehyde, which ensure the great durability of BioGlue™, result in marked glue rigidity. This characteristic was previously mentioned by Belcher et al., where the sealant is described as having “a rigid, inelastic nature that does not expand with the underlying lung”, possibly increasing the risk for inflammation . A study by Azadani et al. on mechanical properties of surgical glues revealed that BioGlue™ had a far lower compliance than comparable products . Our tests provided quantitative data in this regard and could contribute to a deeper understanding of biomechanical properties of BioGlue™.
Concerning the bovine serum contained in Bioglue™, the risk of blood borne diseases exists for treated patients. Other reservations about the sealant concern toxicity of the glutaraldehyde component, as topified by Fürst et al. in a combined in vitro/in vivo experiment . Here a cytotoxic effect of glutaraldehyde was found in vitro for human and mouse growth cells and an inflammation of the liver and aorta in vivo, suggesting BioGlue™ be used with caution.
Possible limitations of this in vitro trial include a certain inevitable variation in parenchymal lesion size. We minimized these potential confounders by using lungs from pigs of equal weight (around 80 kg) and size and by inducing standardized lesions. As all lower lobes were fully inflated with a TVi of 400 ml, no differences were noted in this regard. Judgment of sealing and measurement of lesion length might be prone to information bias. To counteract this, measurements were undertaken by two independent investigators and arbitrated by a third, if necessary. In the end our in vitro tests deliver quantitative results that may contribute to further understanding and use of BioGlue™ as adjunct therapy for AAL.
Our in vitro tests suggest that BioGlue™ might be a useful adjunct in treating AAL, especially in patients requiring high ventilation pressures. Caution should be advised for the utility of this sealant in patients with trapped lung due to its rigidity.
alveolar air leak
maximally tolerated pressure
positive end-expiratory pressure
maximal inspiratory pressure
mean inspiratory pressure
plateau inspiratory pressure
expiratory tidal volume
inspired tidal volume.
Abolhoda A, Liu D, Brooks A, Burt M. Prolonged air leak following radical upper lobectomy. Chest. 1998;113(6):1507–10.
Bardell T, Persikas D. What keeps postpulmonary resection patients in hospital. Can Respir J. 2003;10:86–9.
Okereke I, Murthy SC, Alster JM, Blackstone EH, Rice TW. Characterization and importance of air leak after lobectomy. Ann Thorac Surg. 2005;79:1167–73.
Annegg U, Lindenmann J, Matzi V, Smolle J, Maier A, Smolle-Juttner F. Efficiency of fleece bound sealing (TachoSil) of air leaks in lung surgery: a prospective randomised trial. Eur J Cardiothorac Surg. 2007;31:198–202.
Belda Sanchís J, Serra-Mitjans M, Iglesias Sentis M, Rami R. Surgical sealant for preventing air leaks after pulmonary resections in patients with lung cancer. Cochrane Database Syst Rev. 2010;1:CD003051.
Malapert G, Hanna HA, Pages PB, Bernard A. Surgical sealant for the prevention of prolonged air leak after lung resection: meta-analysis. Ann Thorac Surg. 2010;90:1779–85.
Rocco G, Rendina EA, Venuta F, Mueller MR, Halezeroglu S, Dienemann H, et al. The use of sealants in modern thoracic surgery: a survey. Interact Cardiovasc Thoracic Surg. 2009;9:1–3.
Zhang R, Bures M, Höffler HK, Zinne N, Langer F, Bisdas T, et al. TissuePatch™ as a novel synthetic sealant for repair of superficial lung defect: in vitro tests results. Ann Surg Innov Res. 2012;6:12.
Chao HH, Torchiana DF. BioGlue: albumin/glutaraldehyde sealant in cardiac surgery. J Card Surg. 2003;18(6):500–3.
Raanani E, Georghiou GP, Kogan A, Wandwi B, Shapira Y, Vidne BA. BioGlue for the repair of aortic insufficiency in acute aortic dissection. J Heart Valve Dis. 2004;13:734–7.
Bhamidipati CM, Coselli JS, LaMaire SA. BioGlue in 2011: what is its role in cardiac surgery? J Extra Corpor Technol. 2012;44:6–12.
Tokunaga S, Yasuga S, Masuda M. Devised reinforcement of distal stump in total arch replacement using BioGlue. Asian Cardiovasc Thorac Ann. 2013;22:755–7.
D’Andrilli A, Venuta F, Rendina E. Use of Sealants to Reduce Air Leak Duration and Hospital Stay After Lung Resection. In: Ferguson MK, editor. Difficult Decisions in Thoracic Surgery: An Evidence-Based Approach. London: Springer; 2011. p. 375–83.
Macchiarini P, Wain J, Almy S, Dartevelle P. Experimental and clinical evaluation of a new synthetic, absorbable sealant to reduce air leaks in thoracic operations. J Thorac Cardiovasc Surg. 1999;117:751–8.
Varela G, Jinénez MF, Novoa N, Aranda JL. Estimating hospital cost attributable to prolonged air leak in pulmonary lobectomy. Eur J Cardiothorac Surg. 2005;27:329–33.
Potaris K, Mihos P, Gakidis I. Preliminary results with the use of an albumin-glutaraldehyde tissue adhesive in lung surgery. Med Sci Monit. 2003;9:179–83.
Rathinam S, Naidu BV, Nanjaiah P, Loubani M, Kalkat MS, Rajesh PB. BioGlue and Peri-strips in lung volume reduction surgery: pilot randomised controlled trial. J Cardiothorac Surg. 2009;4:37.
Tansley P, Al Mulhim F, Lim E, Ladas G, Goldstraw P. A prospective, randomized, controlled trial of the effectiveness of BioGlue in treating alveolar air leaks. J Thorac Cardiovasc Surg. 2006;132:105–12.
Allen MS, Wood DE, Hawkinson RW, Harpole DH, McKenna RJ, Walsh GL, et al. Prospective randomized study evaluating a biodegradable polymeric sealant for sealing intraoperative air leaks that occur during pulmonary resection. Ann Thorac Surg. 2004;77:1792–801.
Belcher E, Dusmet M, Jordan S, Ladas G, Lim E, Goldstraw P. A prospective, randomized trial comparing BioGlue and Vivostat for the control of alveolar air leak. J Thorac Cardiovasc Surg. 2010;140:32–8.
Azadani AM, Matthews PB, Ge L, Shen Y, Jhun CS, Guy TS, et al. Mechanical properties of surgical glues used in aortic root replacement. Ann Thorac Surg. 2012;87:1154–60.
Fürst W, Banerjee A. Release of Glutaraldehyde From an Albumin-Glutaraldehyde Tissue Adhesive Causes Significant In Vitro and In Vivo Toxicity. Ann Thorac Surg. 2005;79:1522–9.
The BioGlue™ samples used for this study were provided by CryoLife Europa, Ltd.. Special thanks to Mr. Stefan Pingel for his support in this regard. We also acknowledge support by Deutsche Forschungsgemeinschaft and Open Access Publishing Fund of University of Tübingen.
The authors declare that they have no competing interests.
MB carried out the conception and design of the study, in vitro tests, acquisition of data, analysis and interpretation of the data, statistical analysis as well as drafting of the manuscript. KH participated in the acquisition of data. GF, TK, EB and PZ participated in the analysis and interpretation of the data, critical revision of the manuscript and supervision of the study. FL participated in histological examination. RZ participated in the conception and design of the study, acquisition and interpretation of the data, drafting and critical revision of the manuscript. All authors read and approved the final manuscript.
About this article
Cite this article
Bures, M., Höffler, HK., Friedel, G. et al. Albumin-glutaraldehyde glue for repair of superficial lung defect: an in vitro experiment. J Cardiothorac Surg 11, 63 (2016). https://doi.org/10.1186/s13019-016-0443-x
- Air leak