- Case report
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
Left ventricular non-compaction cardiomyopathy and left ventricular assist device: a word of caution
© The Author(s). 2016
Received: 8 February 2016
Accepted: 6 July 2016
Published: 15 July 2016
In patients with left ventricular non-compaction (LVNC), implantation of a left ventricular assist device (LVAD) may be performed as a bridge to transplantation. In this respect, the particular characteristics of the left ventricular myocardium may represent a challenge.
We report a patient with LVNC who required urgent heart transplantation for inflow cannula obstruction nine months after receiving a LVAD. LVAD parameters, echocardiography and examination of the explanted heart suggested changes of left ventricular configuration brought about by LVAD support as the most likely cause of inflow cannula obstruction.
We conclude that changes experienced by non-compacted myocardium during LVAD support may give rise to inflow cannula obstruction and flow reduction. Presence of LVNC mandates tight surveillance for changes in LV configuration and LVAD flow characteristics and may justify urgent transplantation listing status.
Left ventricular non-compaction (LVNC) is attributed to arrest of normal embryogenesis of the endocardium and myocardium and results in a two-layered structure of the left ventricle (LV). In addition to a compacted outer layer, the LV comprises a spongy non-compacted inner layer with heavy trabeculation and deep intertrabecular recesses [1, 2]. Though its incidence and prevalence are still uncertain, LVNC was recently suggested to represent the third most commonly diagnosed cardiomyopathy . The major clinical findings in LVNC are heart failure, arrhythmias and thromboembolic events [1, 2]. In patients with end-stage heart failure, LVAD implantation may be performed as a bridge to transplantation [4–7].
While previous authors focused on thromboembolic events [5, 7] and arrhythmia  associated with LVAD support in LVNC, the changes non-compacted left ventricular myocardium may experience during LVAD support also merit investigation. Our case sheds light on LVAD support in LVNC from a different angle and raises questions with regard to the duration of support being likely to be limited by support-induced changes of non-compacted myocardium, LV configuration and LVAD flow behaviour.
A 35-year-old male with LVNC underwent off-pump LVAD implantation (HVAD, HeartWare Inc.) through a minimally invasive access. Presence of trabeculation obstructing the inflow cannula was excluded by digital exploration before introduction of the LVAD. Correct inflow cannula position and absence of obstruction were additionally verified by transesophageal echocardiography and evaluation of LVAD flow behaviour before closure. After an uneventful hospital stay and subsequent exercise tolerance and quality of life improvement, the patient presented with low flow alarms and had his LVAD speed re-adjusted several times. After nine months, he was readmitted for perilously low flows that intermittently dropped to 500 ml/min. This was interpreted as the consequence of an excess of outdoor activity and volume depletion, and when LVAD flows increased satisfactorily after administration of fluids, the patient was discharged the next day. Subsequently, LVAD flows kept dropping intermittently over a few weeks, and finally the patient ceased to respond to administration of fluids and developed signs of heart failure. He was re-admitted, and his waiting list status was set to high urgency.
Outcomes of LVAD implantation in patients with left ventricular non-compaction
Duration of support
Maile et al., 2004 
Male, 51 years
DeBakey MicroMed LVAD
Successfully bridged to transplantation
Ivan et al., 2005 
Male, 21 years
Jarvik biventricular assist device
Death before transplantation
Male, 37 years
Thoratec RVAD and
Death after transplantation
Nathan et al., 2010 
Male, 14 years
Abiomed, biventricular assist device
Successfully bridged to transplantation
Ninios et al., 2010 
Male, 28 years
Jarvik 2000 LV
Recurrent thrombosis with successful intravenous lysis
Patient on LVAD support at the date of publication
Hanke et al., 2012 
Male, 12 months
Berlin Heart, Excor, biventricular assist device
Pulmonary haemorrhage requiring use of a membrane oxygenator, thrombosis requiring replacement of the LVAD
Successfully bridged to transplantation
Siehr et al., 2013 
Male, 6 weeks
Berlin Heart, Excor, LVAD
Embolic cerebrovascular accident, 2 pump exchanges due to clot formation
Successfully bridged to transplantation
Uribarri et al., 2015 
4 males (23, 25, 43,54 years)
1 female (24 years)
2 HeartMate II
Mean follow up period 86.5 weeks, range 40–194 weeks
4 cases of pump thrombosis in 3 of 5 patients
2 deaths, 3 patients on LVAD support at the date of publication
Ninios et al.  reported off-pump implantation of a Jarvik 2000 LV via a left posterolateral thoracotomy, while Maile et al.  described a case of implantation of a DeBakey LVAD in a patient with LVNC. 5 patients reported by Uribarri et al.  underwent on-pump HeartMate II implantation through a full sternotomy or on-pump HVAD implantation through an upper hemisternotomy and left-sided anterolateral thoracotomy. In this series, thrombosis was the main issue, with arrhythmia investigated as a second outcome parameter . In the case reported by Ninios et al., the focus was also on the association between LVNC and pump thrombosis . The patient reported by Maile et al.  underwent transplantation after 2.5 months of uneventful LVAD support, and Ivan et al.  reported two cases that took a fatal course after biventricular assist devices were implanted as a last remedy. In none of the cases published to date, however, mention was made of complications resulting from the particular structure of the left ventricular myocardium in patients with LVNC even though some of the patients were on LVAD support for considerably longer periods than our patient [5, 7] and Uribarri et al.  explicitly reported the apex to be the region that was most frequently affected by non-compaction in their patients.
The latter finding is in keeping with previous literature reporting the typical features of LVNC to be located predominantly in the apical and mid-ventricular segments of the LV , which is relevant in that trabecular structures within a nearly obliterated apex may result in obstruction of the inflow cannula. In this respect, the implantation method may be of relevance, because on-pump implantation allows more thorough inspection and palpation of the left ventricle in order to exclude presence of or remove apical trabecular structures that may result in obstruction of the inflow cannula.
In our case, off-pump HVAD implantation was uncomplicated, obstruction of the inflow cannula by apical trabecular structures was excluded by digital palpation as well as by transesophageal echocardiography, and unhindered blood flow into the inflow cannula was reflected by normal LVAD flow characteristics. In hindsight, however, on-pump implantation would have allowed not only inspection of the left ventricular apex, but also excision of excess tissue. On the other hand, the fact that first signs of inflow cannula obstruction occurred after several months of uneventful support and absence, in earlier echocardiographic examinations, of the heavy apical and mid-ventricular trabecular masses seen on echocardiography after readmission for LVAD dysfunction suggest that at least part of the changes that finally resulted in inflow cannula obstruction must have occurred over time.
While decompression of the LV is a normal and desirable consequence of LV unloading, the extreme degree of narrowing of the LV cavity we saw in the explanted heart as well as the heavy mass of rigid trabeculated tissue around the inflow cannula represent unusual findings. They are most likely explained by LV configuration and the geometry and density of pre-existing trabeculated tissue having changed in response to the hemodynamic changes that were brought about by LVAD support.
We conclude that LVNC may turn out to be a cause of inflow cannula obstruction and flow reduction even where LVAD function was normal upon implantation. On-pump implantation may be advisable with a view to inspecting the inflow opening and excising excess tissue, thus preventing subsequent catastrophic inflow obstruction. LVNC requires a high index of suspicion and mandates tight surveillance of LVAD flow characteristics and LV configuration. If LVNC actually turns out to be a limiting factor in LVAD therapy, it may justify more liberal access to high urgency listing for patients afflicted with this particular condition.
ECLS, extracorporeal life support; HVAD, left ventricular assist device supplied by HeartWare Inc; LV, left ventricle, left ventricular; LVAD, left ventricular assist device; LVNC, left ventricular non compaction; VAD, ventricular assist device
We thank the physicians of the intensive care unit for the care they provided for the patient.
The authors have received no funding for the preparation of this paper.
KA conceived the design of the project, collected and interpreted the data and images and drafted the manuscript. RP, SUA and BFA were involved in the surgical treatment and intensive care of the patient, have critically revised the manuscript and contributed to interpreting the data. BFA gave final approval of the version to be published. All authors have read and approved the final manuscript.
The authors declare that they have no competing interests.
Consent for publication
Written informed consent was obtained from the patient for publication of this case report and any accompanying images. A copy of the written consent is available for review by the Editor-in-Chief of this journal.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Oechslin E, Attenhofer Jost CH, Rojas JR, Kaufmann PA, Jenni R. Long-term follow-up of 34 adults with isolated left ventricular noncompaction: a distinct cardiomyopathy with poor prognosis. J Am Coll Cardiol. 2000;36:493–500.View ArticlePubMedGoogle Scholar
- Weiford BC, Subbarao VD, Mulhern KM. Noncompaction of the ventricular myocardium. Circulation. 2004;109:2965–71.View ArticlePubMedGoogle Scholar
- Towbin JA, Lorts A, Jefferies JL. Left ventricular non-compaction cardiomyopathy. Lancet. 2015;386:813–25.View ArticlePubMedGoogle Scholar
- Maile S, Kunz M, Oechslin E, Zund G, Rahn M, Lachat ML, Turnia MI. Intractable ventricular tachycardia and bridging to heart transplantation with a non-pulsatile flow assist device in a patient with isolated left-ventricular non-compaction. J Heart Lung Transpl. 2004;23:147–9.View ArticleGoogle Scholar
- Ninios V, Visouli A, Pitsis A. Images in cardiovascular medicine. Repeated successful thrombolysis of a Jarvik 2000 left ventricular assist device in a patient with noncompaction cardiomyopathy. Circulation. 2010;121:e13–4.View ArticlePubMedGoogle Scholar
- Ivan D, Flamm SD, Abrams J, Kindo M, Heck K, Frazier OH. Isolated Ventricular Non-Compaction in Adults With Idiopathic Cardiomyopathy: Cardiac Magnetic Resonance and Pathologic Characterization of the Anomaly. J Heart Lung Transpl. 2005;24:781–6.View ArticleGoogle Scholar
- Uribarri A, Rojas SV, Avsar M, Hanke JS, Napp LC, Berliner D, Bavendiek U, Bauersachs J, Bara C, Sanchez PL, Haverich A, Schmitto JD. First series of mechanical circulatory support in non-compaction cardiomyopathy: Is LVAD implantation a safe alternative? Int J Cardiol. 2015;197:128–32.View ArticlePubMedGoogle Scholar
- Ritter M, Oechslin E, Sutsch G, et al. Isolated noncompaction of the myocardium in adults. Mayo Clin Proc. 1997;72:26–31.View ArticlePubMedGoogle Scholar
- Nathan M, Camacho M, Hardin JT, Starr JP. Abiomed biventricular assist device as a bridge to transplantation in a 14-year-old with cardiomyopathy resulting from ventricular noncompaction. Pediatr Cardiol. 2010;31:1249–51.View ArticlePubMedGoogle Scholar
- Hanke SP, Gardner AB, Lombardi JP, Manning PB, Nelson DP, Towbin JA, Jefferies JL, Lorts A. Left ventricular noncompaction cardiomyopathy in Barth syndrome: an example of an undulating cardiac phenotype necessitating mechanical circulatory support as a bridge to transplantation. Pediatr Cardiol. 2012;33:1430–4.View ArticlePubMedGoogle Scholar
- Siehr SL, Bernstein D, Yeh J, Berry GJ, Rosenthal DN, Hollander SA. Orthotopic heart transplantation in two infants with histiocytoid cardiomyopathy and left ventricular non-compaction. Pediatr Transplant. 2013;17:E165–167.View ArticlePubMedGoogle Scholar