Serious hazards of transfusion: evaluating the dangers of a wrong patient autologous salvaged blood in cardiac surgery
Journal of Cardiothoracic Surgery volume 17, Article number: 182 (2022)
The past half century has seen the near eradication of transfusion-associated hazards. Intraoperative cell salvage while widely used still poses significant risks and hazards due to human error. We report on a case in which blood collected from a patient with lung cancer was mistakenly administered to a patient undergoing cardiac surgery who should have received his own collected blood. The initial investigation found that the cause of the patient harm was violations of procedures by hospital personnel. A detailed investigation revealed that not only violations were the cause, but also that the underlying causes included haphazard organizational policies, poor communication, workload and staffing deficiencies, human factors and cultural challenges.
On August 14, 2019, a 72-year-old male was admitted to our hospital for angina pectoris and multivessel coronary artery disease. Cardiac surgery was performed using an autologous salvage blood collection system, and there were no major problems other than the prolonged operation time. During the night after the surgery, when the patient’s blood pressure dropped, a nurse retrieved a blood bag from the ICU refrigerator that had been collected during the surgery and administered it at the physician's direction, but at this time neither the physician nor the nurse performed the required checking procedures. The blood administered was another patient's blood taken from another surgery the day before; an ABO mismatch transfusion occurred and the patient was diagnosed with DIC. The patient was discharged 65 days later after numerous interventions to support the patient. An accident investigation committee was convened to analyze the root causes and develop countermeasures to prevent a recurrence.
This adverse event occurred because the protocol for intraoperative blood salvage management was not clearly defined, and the procedure was different from the standard transfusion practices. We developed a new workflow based on a human factors grounded, systems-wide improvement strategy in which intraoperative blood collection would be administered before the patient leaves the operating room to completely prevent recurrence, instead of simply requiring front-line staff to do a double-check. Implementing strong systems processes can reduce the risk of errors, improve the reliability of the work processes and reduce the likelihood of patient harm occurring in the future.
Autotransfusion of patient blood has been widely used in various surgeries since the early 1970s . One method of blood salvage is to collect the blood from the operative field and store it intraoperatively . Intraoperative cell salvage is effective for the conservation of red blood cells . Although technical errors in intraoperative cell salvage have been reported, there have been no reported cases of blood transfusion to a wrong patient, a so called "never event" . In this case report, we describe a wrong patient blood infusion case that resulted in the wrong blood unit given to the patient collected as autologous-salvaged blood (ASB) during cardiac surgery. We investigated the causes of this never event using root cause analysis (RCA) approach to intuitively lay out the incident information and quickly show the cause-and-effect relationships that contributed to the patient harm, in order to prevent similar incidents.
A 72-year-old Patient X with a history of congestive heart failure and chronic atrial fibrillation, requiring dialysis for chronic renal failure due to nephrosclerosis, was admitted to our hospital because of angina pectoris and multi vessel coronary artery disease. The patient’s home medications included warfarin potassium (1 mg), clopidogrel sulfate (50 mg), amiodarone hydrochloride (100 mg), and bisoprolol fumarate (1.25 mg). The patient’s blood type was O RhD positive. The blood was collected by the ASB system and the coronary artery bypass surgery (CABG) was successfully completed. The CABG surgery was prolonged due to a lenghty re-anastomosis of one of the three-vessel cardiac grafts. The operative time was 8 h 19 min, intraoperative blood loss was 790 ml, and no autologous or allogeneic blood transfusions were performed. The hospital usually stocks the collected blood in the intensive care unit (ICU) if the case does not require urgent use of the collected blood. The surgeon usually decides depending on the patient's condition whether to transfuse the blood back to the patient or discard it within 24 h after the operation.
The physician in charge on the night after the operation, decided to administer the salvaged blood of the patient due to a decrease in systolic blood pressure. The low blood pressure was attributed to dehydration as the patient was not bleeding, not in shock, and had no abnormalities in cardiac function, blood gases, or their electrocardiogram. The physician instructed nurse A to bring the patient’s blood from the ICU refrigerator. Because nurse A was engaged in the care of another patient, she requested that nurse B to “Bring it to me.” Nurse B went to the ICU refrigerator and picked up a blood bag. However, the bag retrieved by nurse B was not that of Patient X but was collected from another Patient Y, who had undergone thoracic surgery the day before to remove their lung cancer. The blood type of Patient Y was A RhD negative. When stored in the refrigerator, a note with the patient's name and identification (ID) number was supposed to be put on the basket, but this was not present at the time of the incident. The name was written on the pack, but nurse B did not check this because she was in a hurry. Nurse B believed that the blood in her hand was that of Patient X, and handed it directly to the nurse A, who then connected it to the patient's intravenous line and started the infusion without further ascertainment to asure the correct patient's blood identification. The two-person check at the bedside by nurses or physician and nurse was not performed. For a typical blood transfusion, a physician will use a Personal digital assistant (PDA) for verification, but the salvaged blood bag does not have a barcode and is not verified by a PDA.
The patient became hypotensive with blood pressures in the range of 50 to 60 mmHg systolic pressure range and vasopressors were initiated. Several minutes after the transfusion began, the patient's blood pressure dropped further, with a systolic pressure consistently in the low 50 s. The blood transfusion was stopped due to growing concern about the possibility of wrong blood transfusion. The volume of blood administered was estimated to be approximately 50 mL. The blood bag was removed from the venous line and the transfusion was stopped. Albumin and noradrenaline drop were administered, and over the next 15–20 min, the patient’s systolic blood pressure increased to the 130 mmHg range. There were no physical findings suggestive of an allergic reaction or hemolytic urine. The surgeon thought that the cause of the hypotension was a low circulating blood volume. One hour later, the patient’s serum hemoglobin level decreased to 9.2 g/dl from 11.4 g/dl preoperatively. The nurse was instructed to restart the transfusion. The nurse connected the blood that was hung on the bedside infusion table to the patient and restarted the transfusion. All residual blood was transfused into the patient. A decrease in the number of platelets was noted the next morning, blood pressures were lower and the patient was diagnosed with disseminated intravascular coagulation (DIC) syndrome.
The charge nurse discovered during a regular safety check of the ICU refrigerator that the ASB blood, which should have already been administered, was still stored. This revealed that the blood administered to Patient X was that of Patient Y. Patient X was successfully resuscitated and was discharged from the ICU on day 8. The patient was finally discharged home on day 65 due to persistent pleural effusions and positive CRP and had delayed rehabilitation due to severe back pain.
Adverse incident aftermath: apology and disclosure
An hour after the accident occurred the concerned parties gathered to discuss the best course of action. Three and a half hours after the wrong transfusion was recognized, the surgeon in charge and staff of the patient safety department disclosed the facts of the events to the patient's family and apologized to the family . Nine hours later, a second disclosure and apology were made to the patient's family. From day 12 to 60 day after the accident, several meetings and e-mail conferences were held involving the investigation committee, chaired by an external committee member tasked with analyzing the causes of the incident and proposing countermeasures to prevent a recurrence. The investigation report was submitted to the hospital President, who accepted the change recommendations. The hospital worked to be transparent during the investigation and issued statements about the changes being made to improve the reliability of their work processes and reduce the likelihood of a similar event occurring in the future. A periodic audit of blood transfusions indicated that all the recommended changes in ABS were implemented, and no other ASB related adverse events have occurred since.
Discussion and conclusions
We describe a completely preventable wrong patient blood transfusion in which a patient was administered another patient’s blood, collected as autologous-salvaged blood (ASB) during cardiac surgery. We review the root cause investigation (RCA) and highlight the systems' issues that emerged and the corrective actions implemented in the hospital to prevent similar adverse incidents.
The history of blood supply is one of early, sobering frequency of disease transmission but also remarkable improvement in systems safety in terms of correct blood transitions that is free of infectious agents. Blood product safety has been an improving area of focus over recent decades in many countries . Several methods have been employed to reduce the risk of blood transfusions and improve blood product administration safety . The entire blood harvesting and transfusion process has been redesigned from before blood donation collection through to the post-procedure follow-up with the blood product recipients. There have been improvements made in the collection, storage, management, distribution, utilization, and monitoring of transfusions.
Transfusion system infrastructure
The transfusion service is one of the most highly regulated services within the clinical laboratory. The pathway of blood delivery is inherently complex because multiple patient care areas are involved. The goal of the blood delivery pathway is to deliver the right product to the correct patient .
The pathway can be summarized by three simple steps:
Identify the patient with two unique identifiers (ID).
Connect the patient identifiers to all prepared lab samples, tests, and blood products.
Deliver the right blood product to the right patient at the right time, confirming patient ID again.
These three simple steps comprise numerous processes, each with their own risks of failure, with the highest rates of failure associated with processes outside of the clinical laboratory.
Risk of ABO-incompatible transfusions and hemolytic reactions
The risk of fatality due to an ABO-mismatched red blood cell transfusion is estimated at 1 to 4 per 10,000,000 of each red blood cell unit transfused . But fatal reactions represent the ‘tip of the iceberg’ as most ABO-incompatible near miss transfusions involve small volumes due to early clinical signs/symptoms and rarely do patients not survive .
The risks of a lethal hemolytic transfusion reaction were estimated at 1 per 550,000 units transfused for the time period 1976–1985 in the US . Not all hemolytic reactions are ABO-related and not all wrong transfusion events result in adverse clinical outcomes. Others have estimated that 1 in every 19,000 units of red blood cells is transfused to the wrong patient each year, 1 in 76,000 transfusions results in an acute hemolytic reaction, and, 1 in 1.8 million units of transfused red blood cell units results in death due to acute hemolytic reaction .
When estimating the risks, the best information available indicates that most transfusions to the wrong patient occur as a result of potentially avoidable system failures . The most frequent error leading to transfusion of ABO-incompatible blood occurs during patient identification/verification at the bedside; as a result, although the blood is labeled appropriately, it can result in the wrong blood being given to the patient.
Root cause analysis of the case (RCA)
A linkage diagram (Fig. 1) was created and a root cause analysis of events (Table 1) was developed to link the problem statement to the conditions and actions. An RCA, is a visual format for performing a root cause analysis, allowing us to intuitively lay out the information to quickly show the cause-and-effect relationships that contributed to this adverse incident. These charts help identify a number of factors that predispose the blood transfusion system to errors and revealed multiple contributing factors, including communication problems, human factors problems, inadequate policies and procedures, cultural problems, etc . The downstream effects of the errors led to the wrong blood reaching the patient’s bedside and to a serious patient outcome. The linkage diagram does not contain all of the information, but it provides an overview of how the incident occurred and helps to organize the information in a way that can be quickly and visually understood.
Unfortunately, but importantly, the linkage diagram shows the patient’s adverse event was not the result of a single error. Most errors are caused by a combination of cascading failures, and it is rare that a single error leads to an incident. Initially, the cause of this accident appeared to be a simple case of rule violation, as the blood was administered by a nurse without checking the name against the blood bag label . However, upon deeper reflection, it became clear that identifying this as the cause and taking measures to prevent recurrence, such as double-checking the patient details, may not have prevented a recurrence [16,17,18] in the future. Therefore, the investigation committee searched for system errors at a deeper level. There were multiple process failures and at several missed opportunities where this error could have been either prevented or caught earlier.
Nurse B's action in "administering the wrong infusion" was considered a possible cause of human error, which was investigated using the RCA (see below). Human error is defined as the failure of planned actions to achieve their desired ends without the intervention of some unforeseeable events, and it is not the cause of the accident; it is the factor that causes the human to make the error that is the cause of the accident . Human behavior is the result of a combination of human and environmental factors, known as the Lewin’s equation  or the SHELL model . Figure 1 shows an extract from the RCA. Two factors, human and environmental factors, are involved in Nurse A's behavior. The unique actions of Nurse A are shown on the bottom, and the environmental factors are shown on the top of the figure.
First, we consider the human factors involved . We explored the reasons why Nurse A believed that the blood bag in her hand belonged to Patient X and not to Patient Y. The probable background factors included that the: (1) autologous salvaged blood units from multiple patients were stored in a single refrigerator in the ICU and this information was not shared among the staff; (2) the patient’s ID and name were handwritten on the bag after collection, not labeled as required; and (3) barcode matching system was not applied to the salvaged blood and the ABS administration was not managed by the Blood Transfusion Service.
Next, the environmental factors were discussed. The cause of Patient Y's blood being passed into the hands of Nurse A as Patient X's blood was discussed. The probable background factors for how Patient Y's blood got into the hands of Nurse A were as follows: (1) the blood collected from multiple patients was stored in multiple refrigerators in the ICU; (2) the patient’s name and other information were not written on the basket as required when storing the blood bag in the refrigerator; (3) normally, a paper label with the patient's name and ID number must be affixed to the bag after blood collection before usage, but in this case, this label was not used, and the patient's ID number and name were hand written directly on the transfusion bag with a black magic marker; and (4) the thoracic surgeon who salvaged the blood from Patient Y was not familiar with blood salvage protocols and did not make a decision within 24 h to use or discard the blood.
It is clear from considering factors 1–3 above, that there were no clear SOPs defined in the ICU. Nurse A administered the incorrect blood unit, believing that the blood unit in her hand belonged to Patient X, when in fact it belonged to Patient Y. The investigation committee determined that the "intraoperative salvaged blood, without starting to administer the blood before leaving theatre, and bringing the bag to the ICU," was the lead root cause of this event.
Corrective and preventative action plan
The investigation committee believed that measures addressing the violations of the confirmation process by the staff, background factors such as lack of information sharing, and compliance failures in storage and destruction of bags, while critical, would not have completely prevented this from recurring. Therefore, measures to prevent a recurrence were focused on the upstream causes, which includes how procedures for intraoperative salvaged blood are managed. Transfused salvaged blood is generally recommended to be administered in the operating room but it is not prohibited to be given outside the operating room. However, to prevent blood unit mix-ups, it is necessary to start the ABS administration in the operating room. The draft guidelines for the Implementation of Transfusion of Autologous Salvaged Blood (2020) by the Japanese Society for Autologous Blood Transfusion indicates that “in principle,” administration should be started in the operating room . At our hospital, we decided to go one step further and require starting all ASB administration in the operating room without exception. Additionally, new regulations have been established regarding how to record blood salvage, label ABS bags, and instructions for administration and disposal, which previously were not clearly stated. These regulations will be managed by the Blood Transfusion Service of the hospital. Cell salvage equipment and staff trained are needed to operate immediately and be available 24 h a day when undertaking surgery where blood loss is a potential complication. The hospital should nominate a clinical lead and a coordinator for cell salvage, who oversee a competence-based training program for all involved staff, along with ongoing data collection and regular data audits. This training should be overseen by the Blood Transfusion Division.
In conclusion, we investigated a serious adverse patient incident in which intraoperatively salvaged blood was transfused to the wrong patient with a different blood type. Under the Human Factors theory, failures are not satisfactorily explained by demonstrating human deviation from expected behavior. Instead, the circumstances and underlying pressures are meticulously explored, and systemic deficiencies are identified. It is always easier to see the warning signs in hindsight but in reviewing incidents of missed warning signs we are reminded to look for patterns of smaller incidents and to take near-misses seriously. Interventions to improve blood safety should lean towards design improvements, engineering controls, or process simplification and standardization. We developed and implemented a systems' improvement strategy that requires a stringent verification process in collecting and labeling blood specimens. The intraoperative blood collection is administered before the patient leaves the operating room in order to completely prevent a wrong blood administration recurrence, instead of simply requiring front-line staff to check or double-check the blood compatibility. The hospital has been working to improve the reliability of their work processes, enhancing policy and training improvements, and making all efforts to reduce the likelihood of a similar error occurring in the future. While strict rules are a reminder of what not to do, people’s ability to perceive the situation and adapt is the primary source of safety.
Availability of data and materials
The datasets used and/or analysed during the current study are available from the corresponding author on reasonable request after approval by our Ethics Committee.
Autologous Salvaged Blood
Root Cause Analysis
Intensive Care Unit
Personal Digital Assistant
Roche JK, Stengle JM. Open-heart surgery and the demand for blood. JAMA. 1973;225:1516–21.
Tawes RL Jr. Clinical applications of autotransfusion. Semin Vas Surg. 1994;7:89–90.
UK National Blood Service. Blood Matters. 2004;16.
Waters JH, Sprung J. Erros during intraoperative cell salvage because inappropriate wash solutions. Anesth Analg. 2001;93:1483–5. https://doi.org/10.1097/00000539-200112000-00027.
Cantor M, Barach P, Derse A, Maklan C, Woody G, Fox E. Disclosing adverse events to patients. Jt Comm J Qual Saf. 2005;31:5–12.
Harris JC, Crookston KP. Blood product safety. In: StatPearls. Treasure Island (FL). StatPearls Publishing; 2022. https://www.ncbi.nlm.nih.gov/books/NBK539826/.
Sharma S, Sharma P, Tyler LN. Transfusion of blood and blood products: indications and complications. Am Fam Physician. 2011;83(6):719–24.
Cutts JC, Quinn B, Seed CR, Kotsiou G, Pearson R, Scott N, Wilson DP, Harrod ME, Maher L, Caris S, Thompson AJ, Farrell M, Pink J, Hellard ME. A systematic review of interventions used to increase blood donor compliance with deferral criteria. Transfus Med Hemother. 2021;48:118–29. https://doi.org/10.1159/000509027.
Storch EK, Rogerson B, Eder AF. Trend in ABO-incompatible RBC transfusion-related fatalities reported to the FDA, 2000–2019. Transfusion. 2020;60(12):2867–75. https://doi.org/10.1111/trf.16121.
Janatpour KA, et al. Clinical outcomes of ABO-incompatible RBC transfusions. Am J Clin Pathol. 2008;129:276–81.
Strobel E. Hemolytic transfusion reactions. Transfus Med Hemother. 2008;35(5):346–53.
Vamvakas EC, et al. Transfusion-related mortality: the ongoing risks of allogeneic blood transfusion and the available strategies for their prevention. Blood. 2009;113:3406–17.
Stainsby D, et al. Serious hazards of transfusion: a decade of hemovigilance in the UK. Transfus Med Rev. 2006;20:273–82.
AHRQ’s Patient Safety Initiative: Building Foundations, Reducing Risk. Publication # 04-RG005. Chapter 2. Efforts to Reduce Medical Errors: AHRQ’s Response to Senate Committee on Appropriations Questions. Secondary AHRQ’s Patient Safety Initiative: Building Foundations, Reducing Risk. Publication # 04-RG005. Chapter 2. Efforts to Reduce Medical Errors: AHRQ’s Response to Senate Committee on Appropriations Questions. http://archive.ahrq.gov/research/findings/final-reports/pscongrpt/psini2.html. Accessed 17 Mar 2022.
Ri M, Kasai M, Kohno A, Kondo M, Sawa M, Kinoshita T, et al. A survey of blood transfusion errors in Aichi Prefecture in Japan: Identifying major lapses threatening the safety of transfusion recipients. Transfus Apher Sci. 2020;59(3):102735. https://doi.org/10.1016/j.transci.2020.102735.
Institute of Medicine (US) Committee on Quality of Health Care in America, Kohn LT, Corrigan JM, Donaldson MS, eds. To err is human: building a safer health system. Washington: National Academies Press; 2000.
Storch EK, Rogerson B, Eder AF. Trend in ABO-incompatible RBC transfusion-related fatalities reported to the FDA, 2000–2019. Transfusion. 2020;60:2867–75. https://doi.org/10.1111/trf.16121.
Murphy MF, Jayne Addison J, Poles D, Dhiman P, Bolton-Maggs P. Electronic identification systems reduce the number of wrong components transfused. Transfusion. 2019;59(12):3601–7. https://doi.org/10.1111/trf.15537.
Reason J. Human error models and management. West J Med. 2000;172:393–6. https://doi.org/10.1136/ewjm.172.6.393.
Balkenius C. Natural intelligence in artificial creatures 37. 1st ed. Lund: Lund University Cognitive Studies; 1995. p. 79–126.
Hawkins FH, Orlady HW, editors. Human factors in flight. 2nd ed. England: Avebury Technical; 1993.
Mistry H, Poles D, Watt A, Bolton-Maggs PHB. Human errors in manual techniques for ABO/D grouping are associated with potentially lethal outcomes. Transfus Med. 2019;29:262–7. https://doi.org/10.1111/tme.12616.
The Japanese Society for Autologous Blood Transfusion. The draft guidelines for the Implementation of Transfusion of Autologous Salvaged Blood. 2020. https://www.jsat.jp/jsat_web/down_load/pdf/kaisyushiki_jikoketsukijun2020.pdf. Accessed 28 Apr 2022.
We would like to thank the doctors, nurses, and all of care team members, as well as all participants in the accident investigation committee.
No funding was received to write the paper.
Ethical approval and consent to participate
The Tokyo Medical University Institutional Review Board permits case reports with the following requirements for permission: comprehensive consent at the time of admission to be the subject of a report at conference or journal, and the patient's written consent for the submission to journals. In this case, in addition to obtaining comprehensive consent at the time of admission, written consent was also obtained for publication.
Consent for publication
Attach a written consent in the patient's own handwriting to the instructions written in the native language.
The authors declare no conflicts of interest associated with this manuscript.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Uramatsu, M., Maeda, H., Mishima, S. et al. Serious hazards of transfusion: evaluating the dangers of a wrong patient autologous salvaged blood in cardiac surgery. J Cardiothorac Surg 17, 182 (2022). https://doi.org/10.1186/s13019-022-01931-6
- Patient safety
- Root cause analysis
- Human factors
- Autologous-salvaged blood
- Wrong patient
- Medical error