Posterior mitral annuloplasty for enhancing mitral leaflet coaptation: using a strip designed for placement in the posterior annulus
© Kim et al. 2015
Received: 25 January 2015
Accepted: 28 October 2015
Published: 13 November 2015
In patients with mitral valve regurgitation (MR), posterior mitral annuloplasty (PMA) was performed for mitral valve repair using a strip designed for placement in the posterior annulus, sparing the anterior annulus and anterior half of the commissures.
Between September 2009 and October 2013, we performed PMA using a novel strip in 74 consecutive patients with MR greater than 3+. Procedures associated with mitral valve repairs were performed in 41 patients (56.9 %), including new chord placement for leaflet prolapse (n=30), patch valvuloplasty for posterior chord rupture (n=4), and posterior leaflet augmentation (n=15). All patients were analyzed by serial echocardiographic follow-up, and preoperative and postoperative computed tomography was performed in 10 randomly selected patients.
Hospital death occurred in two patients (2.7 %), and 72 survived patients were completely followed up. At a mean follow-up of 37.2 ± 15.0 months, the MR grade was zero or 1+ in 64 patients (88.9 %), 2+ in 7 patients (9.7 %), and 3+ in one patient (1.4 %). The mean indexed valve area and mean valve gradient were 1.7 ± 0.4 cm2/m2 and 3.5 ± 1.2 mmHg, respectively. The mean leaflet coaptation height in early systole was 12.8 ± 3.5 mm. During the cardiac cycle, the repaired valves exhibited dynamic changes of 19.5 ± 9.3 % in the septo-lateral dimensions. No early conversions to valve replacements or late reoperations occurred. None of the patients with remnant or recurrent MR experienced hemolysis.
PMA using a novel strip showed a sufficient coaptation height secondary to reduction of the septo-lateral annular dimensions and dynamic changes in the dimensions. It can be expected to be an alternative mitral annuloplasty technique with satisfactory results.
KeywordsHeart valve Mitral valve repair Surgery/techniques
Most remodeling annuloplasty rings fix the annular dimensions of the mitral valve (MV) in a settled shape . The partial flexible bands restrict the hinge motion between the anterior and posterior leaflets due to their fixation to trigones, commissures, and the posterior annulus. The rings and bands can restore valve competence, but they restrict most annular motion. In addition, because the annulus is fixed to the ring or band in a flat plane, it loses the commissural hinge work and three-dimensional saddle shape [2, 3]. Such normal annular geometry may be preserved by an annuloplasty strip that spares the anterior annulus and commissures. We retrospectively evaluated in a cohort of patients underwent posterior mitral annuloplasty (PMA) using a novel strip designed for placement in the posterior annulus.
This study was a retrospective review of the prospective follow-up of mitral valve regurgitation (MR) patients who underwent PMA using a novel annuloplasty strip for MV repair. This study was approved by the Institutional Review Board at Chonbuk National University Hospital.
Preoperative characteristics of 74 patients undergoing PMA for mitral valve regurgitation
1.62 ± 0.19
NYHA III and IV
Preop LVEF, %
54.3 ± 10.5
41.8 ± 15.7
Cause of MR
Acute chord rupture
Chronic inflammation (myeloproliferative)
Preop MR grade
3+ (moderate to severe)
Posterior mitral annuloplasty strip
In 10 randomly selected patients, preoperative and postoperative computed tomography was performed to observe the mitral annular shape.
Transthoracic echocardiography was performed at admission, discharge, 6 months postoperatively, and annually. The MR grade was determined according to the following scale: 0, no or trivial MR; 1+, mild; 2+, mild to moderate MR; 3+, moderate to severe MR; and 4+, severe MR. The MV orifice area was assessed by the pressure half-time method. In the parasternal long-axis view, the maximum and minimum septo-lateral dimensions were measured in diastole and systole, respectively. The coaptation height (i.e., the longest coaptation length of the anterior and posterior leaflets) was measured in early systole.
All statistical analyses were performed in SPSS 18.0 (IBM, Armonk, NY, USA). Continuous variables were expressed as the mean ± standard deviation and compared using the Student’s t test and paired t-tests. Categorical variables were expressed as proportions (%) and compared using the χ2 test.
MR grade, at discharge
0 (absent or trivial)
+3 (moderate to severe)
MR grade, the latest
0 (absent or trivial)
+3 (moderate to severe)
2.7 ± 0.5
Index MVA, cm2/m2
1.7 ± 0.4
Valve gradient, mmHg
3.5 ± 1.2
Leaflet coaptation height, mm
12.8 ± 3.5
Perioperative annular dimensions in the parasternal long-axis view
In diastole (max), cm
32.2 ± 7.3
21.1 ± 5.1
In systole (min), cm
27.1 ± 6.7
17.6 ± 4.3
Dynamics, % (max-min/min)
18.8 ± 11.7
19.5 ± 9.3
The PMA strip was designed to lift the middle portion of the posterior annulus and increase coaptation length with placement in the posterior annulus, sparing the anterior annulus and commissures. Two mitral leaflets can yield competent coaptation with reduction of the septo-lateral annular dimensions without reduction of the transverse annular dimensions, particularly in patients with functional MR . Although transverse reduction of the mitral annular dimensions is developed in the usual mitral annuloplasty, a transverse annular reduction is not likely to be a prerequisite for two leaflet coaptation.
The middle part of the strip is lifted by two thick margins into a curvilinear shape. As a result, the middle of the annulus-strip complex is lifted on the base of both commissural planes and the posterior annulus becomes curvilinear with a resultant reduction of the septolateral dimensions. The reduction effect of the septolateral dimensions must be more effective than the usual rings or bands that make a round annular shape. Such a coaptation-enhancing mechanism is seen in the GeoForm annuloplasty ring (Edwards Lifesciences, Irvine, CA, USA) with a diminished antero-posterior distance  used to repair ischemic MR, but the motion of the anterior annulus and both commissures is restricted by the rigid ring fixing the annular circumference.
In both commissures spared from the strip annuloplasty, hinges are made by the lower-placed strip ends, which tend to be straightened. Regurgitation or prolapse from the spared commissures did not occur during the follow-up period.
In our cases of rheumatic or ischemic MR or narrow posterior leaflet < 1.0 cm in height, the posterior leaflet was augmented with an elliptical pericardial patch to obtain a sufficient coaptation area prior to PMA . A tethered leaflet causing ischemic MR  can yield a proper coaptation area with posterior leaflet augmentation. The leaflet augmentation was also useful in rheumatic MR patients with a short leaflet height. The combined procedure of posterior leaflet augmentation and PMA is a two-leaflet repair rather than a monocusp repair.
MR caused by annular dilatation was well repaired with placement of the strip only. In patients with posterior leaflet prolapse due to chordal rupture, however, new chord placement  or patch vavuloplasty  may be more effective for increasing the leaflet coaptation area than the resection techniques for the prolapsed segment.
After the typical mitral annuloplasty using a ring, the redundant posterior leaflet with leaflet augmentation may be a risk for developing new systolic anterior motion , because the posterior leaflet has been fixed to the trigones by the usual rings or bands. However, because the PMA strip is separated from the anterior annular components, the augmented posterior leaflet has nothing to do with systolic anterior motion.
The size of the annuloplasty rings or bands is determined by various methods [12–14], but the ring size or band length recently has not seemed as important [12, 14, 15]. For our patients, the posterior annulus was reduced to approximately 1.5 times the anterior annular length . The strip length is not likely to affect the anterior annular length after strip annuloplasty.
The PMA is a simple procedure that can be easily performed with six interrupted mattress sutures in the posterior annulus, and its reliable coaptation induction results in a high repair rate without re-repair or conversion to valve replacement. Because of its simplicity and reliability, the procedure can be performed liberally for most MR cases that are a questionable decision for annuloplasty, such as moderate MR associated with dilated annulus, during the other main cardiac procedure. We never experienced hemolysis from remnant regurgitation, probably because of the flat shape of the strips and supraannular position.
The present study has some limitations. We did not obtain three-dimensional echocardiographic images that could demonstrate the lifted posterior annulus and preserved commissural hinge. Also, no comparison study with other rings or bands was performed. In the follow-up computed tomography, however, the lifted posterior annulus and the hinged commissures were readily demonstrated.
PMA using a novel strip for placement in the posterior annulus showed a sufficient coaptation height secondary to reduction of the septo-lateral annular dimension and dynamic change of the dimensions. It can be considered as an alternative mitral annuloplasty technique with satisfactory results.
We thank Dr. Eun Young Kim who provided cardiac computed tomographic angiograms. This study was supported by funds from the Research Institute of Clinical Medicine of Chonbuk National University and Biomedical Research Institute of Chonbuk National University Hospital.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Filsoufi F, Chikwe J, Adams DH. Acquired disease of the mitral valve. In: Sellke FW, del Nido PJ, Swanson SJ, editors. Sabiston & Spenser Surgery of the chest. 8th ed. Philadephia: Saunders Elsevier; 2010. p. 1221.Google Scholar
- Vergnat M, Levack MM, Jassar AS, Jackson BM, Acker MA, Woo YJ, et al. The influence of saddle-shaped annuloplasty on leaflet curvature in patients with ischaemic mitral regurgitation. Eur J Cardiothorac Surg. 2012;42:493–9.View ArticlePubMedPubMed CentralGoogle Scholar
- Ryomoto M, Mitsuno M, Yamamura M, Tanaka H, Fukui S, Tsujiya N, et al. Is physiologic annular dynamics preserved after mitral valve repair with rigid or semirigid ring? Ann Thorac Surg. 2014;97:492–7.View ArticlePubMedGoogle Scholar
- Chung JW, Shin JK, Song MG, Chee HK, Kim JS. Patch valvuloplasty in patients with posterior chordae rupture. Int J Cardiol. 2012;154:206–7.View ArticlePubMedGoogle Scholar
- Kim KH, Kim MH, Kim WH. Improvement of mitral valve coaptation with supraannular plication of the posterior annulus: a newly designed strip for posterior annular plication. Ann Thorac Cardiovasc Surg. 2012;18:95–100.View ArticlePubMedGoogle Scholar
- Calafiore AM, Di Mauro M, Iacò AL, Varone E, Romeo A, Mangiafico S, et al. Resecting and nonresecting techniques for posterior mitral leaflet prolapse. J Card Surg. 2011;26:119–23.View ArticlePubMedGoogle Scholar
- Maisano F, Ziskind Z, Grimaldi A, Blasio A, Caldarola A, De Bonis M, et al. Selective reduction of the septolateral dimensions in functional mitral regurgitation by modified-shape ring annuloplasty. J Thorac Cardiovasc Surg. 2005;129:472–4.View ArticlePubMedGoogle Scholar
- De Bonis M, Taramasso M, Grimaldi A, Maisano F, Calabrese MC, Verzini A, et al. The GeoForm annuloplasty ring for the surgical treatment of functional mitral regurgitation in advanced dilated cardiomyopathy. Eur J Cardiothorac Surg. 2011;40:488–95.PubMedGoogle Scholar
- Robb JD, Minakawa M, Koomalsingh KJ, Shuto T, Jassar AS, Ratcliffe SJ, et al. Posterior leaflet augmentation improves leaflet tethering in repair of ischemic mitral regurgitation. Eur J Cardiothorac Surg. 2011;40:1501–7.PubMedPubMed CentralGoogle Scholar
- Nouri M, Tokaldany ML, Shahrzad M, Mardanloo AS, Ahmadi SH, Roomi ZS, et al. Echocardiographic determinants of ischemic mitral regurgitation. J Card Surg. 2013;28:359–65.View ArticlePubMedGoogle Scholar
- Asai T, Kinoshita T, Hosoba S, Takashima N, Kambara A, Suzuki T, et al. Butterfly resection is safe and avoids systolic anterior motion in posterior leaflet prolapse repair. Ann Thorac Surg. 2011;92:2097–103.View ArticlePubMedGoogle Scholar
- David TE, David CM, Manlhiot C. Simplici-T annuloplasty band for mitral valve repair for degenerative disease. Ann Thorac Surg. 2014;98:1551–6.View ArticlePubMedGoogle Scholar
- Brown ML, Schaff HV, Li Z, Suri RM, Daly RC, Orszulak TA. Results of mitral valve annuloplasty with a standard-sized posterior band: is measuring important? J Thorac Cardiovasc Surg. 2009;138:886–91.View ArticlePubMedGoogle Scholar
- Calafiore AM, Di Mauro M, Iacò AL, Mazzei V, Teodori G, Gallina S, et al. Overreduction of the posterior annulus in surgical treatment of degenerative mitral regurgitation. Ann Thorac Surg. 2006;81:1310–6.View ArticlePubMedGoogle Scholar
- Calafiore AM, Iacò AL, Ibrahim A, Al-Amri H, Refaie R, Own A, et al. A novel and simple technique for correction of posterior leaflet prolapse due to chordal elongation or rupture. J Thorac Cardiovasc Surg. 2014;148:1407–12.View ArticlePubMedGoogle Scholar
- Duplessis LA, Marchand P. The anatomy of the mitral valve and its associated structures. Thorax. 1964;19:221–7.View ArticlePubMedGoogle Scholar