- Research article
- Open Access
- Open Peer Review
In-vivo assessment of the morphology and hemodynamic functions of the BioValsalva™ composite valve-conduit graft using cardiac magnetic resonance imaging and computational modelling technology
© Kidher et al.; licensee BioMed Central Ltd. 2014
- Received: 14 July 2014
- Accepted: 27 November 2014
- Published: 9 December 2014
The evaluation of any new cardiac valvular prosthesis should go beyond the classical morbidity and mortality rates and involve hemodynamic assessment. As a proof of concept, the objective of this study was to characterise for the first time the hemodynamics and the blood flow profiles at the aortic root in patients implanted with BioValsalva™ composite valve-conduit using comprehensive MRI and computer technologies.
Four male patients implanted with BioValsalva™ and 2 age-matched normal controls (NC) underwent cardiac magnetic resonance imaging (MRI). Phase-contrast imaging with velocity-mapping in 3 orthogonal directions was performed at the level of the aortic root and descending thoracic aorta. Computational fluid dynamic (CFD) simulations were performed for all the subjects with patient-specific flow information derived from phase-contrast MR data.
The maximum and mean flow rates throughout the cardiac cycle at the aortic root level were very comparable between NC and BioValsalva™ patients (541 ± 199 vs. 567 ± 75 ml/s) and (95 ± 46 vs. 96 ± 10 ml/s), respectively. The maximum velocity (cm/s) was higher in patients (314 ± 49 vs. 223 ± 20; P = 0.06) due to relatively smaller effective orifice area (EOA), 2.99 ± 0.47 vs. 4.40 ± 0.24 cm2 (P = 0.06), however, the BioValsalva™ EOA was comparable to other reported prosthesis. The cross-sectional area and maximum diameter at the root were comparable between the two groups. BioValsalva™ conduit was stiffer than the native aortic wall, compliance (mm2 • mmHg−1 • 10−3) values were (12.6 ± 4.2 vs 25.3 ± 0.4.; P = 0.06). The maximum time-averaged wall shear stress (Pa), at the ascending aorta was equivalent between the two groups, 17.17 ± 2.7 (NC) vs. 17.33 ± 4.7 (BioValsalva™ ). Flow streamlines at the root and ascending aorta were also similar between the two groups apart from a degree of helical flow that occurs at the outer curvature at the angle developed near the suture line.
BioValsalva™ composite valve-conduit prosthesis is potentially comparable to native aortic root in structural design and in many hemodynamic parameters, although it is stiffer. Surgeons should pay more attention to the surgical technique to maximise the reestablishment of normal smooth aortic curvature geometry to prevent unfavourable flow characteristics.
- Composite valve-conduit
- Aortic valve
- Aortic root
- Aortic prosthesis
- Computational fluid dynamic
Aortic root aneurysm, dissection and endocarditis associated with aortic valve dysfunction may be the most common examples of pathological conditions that often need concomitant replacement of aortic valve, aortic root and ascending aorta . Traditionally, the modified Bentall technique with mechanical valve conduit is the standard surgical procedure for such pathologies with excellent short and long-term results ,. The main drawback of the classical modified Bentall procedure is the use of mechanical valve which mandate the use of anticoagulants. In addition, the increasing number of elderly patients requiring modified Bentall procedure make the use of biological composite valve-conduit more appealing option. Hemodynamics is an essential aspect in designing any vascular or valve product. Stentless aortic valve prosthesis seems to have better hemodynamic performance than stented valves  and may improve coronary circulation up to the expected normal level . Stentless homografts have very good hemodynamic performance and durability but their use is limited by their availability , therefore stentless xenografts were designed and introduced to overcome these problems and maintain good hemodynamic profile similar to homograft aortic valves, with excellent mid-term results . Although stentless xenografts roots offer good choice and satisfactory outcome, their use is limited by their length as they cannot be employed when the ascending aorta needs to be replaced unless Dacron graft extension is sutured . This makes the procedure more complex, increasing the ischemic time and risk of bleeding from suture line.
Another important hemodynamic aspect in aortic valve prosthesis design is the presence or absence of sinuses of Valsalva . Sinuses of Valsalva reduce the stress in the leaflet significantly, facilitate vortex formation that seems to help in smooth valve closure with less bending deformation in the longitudinal direction, and reduce the stress on the coronary anastomosis -. All the above factors can affect the durability and performance of the aortic bio-prosthetic valves, however, the effect of sinuses of Valsalva on coronary circulation is still debatable ,.
All the above-mentioned limitations accelerated the designing and introduction of a new prefabricated composite bio-prosthetic valve–conduit graft (BioValsalva™) into the valve- conduit market since 2006. The BioValsalva™ (Vascutek Terumo, Renfrewshire, Scotland) is a composite valve-conduit graft which consists of stentless porcine aortic valve (elan valve™) pre-sewn to self-sealing trilaminated vascular graft (Vascutek Triplex™). It is a one-piece constructed graft making the procedure technically less complex, reducing cross-clamp and by-pass time and providing better haemostasis.
Preliminary data from Europe have been published - demonstrating good early post-operative results with low mortality rate and superior haemostatic properties. However, until date, no study has been conducted to assess the hemodynamic performance of the new generation of BioValsalva™ using advanced MRI and computer technologies. Cardiovascular magnetic resonance imaging (CMR) with phase-contrast velocity mapping and subsequent offline computational fluid dynamics (CFD) processing is a state-of-art effective imaging tool for the visualization of the streamlines and particle traces of the blood flow within the aortic root and thoracic aorta ,.
The objective of the study was to assess the hemodynamics and to characterise the blood flow profiles in patients with implanted BioValsalva™ composite valve-conduit, and compare these with matched normal controls based on detailed analysis of cardiac MRI data and patient-specific computational flow modelling as a proof of concept assessment.
Ethical approval (10/H0717/45, North London Research Ethic Committee 1) was granted and informed consent was obtained from all participants prior to inclusion. All patients with implanted BioValsalva™ pre-assembled composite valve-conduit graft were eligible. Age and gender matched healthy controls were recruited as normal control group (NC).
Cardiovascular magnetic resonance imaging
MRI for patients was performed after the recommended minimum safety period of four weeks post-operatively, mean (49 ± 11 days). A Philips Achieva 1.5 T scanner (Philips Achieva, Royal Philips, Amsterdam, The Netherlands) with a 32 channel cardiac coil was used for all studies. Two-dimensional phase contrast imaging was performed in the aortic root, at the level of the annulus, and in the descending thoracic aorta at the level of the pulmonary bifurcation. Double oblique sections were obtained to image the aorta in cross-section. The velocity-encoding value was chosen as 10% above peak velocity in the direction measured and 100 phases per cardiac cycle were obtained. Reconstructed voxel size was 1.4 × 1.4 × 10 mm. Anatomic imaging of the thoracic aorta was achieved using a navigator-gated balanced steady state free precession angiogram with a reconstructed voxel size of 0.5 × 0.5 × 2 mm.
Two -dimensional cine phase-contrast MRI of blood flow was performed in three orthogonal directions: feet-head, left-right and anterior-posterior separately on the selected imaging planes. The phase contrast images consisted of magnitude and phase images in pairs, and the magnitude images were utilized to identify and segment the anatomic region of the aortic root. The segmented region was then mapped to the corresponding phase image to calculate the velocity values in reference to encoded flow velocity. Velocity mapping was performed for all three components, and was converted to the through-plane as well as in-plane velocity maps over the cross section of aortic root. By integrating the through-plane velocity map over the cross-sectional area, the volumetric flow was calculated at each of the 100 time points; then the temporal volumetric flow waveform was reconstructed for a complete cardiac cycle. In this study, the temporal maximum volumetric flow rate (Q max ), the time-averaged volumetric flow rate (Q mean ) and the spatial maximum velocity at peak flow rate (V max ) were examined. Additionally, the effective orifice area, EOA (A orifice ) was calculated, which is defined as the area where the velocities are larger than the spatial averaged velocity at peak flow rate . The ratio of the EOA to sinus area (A orifice \A sinus ) was also calculated to further assess the function of aortic valves in both control and patient groups. A MATLAB program (The MathWorks, Inc, Natick, MA, USA) was developed to carry out all the analysis of phase contrast MR images in this study.
Aortic wall compliance
The blood pressure at the root of aorta was measured by a novel device Pulsecor (Pulsecor Ltd, Auckland, New Zealand) that estimates central aortic pressures from the brachial cuff pressure fluctuations . It employs a physics-based model involving pressure wave reflection simulation to reconstruct the central pressure waveform according to the small oscillations of the intra-arterial pressure . The Pulsecor device has been evaluated to be an accurate non-invasive method to estimate central aortic pressure . The pressure of each subject was measured by Pulsecor device just before their MR scans, and the derived maximum systolic (P s) and minimum diastolic (P d) pressure were utilized for the calculation of compliance, distensibility and wall stiffness of the aortic root.
Computational fluid dynamic analysis
Based on the reconstructed three-dimensional geometry of the aortas, computational fluid dynamic (CFD) models were built for all the subjects in normal and patient groups. The patient-specific flow information obtained from phase-contrast MR data and the aortic pressure data measured with the Pulsecor device was applied to the computational model as boundary conditions. The three-directional encoded velocity profiles were mapped to the cross section of the aortic root in the model both spatially and temporally. The aortic pressure waveform was applied to the model outlet at the middle of descending aorta. The flow exiting through the three branches on the aortic arch was determined by the flow difference between the two MR imaging planes at the ascending and thoracic descending aorta. Blood was treated as a Newtonian and incompressible fluid with a dynamic viscosity of 4.0 mPa and a density of 1060 kg/m3, and the aortic walls were assumed to be rigid with no slip condition. The 3D geometries were exported into ICEM CFX (ANSYS) for mesh generation. The computational mesh consisted of 3D hexahedral cells with higher resolution in the fluid boundary layer at the wall, and mesh sensitivity test was carried out to ensure mesh independence of the simulation results.
The Navier–Stokes equations were solved numerically with a commercial finite-volume-based CFD solver (ANSYS CFX 14). The laminar-turbulent transitional version of the hybrid shear stress transport turbulence model , which incorporates empirical correlations to cover a range of transition mechanisms that lead to turbulent (chaotic) flow, was employed in the simulations to capture possible transitional flow in the aorta.
All simulations were carried out for three cardiac cycles to achieve a periodic solution with optimized time step and convergence criteria, and the results presented here were obtained in the third cycle. Results were analysed using post-processing software ENSIGHT 10 (CEI Inc, NC, USA). One of the most important parameters, time-averaged wall shear stress (TAWSS), was calculated by averaging the time-varying wall shear stress (WSS) over the cardiac cycle.
Formal statistical analysis was not clinically important in such proof of concept study with small number of subjects. However for descriptive purpose, statistical analysis was performed using IBM SPSS Statistics software version 20.0 (IBM Corp., Armonk, NY, USA) and Microsoft Excel 2010. Participants’ characteristics were expressed as mean ± standard deviation (continuous variables) and as frequencies (categorical variables). Statistical significance was considered where P ≤ 0.05. Comparative analysis between patients and control groups was carried out using non- parametric independent samples test (Mann–Whitney U) for continuous variables, and Fisher’s exact test for categorical variables.
Demographic and clinical characteristics
Control (n = 2)
Patients (n = 4)
Male [n (%)]
62 ± 4
66 ± 5
Age range (years)
White Caucasian [n (%)]
DM [n (%)]
Hypertension [n (%)]
Statin treatment [n (%)]
30 ± 1.8
28.3 ± 2.8
68 ± 13
64 ± 7
131 ± 1
129 ± 8
79 ± 1
68 ± 14
53 ± 2
61 ± 21
96 ± 1
88 ± 7
183 ± 5
176 ± 3
101 ± 11
90 ± 12
PVD [n (%)]
Bicuspid aortic valve
NYHA (pre-op.) [n (%)]
NYHA (post-op.) [n (%)]
Indication for surgery, dilated aortic root and
Aortic stenosis and regurgitation
CPB time (minutes)
190 ± 51
Inotropes [n (%)]
Atrial Fibrillation [n (%)]
> 24 intubation
Hospital stay (days)
10.5 ± 2.6
Control (n = 2)
Patients (n = 4)
A mean (cm2), ascending aorta
8.1 ± 2.2
6.5 ± 0.31
D max (cm), ascending aorta
3.2 ± 0.28
2.9 ± 0.12
A sinus (cm2), aortic sinus
8.9 ± 1.6
7.9 ± 0.9
D sinus (cm), aortic sinus
3.36 ± 0.3
3.18 ± 0.15
Q max (ml/s)
541 ± 199
567 ± 75
Q mean (ml/s)
95 ± 46
96 ± 10
V max (cm/s)
223 ± 20
314 ± 49
A orifice (cm2)
4.40 ± 0.24
2.99 ± 0.47
A orifice \A sinus
0.54 ± 0.03
0.42 ± 0.04
17.17 ± 2.7
17.33 ± 4.7
Compliance (mm2 • mmHg−1 • 10−3)
25.3 ± 0.4
12.6 ± 4.2
Distensibility (mmHg−1• 10−3)
3.33 ± 1.03
1.7 ± 0.62
Stiffness index β
0.23 ± 0.03
0.43 ± 0.11
Flow rate and velocity
Spatiotemporal velocity profile
Effective orifice area
Aortic wall elasticity
Native aortic wall is more elastic than the BioValsalva™ conduit with twofold compliance (mm2 • mmHg−1 • 10−3) value (25.3 ± 0.4 vs. 12.6 ± 4.2). Superior elastic properties of the native aortic wall were also confirmed by other aortic stiffness indices such as distensibility and stiffness index as shown in Table 2.
Wall shear stress (WSS)
Blood flow pattern
Our literature search could not identify any study investigating BioValsalva™ functional performance using similar methodology. Therefore there were no previous results to compare with. The number of studies investigated BioValsalva™ is limited and focusing mainly on mortality and operative morbidity rates -,. Few of them have performed routine post-operative trans-thoracic echocardiography to assess cardiac function, trans-valvular gradient paravalvular leak, and valvular regurgitation ,,. Two studies assessed the EOA of earlier generation, one use CT scan , and the other used trans-thoracic echocardiography .
The design of valvular or vascular prosthesis is usually inspired by the design of the native tissue to achieve maximum possible similarity in structure and function. The availability of advanced imaging and computational technology makes the assessment of the degree of similarity between such prosthesis and normal tissue achievable. From structural design point, BioValsalva™ composite valve-conduit graft is comparable to native tissue in terms of biological valve, tricuspid design, stentless valve frame, coronary sinuses and proportionally sized conduit. From functional point, they seem very comparable to normal tissue in term of maximum flow rate (ml/s), mean flow rate, flow rate waveform (Table 2 and Figure 2). In addition, cross-sectional area and maximum diameter of aortic root and ascending aorta, were also comparable between the two groups (to some extent higher in the control group).
However, though the overall cross-sectional area and maximum diameter at the sinus level was comparable between the two groups which may indicate correct intra-operative prosthetic size selection, the EOA (A orifice ) at peak flow rate was smaller in BioValsalva™ patients (4.40 ± 0.24 vs. 2.99 ± 0.47 cm2). This difference might be inevitable due the space occupied by the prosthetic valve frame and suturing mattress as indicated by the wide margin between the effective orifice and the edge of the sinus (Figure 4), it may also suggest that BioValsalva™ valve opening is not free and wide as the normal valve. Other factor that might contribute to the difference in EOA is the body size, the height (cm) of healthy control was markedly taller than the patients (183 ± 5 vs. 176 ± 3), which may also explain the differences in the aortic cross-sectional area and aortic maximum diameter (A mean : D max.; A sinus and D sinus ). Two previous studies assessed the EOA of earlier generation, the results of both were smaller than what we found using MRI scan in newer generation of the prosthesis. The EOA was 2.2 ± 0.3 and 2.4 ± 0.4 cm2 for the 25- and 27-mm prostheses when assessed with CT scan , and 1.9 ± 0.3 cm2 when assessed with trans-thoracic echocardiography .
However, the global hemodynamic performance of the BioValsalva™ was not affected by this difference in the EOA. Smaller EOA has caused higher maximum velocity (cm/s) in the patients group (314 ± 49 vs. 223 ± 20), as a small orifice produces high-velocity jet into the aortic root, thus, the spatiotemporal velocity profile of the BioValsalva™ was steeper and faster than the control (Figure 2).
Another difference was in the elastic property of the vascular (aortic) conduit. As expected, the BioValsalva™ conduit is stiffer than the native aortic wall which has higher compliance and distensibility. This is because native aorta is a viable tissue and, histologically, particularly at the proximal location, is rich in elastin tissue . Whether or not the rest of the normal aorta can counterbalance this stiffness to deliver steady normal flow at the periphery need to be assessed in future study.
Flow streamlines demonstrated that the flow pattern was also comparable between the two groups apart from the more disturbed and helical flow that occurs at the outer curvature of the nearly right-angle turn at the ascending aorta level. From the site and level (high above the valve) of this helical flow, it seems to be related to the sharp angle in the flow pathway, makes the surgical technique more accountable to this difference than the prosthetic design. Though the TAWSS max (Pa), were also comparable between the two groups, the spatial distribution of areas with high wall shear stress was different between the two groups. By examining the close correlation between flow streamlines and TAWSS maps (Figures 5 and 6), the difference in the TAWSS could be due to: the difference in the maximum flow velocity (high-speed jet in the BioValsalva™) through the valve would result in higher shear stress on the aortic wall; the development of nearly right-angle turn at the ascending aorta near the anastomosis (suture) line which might cause more disturbance to the aortic blood flow.
Our findings are in line with previously published results using similar technology in normal controls and xenograft aortic root replacement patients (Medtronic Freestyle, Medtronic Inc, Minneapolis, MN, USA), which validates our results and methodology . The shapes of the flow rate waveforms were also very similar between the studies. Interestingly, the mean EOA was also comparable between the two studies for both patients and healthy controls, and the spatiotemporal velocity map of our control and BioValsalva™ were also similar to the published normal and xenograft respectively .
One could argue the small number of participants recruited, however, considering the proof of concept nature of the study and the significant time and effort needed for imaging data reconstruction and computational modelling, bigger number of subjects would not be necessary. Though unlikely to change the general direction of the results, coincidently, all BioValsalva™ patients were male, therefore the effect of gender on the results cannot be confirmed. Another limitation is that coronary flow rate was not assessed, this could shed some light on whether the difference in aorta geometry and conduit compliance can affect coronary circulation.
Combined Cardiac MRI and Computational Modelling Technology is an effective method of assessing valvular and vascular prosthesis from structural and functional point of view and it should be considered as a part of validation assessment of any new prosthesis. BioValsalva™ composite valve-conduit prosthesis is potentially comparable to native aortic valve and aorta in structural design and in many functional hemodynamic patterns, further larger study to confirm these finding is desired. The functional differences between BioValsalva™ and normal aortic valve are mainly in the elasticity of the vascular conduit, therefore future design development should focus on this aspect. Surgeons should pay more attention to the surgical technique they use in sizing and suturing to maximise the reestablishment of normal smooth aortic curvature to prevent unfavourable flow characteristics. Future larger studies using such technology should also consider exploring the clinical impact of unfavourable hemodynamic parameters and the effect of proximal hemodynamic on distal circulation.
Emaddin Kidher and Zhuo Cheng Sharing first authorship.
Institution of work: Imperial College London and Imperial College Healthcare NHS Trust, London, UK.
This work was supported by Vascutek Terumo, Renfrewshire, Scotland [Grant Number P34939] and the National Institute for Health Research (NIHR) Imperial Biomedical Research Centre (BRC) [Grant Number P50508].
- Zannis K, Deux JF, Tzvetkov B, Nakashima K, Loisance D, Rahmouni A, Kirsch ME: Composite freestyle stentless xenograft with Dacron graft extension for ascending aortic replacement. Ann Thorac Surg. 2009, 87: 1789-1794. 10.1016/j.athoracsur.2009.03.069.View ArticlePubMedGoogle Scholar
- Hagl C, Strauch JT, Spielvogel D, Galla JD, Lansman SL, Squitieri R, Bodian CA, Griepp RB: Is the Bentall procedure for ascending aorta or aortic valve replacement the best approach for long-term event-free survival?. Ann Thorac Surg. 2003, 76: 698-703. 10.1016/S0003-4975(03)00568-X.View ArticlePubMedGoogle Scholar
- Zehr KJ, Orszulak TA, Mullany CJ, Matloobi A, Daly RC, Dearani JA, Sundt TM, Puga FJ, Danielson GK, Schaff HV: Surgery for aneurysms of the aortic root: a 30-year experience. Circulation. 2004, 110: 1364-1371. 10.1161/01.CIR.0000141593.05085.87.View ArticlePubMedGoogle Scholar
- Borger MA, Carson SM, Ivanov J, Rao V, Scully HE, Feindel CM, David TE: Stentless aortic valves are hemodynamically superior to stented valves during mid-term follow-up: a large retrospective study. Ann Thorac Surg. 2005, 80: 2180-2185. 10.1016/j.athoracsur.2005.05.055.View ArticlePubMedGoogle Scholar
- Bakhtiary F, Schiemann M, Dzemali O, Dogan S, Schachinger V, Ackermann H, Moritz A, Kleine P: Impact of patient-prosthesis mismatch and aortic valve design on coronary flow reserve after aortic valve replacement. J Am Coll Cardiol. 2007, 49: 790-796. 10.1016/j.jacc.2006.10.052.View ArticlePubMedGoogle Scholar
- Yacoub M, Rasmi NR, Sundt TM, Lund O, Boyland E, Radley-Smith R, Khaghani A, Mitchell A: Fourteen-year experience with homovital homografts for aortic valve replacement. J Thorac Cardiovasc Surg. 1995, 110: 186-193. 10.1016/S0022-5223(05)80025-X.View ArticlePubMedGoogle Scholar
- Melina G, De Robertis F, Gaer JA, Amrani M, Khaghani A, Yacoub MH: Mid-term pattern of survival, hemodynamic performance and rate of complications after medtronic freestyle versus homograft full aortic root replacement: results from a prospective randomized trial. J Heart Valve Dis. 2004, 13: 972-975.PubMedGoogle Scholar
- Beck A, Thubrikar MJ, Robicsek F: Stress analysis of the aortic valve with and without the sinuses of valsalva. J Heart Valve Dis. 2001, 10: 1-11.PubMedGoogle Scholar
- Katayama S, Umetani N, Sugiura S, Hisada T: The sinus of Valsalva relieves abnormal stress on aortic valve leaflets by facilitating smooth closure. J Thorac Cardiovasc Surg. 2008, 136: 1528-1535. 10.1016/j.jtcvs.2008.05.054.View ArticlePubMedGoogle Scholar
- Weltert L, De Paulis R, Scaffa R, Maselli D, Bellisario A, D'Alessandro S: Re-creation of a sinuslike graft expansion in Bentall procedure reduces stress at the coronary button anastomoses: A finite element study. J Thorac Cardiovasc Surg. 2009, 137: 1082-1087. 10.1016/j.jtcvs.2008.10.013.View ArticlePubMedGoogle Scholar
- de Paulis R, Tomai F, Bertoldo F, Ghini AS, Scaffa R, Nardi P, Chiariello L: Coronary flow characteristics after a Bentall procedure with or without sinuses of Valsalva. Eur J Cardiothorac Surg. 2004, 26: 66-72. 10.1016/j.ejcts.2004.04.031.View ArticlePubMedGoogle Scholar
- Bara CL, Verhey JF: Simulation of the fluid dynamics in artificial aortic roots: comparison of two different types of prostheses. J Artif Organs. 2008, 11: 123-129. 10.1007/s10047-008-0416-5.View ArticlePubMedGoogle Scholar
- Di Bartolomeo R, Botta L, Leone A, Pilato E, Martin-Suarez S, Bacchini M, Pacini D: Bio-Valsalva prosthesis: 'new' conduit for 'old' patients. Interact Cardiovasc Thorac Surg. 2008, 7: 1062-1066. 10.1510/icvts.2008.187849.View ArticlePubMedGoogle Scholar
- Kaya A, Heijmen RH, Kelder JC, Morshuis WJ: First 102 patients with the BioValsalva conduit for aortic root replacement. Ann Thorac Surg. 2012, 94: 72-77. 10.1016/j.athoracsur.2012.03.032.View ArticlePubMedGoogle Scholar
- Bochenek-Klimczyk K, Lau KK, Galinanes M, Sosnowski AW: Preassembled stentless valved-conduit for the replacement of the ascending aorta and aortic root. Interact Cardiovasc Thorac Surg. 2008, 7: 964-968. 10.1510/icvts.2008.184457.View ArticlePubMedGoogle Scholar
- Lau KK, Bochenek-Klimczyk K, Galinanes M, Sosnowski AW: Replacement of the ascending aorta, aortic root, and valve with a novel stentless valved conduit. Ann Thorac Surg. 2008, 86: 278-281. 10.1016/j.athoracsur.2008.01.058.View ArticlePubMedGoogle Scholar
- Giannakoulas G, Dimopoulos K, Xu XY: Modelling in congenital heart disease. Art or science?. Int J Cardiol. 2009, 133: 141-144. 10.1016/j.ijcard.2008.10.039.View ArticlePubMedGoogle Scholar
- Torii R, El-Hamamsy I, Donya M, Babu-Narayan SV, Ibrahim M, Kilner PJ, Mohiaddin RH, Xu XY, Yacoub MH: Integrated morphologic and functional assessment of the aortic root after different tissue valve root replacement procedures. J Thorac Cardiovasc Surg. 2012, 143: 1422-1428. 10.1016/j.jtcvs.2011.12.034.View ArticlePubMedGoogle Scholar
- Lowe A, Harrison W, El-Aklouk E, Ruygrok P, Al-Jumaily AM: Non-invasive model-based estimation of aortic pulse pressure using suprasystolic brachial pressure waveforms. J Biomech. 2009, 42: 2111-2115. 10.1016/j.jbiomech.2009.05.029.View ArticlePubMedGoogle Scholar
- Lin AC, Lowe A, Sidhu K, Harrison W, Ruygrok P, Stewart R: Evaluation of a novel sphygmomanometer, which estimates central aortic blood pressure from analysis of brachial artery suprasystolic pressure waves. J Hypertens. 2012, 30: 1743-1750. 10.1097/HJH.0b013e3283567b94.View ArticlePubMedGoogle Scholar
- Xu J, Shiota T, Omoto R, Zhou X, Kyo S, Ishii M, Rice MJ, Sahn DJ: Intravascular ultrasound assessment of regional aortic wall stiffness, distensibility, and compliance in patients with coarctation of the aorta. Am Heart J. 1997, 134: 93-98. 10.1016/S0002-8703(97)70111-X.View ArticlePubMedGoogle Scholar
- Cheng Z, Tan FP, Riga CV, Bicknell CD, Hamady MS, Gibbs RG, Wood NB, Xu XY: Analysis of flow patterns in a patient-specific aortic dissection model. J Biomech Eng. 2010, 132: 051007-10.1115/1.4000964.View ArticlePubMedGoogle Scholar
- Moorjani N, Modi A, Mattam K, Barlow C, Tsang G, Haw M, Livesey S, Ohri S: Aortic root replacement using a biovalsalva prosthesis in comparison to a "handsewn" composite bioprosthesis. J Card Surg. 2010, 25: 321-326. 10.1111/j.1540-8191.2010.01015.x.View ArticlePubMedGoogle Scholar
- Lee RT, Kamm RD: Vascular mechanics for the cardiologist. J Am Coll Cardiol. 1994, 23: 1289-1295. 10.1016/0735-1097(94)90369-7.View ArticlePubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.